Hospital bed

ABSTRACT

A bed having a frame and a deck is disclosed. The deck has at least a horizontal position and a chair position. The deck has an adjustable-length foot section.

RELATED APPLICATIONS

This application is a continuation of U.S. application Ser. No.12/351,089, filed Jan. 9, 2009, projected U.S. Pat. No. 7,784,128, whichis a continuation of U.S. patent application Ser. No. 11/734,954, filedApr. 13, 2007, now U.S. Pat. No. 7,480,951, which is a continuation ofU.S. patent application Ser. No. 11/393,604, filed Mar. 30, 2006, nowU.S. Pat. No. 7,213,279, which is a continuation of U.S. patentapplication Ser. No. 11/353,743, filed Feb. 13, 2006, now U.S. Pat. No.7,237,287, which is a continuation of U.S. patent application Ser. No.10/028,833, filed Dec. 20, 2001, now U.S. Pat. No. 7,017,208, which is acontinuation of U.S. patent application Ser. No. 09/655,127, filed Sep.5, 2000, now U.S. Pat. No. 6,336,235, which is a continuation of U.S.patent application Ser. No. 09/018,542, filed Feb. 4, 1998, now U.S.Pat. No. 6,163,903, which is a continuation of U.S. patent applicationSer. No. 08/511,711, filed Aug. 4, 1995, now U.S. Pat. No. 5,715,548,the disclosures of all of which are expressly incorporated by referenceherein.

BACKGROUND AND SUMMARY OF THE INVENTION

The present invention relates to a bed, and particularly to a chair bedthat can be manipulated to achieve both a conventional bed positionhaving a horizontal sleeping surface upon which a person lies in asupine position and a sitting position having the feet of the person onor adjacent to the floor and the head and back of the person supportedabove a seat formed by the bed. More particularly, the present inventionrelates to a hospital bed or a patient-care bed which is convertible toa chair and which is configured to facilitate several activities thatmay be performed by a caregiver for a person on the sleeping surface ofthe bed.

Many hospital beds are positionable to a configuration having thesleeping surface of the bed at a predetermined height above the floorand having siderails positioned to restrain the movement of a personlying on the sleeping surface past sides of the sleeping surface and offof the bed. The sleeping surfaces of many such hospital beds cantypically be lowered to reduce the distance between the sleeping surfaceand the floor, and the sleeping surfaces of such beds can often bemanipulated to adjust the position of the person on the sleepingsurface. In addition, the siderails of these hospital beds can typicallybe moved to a position away from the sleeping surface to facilitatemovement of the person on the sleeping surface from the supine positionon the sleeping surface to a standing position on the floor near thebed.

It may sometimes be desirable to have two caregivers assist a persontrying to move from the supine position on the sleeping surface of suchhospital bed to the standing position. After moving the siderails awayfrom the sleeping surface, the caregivers may pivot the person so thatthe legs of the person hang over the side of the sleeping surface. Thecaregivers can then assist the person as the person slides past one sideof the sleeping surface until the feet of the person touch the floor.The caregivers typically hold the person firmly while also bracingthemselves to prevent a fall or other injury to the person as the personstands.

Beds and examination tables having articulating decks to adjust theposition of the person on the surface are known in the art. See, forexample, U.S. Pat. No. 5,077,843 to Foster L. Dale et al. and U.S. Pat.No. 4,751,754 to Baily et al., which are assigned to the assignee of thepresent invention, and U.S. Pat. No. 3,281,141 to Smiley et al. andGerman publication No. 716981. Each of these references discloses a bedor an examination table having a top surface that articulates to adjustthe position of the person on the surface.

In addition, beds and examination tables that are convertible to chairsin order to simplify the task of moving a person on the sleeping surfacefrom the supine position to the standing position are known in the art.See, for example, U.S. Pat. No. 5,157,800 to Borders, U.S. Pat. No.5,129,177 to Celestina et al., and U.S. Pat. No. 4,862,529 to Peck, allof which are assigned to the assignee of the present invention, and U.S.Pat. No. 5,279,010 to Ferrand et al., U.S. Pat. No. 4,183,109 to Howell,U.S. Pat. No. 4,411,035 to Fenwick, and U.S. Pat. No. 3,220,022 toNelson. Each of these references discloses a bed that can be convertedto a chair-like configuration.

What is needed is a bed that can be converted to a chair and that canalso facilitate activities that are typically performed by caregivers.For example, caregivers would welcome a bed that, in addition to beingconvertible to a chair, can be configured to weigh a person, to rapidlymove the person from an upright position to a generally horizontalposition when emergency procedures are initiated, and that canfacilitate varied procedures that may be performed on a person carriedby the bed, thereby reducing the number of times the person istransferred from one bed or surface to another.

According to the present invention, a chair bed for a person isprovided, the bed having a head end, a foot end, and sides. The bedincludes a base frame, an intermediate frame coupled to the base frame,a weigh frame coupled to the intermediate frame, and an articulatingdeck coupled to the weigh frame. A plurality of load cell supportscouple the weigh frame to the intermediate frame. The load cell supportsinclude means for determining the weight of objects supported by theweigh frame. Alternatively, the weigh frame and the intermediate framemay be fixed together to form a common frame or may be replaced by asingle common frame.

In preferred embodiments, the articulating deck has longitudinallyspaced head, seat, thigh, and foot sections. The head, thigh, and footsections are movable relative to each other and are movable relative tothe seat section which is fixed relative to the weigh frame. The head,thigh, and foot sections are infinitely adjustable to allow the bed toattain any desired position within the range of movement of the head,thigh, and foot sections, thus accommodating changes of position of aperson on the bed. Of course, the articulating deck can provide aplanar, horizontal sleeping surface, a planar sleeping surface that istilted toward either the head end of the bed or the foot end of the bed,and a non-planar chair-shaped seating surface, in addition to theintermediate positions therebetween.

The bed can include a mechanism for raising and lowering thearticulating deck and the sleeping surface between a low position and araised position relative to the base of the bed. In addition, the bedcan also include mechanisms for independently raising and lowering eachof the head section, the thigh section, and the foot section so that thebed can assume many positions to suit the specific needs of individuals.

The sleeping surface of the chair bed can also be easily moved between agenerally horizontal position and a Trendelenburg position. When in theTrendelenburg position, the intermediate frame is tilted such that thehead end of the sleeping surface is higher than the foot end. Thesleeping surface can also be moved to a reverse Trendelenburg positionhaving the foot end above the head end.

The power required to raise and lower the sleeping surface and to movethe head, thigh, and foot sections can be provided by a primaryelectrical power source such as the main AC power supply of a hospitalin combination with a fluid system. In addition, a foot pedal can bepivotably mounted to the base frame and coupled to the bed to allow thecaregiver to manually provide power to the bed for remote operation ofthe fluid system of the bed.

In one preferred embodiment of the bed, the articulating deck and themechanism for raising and lowering the articulating deck are powered bya fluid system having a pump, valves, and actuators for positioning theintermediate frame relative to the base frame and for positioning thearticulating deck sections. The actuators must be supplied withpressurized fluid to manipulate the bed and the valves control the flowof fluid to the actuators.

The bed is additionally provided with an electrical control system foroperating a pump and controlling control valves. The electrical controlsystem controls the operation of the bed when the bed is connected to anindependent source of power. In addition, the electrical control systemincludes a battery source for operating the valves when the bed is notconnected to an independent source of power.

The articulating deck can be a longitudinal step deck that includesupper deck side portions and a central, longitudinally extending recessbetween the side portions. The recess is defined by a lower deck andside walls connecting the lower deck and the upper portion of the deck.

A companion mattress is carried by the deck. The mattress has a planar,upwardly-facing sleeping surface, side portions resting on the side deckportions, and a projection beneath the sleeping surface. The projectionextends downwardly into the recess and can engage at least a portion ofthe side wall of the deck. The varied thickness of the mattress providesthe mattress with “zones” including a thick zone adjacent to theprojection and a thin zone in areas away from the projection. Themattress includes a head mattress portion, a seat mattress portion, athigh mattress portion, and a foot mattress portion. Each named mattressportion is associated respectively with the head, seat, thighs, and feetof the person resting on the sleeping surface of the bed as well as withthe underlying head, seat, thigh, and foot sections of the deck.

A step deck and a mattress configured for use with a step deck can beused independently of the bed and the articulating deck. For example, astep deck can be provided for a stretcher. Such a stretcher, forexample, can include a frame, a step deck mounted to the frame, the stepdeck having longitudinal upper side portions engaging the frame, and amattress having a generally planar sleeping surface and a bottom surfaceincluding a projection configured so that the shape of the bottomsurface generally conforms to the shape of the step deck. In the samemanner, a step deck and associated mattress could be provided for agurney. Such a gurney would be similar to the illustrative stretcherdescribed above except that the frame would include wheels so that thegurney could be transported by rolling it from place to place.

Though there are many potential variations of step deck shapes andcorresponding mattress shapes and numbers and types of mattress piecesthat could be devised, any step deck having an upper deck portion and arecess defined by a bottom deck portion and walls connecting the bottomand the upper deck portions would achieve the desired results. Likewise,any mattress or combination of mattress pieces that provide a bottomsurface generally conforming to the shape of the step deck would achievethe desired results.

The movable deck head section is coupled to the intermediate frame andis configured to pivot relative to the weigh frame about an effectivepivot axis positioned to lie above the lower deck portion. Preferably,the effective pivot axis is located generally adjacent to a pivot axisdefined by the hip of a person lying on the sleeping surface in order tominimize the shear between the sleeping surface and the back of theperson in the bed as the head section moves between the down positionand the back-support position. To achieve this “reduced-shear pivot,”the head section is mounted to the weigh frame for both translationalmovement and pivoting movement relative to the weigh frame. The pivotingand translational movements combine to produce a motion in which thehead portion pivots relative to the frame about the effective pivotaxis.

The head section of the articulating deck can pivot relative to theintermediate frame between a down position generally parallel to theweigh frame and an upward back-support position. When a person on asleeping surface moves from a flat position to a sitting position, theback and legs of the person engaging the sleeping surface lengthen. Thereduced-shear pivot accommodates this lengthening to reduce the shearbetween the back and legs of the person and the sleeping surface as thehead section pivots between the down position and the back-supportposition by expanding the deck and the sleeping surface. Thereduced-shear pivot allows the upper body of the person on the sleepingsurface to be tilted upwardly without moving the lower body of theperson. This reduces the tendency of the person to slide relative to thesleeping surface during articulation of the head section, therebyreducing the shear between the back and legs of the person and thesleeping surface.

The step deck can also include an upper deck end portion adjacent to thefoot end of the bed. The foot section can be coupled to the upper deckend portion for pivoting movement about a pivot axis that is positionedto lie above the lower deck. The foot section can also be configured tocontract and expand to vary the length of the foot section as the footsection pivots about the pivot axis so that the foot section can pivotdownwardly when the bed is in the low position to place the feet of theperson supported on the sleeping surface on the floor. In addition, thefoot portion of the mattress can be configured to shorten in conjunctionwith the contraction of the foot section. Also, the seat and footportions of the mattress can automatically become thinner to maintain anappropriately sized seat area as the foot section pivots downwardly.

A pair of siderails can be provided on each side of the bed. Each pairof siderails includes a head section siderail that is movable with thepivoting head section of the deck and a body section siderail that ismovable with the seat section. Each siderail is preferably maintained ina generally vertical orientation adjacent to the sides of the bed.

The siderails are each movable between an upward patient-restrainingposition restraining the movement of a person supported on the sleepingsurface past the sides of the sleeping surface and a downward tuckedposition. When in the patient-restraining position, the bottoms of thesiderails are positioned to lie above the upper deck side portions andthe siderails abut the sides of the mattress. When in the tuckedposition, the tops of the siderails are positioned to lie beneath theupper deck side portions in a niche defined by the upper deck sideportions and the side walls connecting the lower deck to the upper deckside portions.

When moving between the patient-restraining position and the tuckedposition, the siderails rotate outwardly and downwardly from thepatient-restraining position away from the side of the bed, and thenback inwardly and downwardly to the tucked position beneath the upperdeck portion. The siderails trace the same path when moving from thetucked position to the patient-restraining position. The rotatingmechanism, which holds the siderails in vertical orientations parallelto the side of the mattress through the entire range of movement, placesthe siderails against the sides of the mattress when the siderails arein the patient-restraining position, allowing for the provision of anarrower bed than would otherwise be provided, even though the mattressof the bed has a standard width.

The body section siderails are coupled to brackets that are fixed to theframe. The head section siderails are coupled to brackets fixed to theside wall of the deck. However, the bed may be provided with breakawayhead section siderails, each head section siderail being mounted on apivotable collateral deck section to move the siderail from the tuckedposition to a generally vertically downwardly extendingdown-out-of-the-way position, preferably extending downwardly along thehead of the bed to provide clear access to space beneath theintermediate frame. Breakaway siderails provide the caregiver with evengreater access to the space under the sleeping surface of the bed whenthe siderails are in the down-out-of-the-way position while alsoimproving access across the sleeping surface for equipment that may bedesired for use on a person on the sleeping surface.

The head section and body section siderails are configured to maintain abetween-rail gap of approximately 2-3 inches as the head sectionsiderail moves relative to the body section siderail. Also in preferredembodiments, the head section siderails are shorter than the bodysection siderails and shorter than typical siderails making it possiblefor a person to enter the bed from the side using the head sectionsiderail as a support and to properly position their hip on the sleepingsurface.

The bed can also be used to monitor the total weight of objects,including a person, carried by the bed. The bed is configured so thathospital equipment such as IV poles can be attached to the bed such thatthe weight of these items is not included in the weight monitored by thebed, thereby facilitating the convenient use of such equipment.

In this specification and in the claims, such terms as “chair bed,”“hospital bed,” “patient-care bed,” and “examination table” are used ina general sense and not in a limiting sense. The bed of the presentinvention has wide application and may be used in a variety ofsituations. The improvements disclosed herein may be used on beds ingeneral, on medical tables, stretchers, gurneys, and so forth asappropriate. However, the bed of the present invention providessignificant improvements in caregiver productivity and patient outcomes.The following capabilities are included in the bed of the presentinvention:

1. Full chair capability as a built-in feature, enabling a singlecaregiver to execute an order from a physician to place the person in achair by operating controls to convert the bed into a chair while thepatent is on the bed.

2. Infinitely adjustable head, thigh, and foot section angles that allowany desired position of the head, thigh, and foot section with in therange of movement of those sections.

3. Foot egress capability, enabling a single caregiver to transfer aperson from the bed to a wheelchair or ambulate the patient.

4. Modular surface capability, allowing a single caregiver to adapt themattress surface to provide decubitus prevention and pulmonary treatmentcapability as the needs of the person change.

5. In-bed weighing capability, enabling a caregiver to monitor theweight of a person on the sleeping surface.

6. Electric or manual Trendelenburg capability, allowing the sleepingsurface to assume the Trendelenburg position having the feet of theperson carried by the bed slightly elevated above the head. In addition,the sleeping surface can assume a reverse Trendelenburg position havingthe head of the person slightly elevated above the feet.

7. Reduced-shear pivot capability, causing the articulating sections tominimize shear forces between the sections and the person so that theperson does not slide appreciably relative to the sleeping surface ofthe mattress as the articulating sections pivot.

8. Siderails having a one-handed release mechanism and that rotateoutwardly and rotate from the patient-restraining position to a tuckedposition underneath the patient-restraining position.

9. Breakaway siderail capability, having a pin that can be removed fromthe foot end of each head section siderail after the siderail has beenrotated under to the tucked position so that the foot end of the headrail pivots downwardly and toward the head end of the bed, therebyproviding a caregiver with additional access to the sleeping surface,additional clearance around the deck so that the caregiver has improvedaccess to the patient, and allowing for the insertion of a C-armlaterally further across the person than may be achieved without thebreakaway siderail capability.

10. CPR foot pedal capability, enabling rapid movement of the headsection from the upward sitting position to the horizontal CPR positionby activation of a CPR pedal so that the caregiver has instantaneouscontrol with one-step operation that requires constant activation.

11. Manual pump articulation and a battery for controlling the valves,allowing the caregiver to raise or lower the sleeping surface or thehead, thigh, and foot sections of the articulating deck and allowingmovement of the sleeping surface to the Trendelenburg position duringremote operation of the bed when the bed is disconnected from externalpower sources.

12. In-bed X-ray capability including a radiolucent window and head andabdomen access for a separate C-arm of X-ray equipment allowing thecaregiver to obtain X-rays of the head, chest, and abdomen withoutremoving the person from the sleeping surface of the bed.

13. Four wheel braking capability for braking four caster wheels mountedto the base, the braking capability being activated by a butterfly wheelpedal that can move between a braking position, a neutral position, anda steering position that allows for the steering of the caster wheelsduring movement of the bed.

14. Mattress including pressure-reducing capability in the seat sectionwhen in the sitting position to reduce the chance of skin breakdown.

15. Mattress including a modular design so that several surfaces/airtherapies are driven by a common air source, a common graphicalcaregiver interface, and a common distributed network, and so that acaregiver can install and initiate an air therapy without moving theperson off of the original sleeping surface.

The bed of the present invention includes a base frame or a main frameupon which several basic components are mounted such as the systemdisplays and the air compressor for the various air-driven treatmenttechnologies. The base frame provides various care modules which aremountable on the bed and usable with the control network, display, andair compressors built therein. Illustratively, a common air power sourceand handling unit is located on the main frame of the bed to supply airto all of several selected mattress surface therapies. Therapy frameheader connectors including a plurality of air lines for coupling theair source to selected air surface modules is provided. A therapysurface control module is mounted on the bed. A microprocessor ormicrocontroller-based electronic module is configured to be electricallycoupled to electronics residing in each of the separate treatmentsurface modules. The control module on the bed will control power, airdistribution, and graphical display, and the control module containsvalves and electronic controls to be described hereinafter. The controlmodule is capable of recognizing the specific surface which is connectedto the control module and will then control the air handling unit anddisplay according to the selected surface.

The bed will primarily be powered from the main AC power input for thehospital or clinic in which it is installed. When disconnected from themain AC power input, a battery may be provided on the base frame forlimited functionality.

The bed including the features described above meets the needs ofmultiple acute care areas including critical care, step down,medical/surgical, and subacute care. This flexibility results in reducedhandling and transport of the person since mobility can be providedcloser to the stretcher, and the person can stay on one bed fortransport so that seamless care can be provided. In addition,convertibility of the bed to the sitting position provides benefitsincluding that the upright position provides physiological benefitsmeeting M.D. orders, it minimizes handling of the person and minimizesthe number of caregivers required for handling people, it speeds patientrecovery, it minimizes the length of stay satisfying the criticalpathway, and it enhances patient safety.

Bed with Deflatable Foot Section

It is an object of the present invention to provide a bed for supportinga person, the bed having a head end, a foot end, and sides, and the bedcomprising a frame, and an articulating deck supported on the frame. Thearticulating deck comprises longitudinally spaced apart head, seat, andfoot sections. The head, seat, and foot sections are all movablerelative to each other to accommodate changes of the position of theperson on the bed.

A mattress is supported on the deck and the mattress has head, seat, andfoot mattress portions associated with or corresponding, respectively,to the head, seat, and foot sections of the deck. The words“corresponding” and “associated” are used in a general sense herein toassociate portions of the mattress with sections of the articulatingdeck and/or with areas of the body of the person resting on the sleepingsurface of the bed. In some cases, a mattress will be selected for aparticular person and the mattress will have longitudinally spacedportions corresponding respectively to portions of the person.

The foot section of the deck is movable from a generally horizontal upposition to a generally vertically downwardly extending down position topermit the lower legs and feet of the person to be lowered, for example,when the bed moves to a sitting position. This foot section may beselectively stopped in its various positions between the generallyhorizontal up position and the generally vertically downwardly extendingdown position to permit the lower legs of the person to be inclined in aconventional recliner fashion. The mattress foot portion is inflatableto serve as a sleeping surface when inflated and when the foot sectionof the deck is generally horizontal, and is declined downwardly anddeflated when the foot section of the deck is in the down position toprovide clearance for the lower legs and feet of the person.

The head section of the deck preferably translates toward the head endof the bed and pivots upwardly to provide a pivotable backrest orback-support portion for the person when the bed moves to the sittingposition. In the illustrative and preferred embodiment, the seat sectionincludes a thigh section that pivots upwardly relative to the frame.When the bed serves as a chair, the thigh section pivots upwardly toform an angle with the frame and cooperates with the head section tocradle the person on the bed thus providing a secure seat for theperson.

The seat section of the articulating deck can be lowered to a lowposition at which the sleeping surface adjacent to the seat portion ofthe mattress is approximately 15 inches (38 cm) above the floor. Theseat section is typically in the low position when the bed is in thesitting position. Thus, although the preferred and illustrative footsection of the articulating deck may be longer than 15 inches (38 cm)when the foot section is in the up position, the foot section preferablyand illustratively can be manipulated to have a longitudinal dimensionof 15 inches (38 cm) or less that will “clear” the floor when the bed isconverted to the sitting position and the foot section moves to the downposition.

Articulating Deck Having a Contracting Foot Section

It is still another object of the present invention to provide a bedhaving an articulating deck having a unique contracting foot section.The bed has an articulating deck on a frame and a mattress on the deck.The deck has longitudinally spaced head, seat, and foot sections thatare movable relative to each other to accommodate changes of theposition of the person on the mattress. The mattress has head, seat, andfoot mattress portions corresponding, respectively, to the head, seat,and foot deck sections.

The foot section of the deck is movable from a generally horizontal upposition to a generally vertically downwardly extending down position topermit the lower legs and feet of the person to be lowered. The footsection of the deck and the mattress foot portion have a first lengthwhen the foot section is in the up position and a second length shorterthan the first length when the foot section is in the down position.This foot section may be selectively stopped in its various positionsbetween the generally horizontal up position and the generallyvertically downwardly extending down position to permit the lower legsof the person to be inclined in a conventional recliner fashion.

The seat section of the articulating deck can be lowered to a lowposition in which the sleeping surface adjacent to the seat portion ofthe mattress is approximately 15 inches (38 cm) above the floor. Theseat section is typically in the low position when the bed is in thesitting position. Thus, although the preferred and illustrative footsection of the articulating deck may be longer than 15 inches (38 cm)when the foot section is in the up position, the foot section cancontract so that the foot section has a longitudinal dimension that will“clear” the floor when the bed is converted to the sitting position andthe foot section moves to the down position. Thus, the foot section canexpand and contract so that the length of the foot section variesbetween a first length and a second length, the second length beinggreater than the first length.

Step Deck in Combination with a Reduced-Shear Pivot Assembly

It is also an object of the present invention to provide such a bedhaving a head, a foot, and sides, and including a unique combination ofa step deck with a reduced-shear pivot feature. The bed includes a frameand a deck carried by the frame. The deck includes an upper deck portionand a central, longitudinal recess in the upper deck portion, the recessbeing defined by a lower deck portion and walls connecting the upper andlower deck portions. In addition, the bed includes a mattress having aplanar, upwardly-facing support surface, side portions resting on theside deck portions, and a central projection extending downwardly intothe recess. The bed also includes a first longitudinal deck sectioncoupled to the deck to pivot about a pivot axis above the lower deckportion between a generally horizontal position and a tilted position.

The head section is coupled to the walls adjacent to the seat sectionand above the lower deck so that the head section is movable from agenerally horizontal down position to a back-support position providinga pivotable backrest. Preferably, the head section simultaneouslytranslates toward the head end of the bed and pivots upwardly whenmoving from the down position to the back-support position. Thetranslation and the pivoting motions combine to produce a motion whereinthe head section pivots relative to the seat section about an effectivepivot axis positioned to lie above the lower deck.

The vertical distance between the support surface and the reduced-shearpivot assembly can be minimized when the bed includes a step deck havingupper deck side portions and a corresponding thin mattress portion.Mounting the reduced-shear pivot assembly to the walls connecting thelower deck and the upper deck portion minimizes the extent that thereduced-shear pivot assembly is required to raise the effective pivotaxis above the reduced-shear pivot assembly as compared to areduced-shear pivot assembly mounted to the bottom of a deck.

Bed Base Frame, Intermediate Frame, and Power Package

It is still another object of the present invention to provide a bed forsupporting a patient, the bed being convertible between a bed positionand a sitting position and having a head end, a foot end, and oppositesides as well as a unique base frame and power unit arrangement. The bedcomprises a base frame, casters for supporting the base frame formovement of the bed by a caregiver, and an intermediate frame mounted onthe base frame for movement upwardly and downwardly to selected heightsand orientations relative to the base frame. An articulated deck ismounted on the intermediate frame, the deck having head, foot, and seatsections that are movable relative to each other.

The bed also includes a fluid system including a pump, valves, andactuators for positioning the intermediate frame relative to the baseframe and for articulating the deck sections. In addition, an electricalcontrol system for operating the pump and controlling the valves of thefluid system when connected to an independent source of power and foroperating only the valves using a battery source on the bed when notconnected to an independent source of power is provided on the bed.

The deck has head, foot, and seat sections, the head section beingmovable between a down position generally parallel to the intermediateframe and an upward back-support position propping up the person andserving as a chair back. The seat section includes a thigh section thatis movable between a generally horizontal down position and an upposition to prop up the thighs of the patient. The foot section ismovable between a generally horizontal up position and a generallyvertically downwardly extending down position to lower the lower legsand feet of the patient.

First means are provided for raising and lowering the intermediate framerelative to the base frame to raise and lower the articulating deck.Second means are provided for raising and lowering the head section ofthe deck relative to the intermediate frame, third means are providedfor raising and lowering the thigh section relative to the intermediateframe, and fourth means are provided for raising and lowering the footsection relative to the intermediate frame. In the illustrative andpresently preferred embodiment, each of the first, second, third andfourth means comprises a hydraulic actuator such as a hydraulic pistonand cylinder arrangement.

The intermediate frame and a weigh frame carried by the intermediateframe support the articulating deck sections for movement to theirvarious positions. The seat section of the deck is illustratively andpreferably fixed to the weigh frame. Power actuators act between theweigh frame and the articulating deck head, thigh, and foot sections.Additional power actuators act between the intermediate frame and thebase frame to raise and lower the intermediate frame relative to thebase frame.

Illustratively, the weigh frame is supported on the intermediate frameby a plurality of load beams which serve to weigh the person as will bedescribed hereinafter. If a weighing capability is not provided, theweigh frame may be fixedly secured to the intermediate frame by “dummy”beams or members which are not load-cell members to provide a“non-scale” bed. In such a non-scale bed, the weigh frame and theintermediate frame are linked together such that they both may beconsidered as a common frame.

The bed further comprises a hydraulic power unit carried on the baseframe to provide pressurized fluid for activating the actuators, conduitfor connecting the power unit to the actuators, and a plurality ofvalves for controlling the flow of fluid between the hydraulic powerunit and each actuator. While hydraulic actuators are shown in theillustrative embodiment, it will be appreciated that, in accordance withthe present invention, various mechanical and electromechanicalactuators and drivers may be used to raise and lower the intermediateframe on the base frame as well as to raise and lower individual decksections relative to the intermediate frame.

It is well known in the hospital bed art that electric drive motors withvarious types of transmission elements including lead screw drives andvarious types of mechanical linkages may be used to cause relativemovement of portions of hospital beds. It is also well known to usepneumatic actuators to actuate and/or move individual portions ofhospital beds. As a result, the terms “means for raising and lowering”in the specification and in the claims, therefore, are intended to coverall types of mechanical, electromechanical, hydraulic and pneumaticmechanisms, including manual cranking mechanisms of all types, forraising and lowering portions of the hospital bed of the presentinvention.

It is an object of the present invention also to provide such ahydraulic power unit comprising electrically driven pump means forsupplying hydraulic power when sufficient electrical power is availableand connected to the bed, as well as manually driven pump means forsupplying hydraulic power when sufficient electrical power is notavailable. The bed of the present invention may preferably comprise abattery power pack for supplying electrical power sufficient to operatethe above-said valves, whereby, with the manually-driven pump means andthe battery powered valves, the deck can be raised and lowered and thehead, thigh, and foot sections can be raised and lowered when the bed isdisconnected from the primary power source, for example, the main ACelectrical source provided by a hospital.

The bed of the present invention may be lowered to a position such thatthe mattress supporting deck is 15 inches (38 cm) from the floor andraised to a work position such that the deck is 34 inches (86.4 cm) fromthe floor. The head section of the mattress-supporting deck is connectedto the weigh frame of the bed by a reduced-shear pivot arrangement suchthat the head section, with the mattress thereon, travels toward thehead of the bed and simultaneously pivots upwardly simulating the pivotof the hip of the human body. The head of the bed may have, for example,85° of articulation.

The thigh section of the mattress-supporting deck is connected to theweigh frame of the bed for pivoting movement so that the end of thethigh section adjacent to the foot section may have, for example, 10° ofarticulation upwardly away from the weigh frame. The end of the footsection of the mattress-supporting deck nearest to the thigh section isconnected to the weigh frame of the bed and may have, for example, 90°of articulation downwardly away from the weigh frame. In addition, thefoot section may contract and expand so that the length of the footsection can vary, for example, between a first length when the footsection is in the up position and a second length when the foot sectionis in the down position, the first length being longer than the secondlength.

Emergency Trendelenburg Positioning

It is further an object of the present invention to provide a bed havingan emergency Trendelenburg positioning feature. The bed is convertiblebetween a bed position and a sitting position, and has a head end, afoot end, and opposing sides. The bed comprises a base frame and anintermediate frame mounted on the base frame for upward and downwardmovement to selected heights and orientations relative to the baseframe. An articulated deck is mounted on the intermediate frame, thearticulated deck having head, foot, and seat sections movable relativeto each other.

The bed further comprises a fluid system including a pump, valves, andactuators for positioning the intermediate frame relative to the baseframe and articulating deck sections and an electrical control systemfor operating the pump and controlling the valves of the fluid system.The fluid system includes a manual valve for operating an actuator tolower an end of the intermediate frame relative to the other end of theintermediate frame independent of the electrical control system.

The bed can alternatively comprise a mattress supported on the frame andhaving a support surface and a positioning system for positioning theintermediate frame relative to the base frame between a generallyhorizontal position and a Trendelenburg position having one end of thesupport surface inclined with respect to the other end of the supportsurface and including a first lock for locking and unlocking theintermediate frame in the horizontal position. A manual actuator can becoupled to a second lock in the positioning system for unlocking theintermediate frame independent of the first lock so that the positioningsystem can move the intermediate frame to the Trendelenburg position.

CPR Foot Pedal

It is further an object of the present invention to provide a bed havinga unique CPR foot pedal feature. The bed has a head, a foot, and twosides, and includes a frame having a top and a bottom. The bed alsoincludes an articulating deck coupled to the frame, the articulatingdeck having a head section that is movable relative to the frame. Thehead section can move between an upward back-support position providinga backrest and a generally horizontal bed position. The bed alsoincludes locking means for securing the head section in the back-supportposition.

A CPR foot pedal is coupled to the locking means and is positioned tolie beneath the articulating deck so that the foot pedal is accessibleto the foot of a caregiver. The foot pedal is movable between an upposition and a downward releasing position releasing the locking meansso that the head section can move downwardly to the bed position whenthe foot pedal is in the releasing position.

The head section of the articulating deck can quickly drop from theback-support position to the down position, for example, to allow acaregiver to quickly administer cardiopulmonary resuscitation to aperson on the sleeping surface who experiences cardiac arrest when thebed is in the sitting position. The CPR foot pedal can be activated by acaregiver by pressing the pedal to cause the head section to rapidlypivot downwardly from the back-support position. Preferably, constantactivation is required and the head section will continue to drop onlyso long as the pedal is activated, leaving the hands of the caregiverfree to conduct other activities as the head section moves toward thedown position.

Docking Site for Docking to the Bed when the Bed is in the SittingPosition

It is further an object of the present invention to provide apatient-care bed with a unique built-in docking capability. The bed hasa head end, a foot end, and two opposing sides, and is convertiblebetween a sitting position and a bed position. The bed includes a baseand a frame coupled to the base. The frame is movable relative to thebase between a low position having the frame a first distance from thefloor and a high position having the frame a second distance from thefloor, the first distance being less than the second distance.

An articulating deck is coupled to the frame. The articulating deckincludes longitudinally spaced head and foot sections that are movablerelative to the frame and movable relative to each other to accommodatechanges of position of a person on the bed. The bed is movable to asitting position having the frame in the low position, the head sectionin an upward back-support position providing a backrest, and the footsection in a generally vertically downwardly extending down position.

A latch-receiver post is appended to the frame and is configured toreceive a latch connected to a portable equipment module so that theportable equipment module can dock with the bed when the bed is in thesitting position. The portable equipment module can include anyequipment that is portable and that may be docked with the bed tomaintain the relative position of the portable equipment module and thebed while taking advantage of the mobility of the bed and the stabilityof the bed when the bed is in the sitting position. For example, amobile power module and a mobile toilet facility would each be portableequipment modules that could be docked to the bed when the bed is in thesitting position.

Bed with a Weigh Frame Supporting an Articulating Deck

It is further an object of the present invention to provide apatient-care bed with a built-in weighing feature. The bed has a head, afoot, and two sides. The patient-care bed includes a first frame, aweigh frame, and a plurality of load cell supports coupling the weighframe to the first frame. In the context of the preferred embodiment,the “first frame” is the “intermediate frame.” The load cell supportsinclude means for determining the weight of objects supported by theweigh frame.

An articulating deck is coupled to the weigh frame. The articulatingdeck includes longitudinally spaced head, seat, thigh, and footsections, the head, thigh, and foot sections being movable relative tothe weigh frame and movable relative to each other. The movements of thesections of the deck accommodate changes of the position of the personon the bed.

Bed with Extended Frame

It is further an object of the present invention to provide a bed havinga head, foot and opposite sides with a unique extended frame feature.The bed comprises a frame, a deck coupled to the frame, a mattressresting on the deck to cushion the patient, and an extended framecoupled to the frame at the foot of the bed. The extended framecomprises two gate assemblies, each gate assembly including a gate, aframe-extender member connected to the frame, and a swing memberconnecting the gate to the frame-extender member.

The gates have closed positions transverse to the bed sides. When in theclosed position, the gates cooperate to close the foot of the bed. Theframe-extender members are connected to the frame adjacent to the sidesof the bed and extend in a direction outwardly and away from the headend of the bed.

A first swing member has a first end pivotably connected to the firstframe extender member. The second end of the swing member swings betweena storage position adjacent to the frame and a closed position adjacentto the deck near the foot of the bed. The first gate is rotatablycoupled to the second end of the swing member. Likewise, a second swingmember has a first end pivotably connected to the second frame-extendermember and the second gate is rotatably coupled to the second end of thesecond frame-extender member. The gates have normal positions transverseto the bed sides acting together to close the foot of the bed, and thegates cooperate with the swing members to move to positions extendinggenerally adjacent to the frame along the sides of the bed, therebyopening the foot of the bed.

The gates are configured to move with the sleeping surface as thesleeping surface is raised and lowered so that if the bed is in thesitting position, the person can grasp the gates for support. Thesleeping surface can then be raised to assist the person as they standand the gates will raise with the sleeping surface, providing support tothe person as they stand.

Typically, the extended frame is carried by the weigh frame. Fornon-scale embodiments of the bed having the common frame configuration,the extended frame is carried by the common frame. Mounting the extendedframe to the weigh frame or to the common frame causes the extendedframe and foot gates to move with the weigh frame or the common frameand to remain stationary relative to the person supported on thesleeping surface. The gates of the extended frame can swing outwardlyfrom the closed position to an open position having each gate positionedto lie adjacent to the swing member. When gates are in the openposition, the caregiver has clear access to the foot section of the bed.When the bed is in the sitting position and the gates are in the openposition, the person carried by the sleeping surface has clear path foregress from the foot end of the bed.

Additionally, the extended frame can fold like an accordion against thebed with the swing members swinging outwardly and around to the storageposition and the gates swinging inwardly against the swing members to aside-grip position next to the swing members. In the side-grip position,the gates serve as a protective “crib-like” perimeter and provide handsupports for the person egressing from the foot of the bed when the bedis in the sitting position. Additionally, the frame and gates can easilybe removed entirely from the foot end of the bed by folding the frameback and folding the gates back.

Swing members in the extended frame minimize the radius of the arc ofthe gate as it swings between the side-grip position and the closedposition. Also, use of the swing members allows the length of theframe-extender members to be minimized while providing the caregiverwith satisfactory access to the person when the bed is in either the bedposition or the sitting position.

The present invention includes several combinations of individualfeatures disclosed herein. For example, the combination of the step deckwith several features such as the reduced-shear pivot, the siderails,the pivoting foot section, the mattress having the projection, and themattress having the deflatable foot portion are all combinations thatare included in the bed in accordance with present invention.

The electronic system architecture for the hospital bed of the presentinvention includes a plurality of electronically controlled moduleslocated on the bed which are interconnected in a peer-to-peerconfiguration. This peer-to-peer communication network configurationenables any of the plurality of modules to communicate directly withanother module in the network without the need for a master controller.In the preferred embodiment, information flow between the electronicmodules is primarily accomplished through the use of a twisted pairnetwork channel, although other physical protocols would be acceptable.

One feature of the control system of the present invention is improvedupgradeability. The peer-to-peer network configuration of the electroniccontrol modules of the present invention facilitates adding or removingmodules from the bed. In conventional bed control systems which use amaster controller, the master controller must be initially designed orsubsequently redesigned to accommodate additional modules. Since nomaster controller is required in the peer-to-peer network configuration,the electronic control system of the present invention does not have tobe redesigned or reprogrammed each time a module is added or removedfrom the bed.

An open product architecture for the communication control network andair controls provides substantial flexibility for future additions ofnew modules. A graphic caregiver interface control module is providedfor controlling the operation of various modules of the hospital bed.This control module is coupled to the peer-to-peer communicationnetwork. The control module includes a user input control panel and adisplay. The control module is programmed to recognize when a new moduleis added to the network automatically and to permit control of the newmodule from the user input control panel. The control module alsodisplays specific control options for the added new module on thedisplay automatically. Therefore, this new module recognition andcontrol apparatus eliminates the need for separate controls on eachindividual module.

The network of the present invention also includes a bed statusinformation charting feature. The network allows all data from each ofthe modules coupled to the network to be available at any time to theother modules. An optional module allows the network to supplyinformation to a remote location through a data link. This informationincludes information from any of the modules communicating on thenetwork. The peer-to-peer communication network of the present inventiontransmits electrical signals representing bed status variables thatindicate the current position, status, and configuration of the bed.These variables include bed articulation angles, brakes, bed exit,scale, surface therapy attributes, as well as other variables. Bydetecting and storing changes in these bed status variables in thememory of a module or by transmitting these variables via the data linkto a remote location, the present invention permits automatic chartingof the bed status variables. Therefore, the hospital information systemcan monitor and record changes in the bed status variables continuouslyduring the patient's stay for billing, legal, insurance, clinical/careplan studies, etc. The caregiver can also routinely check a nurse callbed status at a remote nurse master station rather than making bed checkrounds. A history of the bed status for a particular patient can bedisplayed on the graphical user interface module, downloaded to a datafile, and/or routed via the data link to a remote location.

The peer-to-peer communication network of the present invention is adistributed network. This distributed design allows for peer-to-peercommunications between any of the nodes or modules connected to thenetwork. Failure of a single module does not cause failure or impairmentof the entire peer-to-peer communication network.

The peer-to-peer communication network of the present invention includesembedded self diagnostic capability. The network is capable ofinternally diagnosing hardware and software failures and recommending acorrective action. A signal for this corrective action can be suppliedto a troubleshooting screen on the graphical user interface module,downloaded to a data file, and/or transmitted via a data link to aremote location.

Alternately, a service indicator can be lit to indicate the need forservicing of a specific system failure. Remote troubleshooting ordiagnostics is also possible through a modem connected to an accessorymodule of the bed. A remote computer can run tests and interrogate othermodules of the bed to indicate problems and suggest solutions.

This diagnostic capability also enhances serviceability of the bed. Thelighted LEDs indicate a specific system failure. The graphic caregiverinterface provides detailed information related to product failures onthe bed. In addition, after diagnosis of the bed is performed from aremote location, a company service technician at the remote location cancall an engineer at the hospital to help service the bed.

According to yet another aspect of the present invention, the bedincludes a plurality of different air therapy and support surfaces, allof which can be connected to the bed to provide a complete therapy linethat is rapidly installed or exchanged on demand as census or diagnosticpopulation varies. In an acute care environment, a hospital typicallyneeds decubitus prevention, decubitus treatment (stage one and twominimum), pulmonary therapies including rotation therapy and percussionand vibration therapy, and venous compression therapy capabilities.

The modular therapy and support surface design of the present inventionallows several air support surfaces and air therapy devices to be drivenby a common air source, a common graphical interactive display device,and a distributed communication network. The modular therapy and surfacesupport system of the present invention is designed to provide a one bedsolution for acute care including critical care, step down/progressivecare, med-surg, high acuity subacute care, PACU, and sections of ED. Themodular therapy and support surface system of the present inventionprovides therapies that benefit a large percentage of the patientpopulation in an acute care hospital.

The bed of the present invention includes an air handling unit locatedon a bed frame which is capable of supplying air pressure and/or avacuum to all the therapy and support surface modules. Typically, theair handling unit is mounted on the base frame of the bed. Preferably,the air handling unit drives two lines simultaneously for supplying bothair pressure and vacuum to the air therapy modules. A header connectoris coupled to the air handling unit by a plurality of air lines. Theheader connector is configured to couple the air handling unit to aselected modular air therapy device support surface.

The modular therapy and support surface components for the differenttherapies are contained within the sleep surface on the bed, enabling acaregiver to install, initiate, or remove a desired air therapy from thebed without moving the patient off the original support surface. Themodular design of the present invention allows modules for air therapyto have reduced size. Therefore, the modules can be delivered after thebed and stored easily. The air handling unit of the present invention iscoupled to therapy control modules that contain air distribution meanssuch as adjustable valves and sensors by a simple connection ofpneumatic lines to the control modules.

According to one aspect of the present invention, a bed includes a baseframe, a deck coupled to the base frame, an electrical communicationnetwork, and an air handling unit mounted on the base frame. The bedalso includes a plurality of air therapy devices located on the bed, anda plurality of control modules. Each control module includes a connectorfor coupling a corresponding air therapy device to the air handling unitand to the electrical communication network. Each control module alsoincludes a controller for operating the corresponding air therapy devicewith the air handling unit based on command signals received from theelectrical communication network.

The bed further includes a control unit coupled to the electricalcommunication network for transmitting command signals for the pluralityof air therapy devices over the electrical communication network tocontrol operation of the plurality of air therapy devices. The controlunit includes a display and a user input. Each control module transmitsdisplay commands to the display related to the corresponding air therapydevice. The display commands from the control modules provide a menudriven list of options to the display to permit selection of controloptions for the plurality of air therapy devices from the user input.

In the illustrated embodiment, one of the plurality of air therapydevices is a support surface air bladder located on the deck. Thesupport surface air bladder includes a plurality of independentlycontrolled air zones. One of the plurality of control modules is adecubitus prevention control module coupled to the support surface airbladder to control each of the plurality of air zones of the supportsurface with a common connection to the air handling unit. Another ofthe plurality of control modules is a decubitus treatment control modulefor independently coupling the plurality of air zones of the supportsurface air bladder to the air handling unit.

Another of the plurality of air therapy devices is a pulmonary rotationbladder located between the deck and the support surface air bladder. Apulmonary rotation control module is provided for coupling the pulmonaryrotation air bladder to the air handling unit. The pulmonary rotationcontrol module is coupled to the electrical communication network.

Yet another of the plurality of air therapy devices is a sequentialcompression therapy device. A sequential compression device air controlmodule is provided for coupling the sequential compression device to theair handling unit. The sequential compression device air control moduleis coupled to the electrical communication network.

Still another of the plurality of air therapy devices is a pulmonarypercussion and vibration bladder located on the deck for providingpulmonary percussion and vibration therapy. A pulmonary percussion andvibration control module is provided for coupling the percussion andvibration bladder to the air handling unit. The percussion and vibrationmodule is coupled to the electrical communication network.Alternatively, the percussion and vibration control module is configuredto couple a selected air zone of the support surface air bladder to theair handling unit to provide percussion and vibration therapy in theselected air zone.

An auxiliary air port control module is coupled to the air handling unitand to the electrical communication network. The air port control moduleprovides an auxiliary air outlet on the bed.

According to another aspect of the present invention, a control moduleis provided for activating an air therapy device on a bed which includesa base frame, a deck coupled to the base frame, an electricalcommunication network, an air handling unit mounted on the base frame, agraphical interactive display coupled to the electrical communicationnetwork for transmitting and receiving command signals from thecommunication network, and a plurality of air therapy devices stored onthe bed. The control module includes at least one electricallycontrolled valve having an input and an output, at least one pressuresensor having an input and an output, and an electronic controllercoupled to and controlling the at least one electrically controlledvalve and coupled to the output of the at least one pressure sensor. Thecontrol module also includes a connector for coupling the input of thevalve to the air handling unit on the bed, for coupling the output ofthe valve to the selected air therapy device, for coupling the input ofthe pressure sensor to the selected air therapy device, and for couplingthe controller to the electrical communication network on the bed sothat the controller receives the command signals from the graphicalinteractive display to control the selected air therapy device.

The graphical interactive display includes a display and a user input.The controller transmits display command signals to the graphicalinteractive display to display information related to the selected airtherapy device on the display. The display commands from the controllerprovide a menu driven list of control options for the selected airtherapy device to the display to prompt selection of various controloptions for the selected air therapy device from the user input.

If the selected air therapy device includes a plurality of air zones,the control module includes an electrically controlled valve for each ofthe plurality of air zones to couple the plurality of air zones to theair handling unit on the bed independently. The control module alsoincludes a separate pressure sensor for each of the plurality of airzones.

According to yet another aspect of the present invention, a bed includesa base frame, a deck coupled to the base frame, an electricalcommunication network, an air handling unit mounted on the base frame,and a header connector including an electrical connector coupled to theelectrical communication network and a pneumatic connector coupled tothe air handling unit. The bed also includes a plurality of exchangeableair therapy devices. Each of the air therapy devices includes at leastone air zone, a therapy control module having a controller, a valvecoupled to each air zone of the air therapy device, and a moduleconnector configured to mate with the header connector to couple thevalve to the air handling unit and to couple the controller to theelectrical communication network so that each of the plurality ofexchangeable air therapy devices use the same air handling unit andelectrical communication network.

In the illustrated embodiment, the module connector includes a firstconnector coupled to an input of the valve and a second connectorcoupled to the controller, the first connector of the module connectorbeing configured to mate with the pneumatic connector of the headerconnector on the bed to couple the air handling unit to the at least oneair zone of the air therapy device through the corresponding valve andthe second connector being configured to mate with the electricalconnector of the header connector on the bed to couple the electricalcommunication network to the controller so that the controller receivescommands from the electrical communication network to control air flowto the air therapy device through the valve.

According to still another aspect of the present invention, the modularsupport surface of the present invention includes an improved surfacefoot section specifically designed for use with a bed having anarticulating deck movable from a normal bed position to a chairposition. The surface foot section is configured to retract or shortenas the bed moves to the chair position to enable a patient's feet to beplaced on the floor or on a foot prop. The foot section also collapsesor thins to maintain an acceptable chair seat size which also enablesthe patient's feet to be placed on the floor or foot prop.

In the illustrated embodiment, a surface foot section apparatus isprovided for a bed including a base frame, an articulating deck coupledto the base frame, the articulating deck including a generally planarfoot deck section, the articulating deck being movable from a bedconfiguration to a chair configuration. The surface foot sectionapparatus includes a first set of air bladders configured to collapse ina first direction generally parallel to the foot deck section when thefirst set of air bladders is deflated, and a second set of air bladderslocated adjacent the first set of air bladders. The second set of airbladders is configured to collapse in a second direction normal to thefoot deck section when the second set of air bladders is deflated sothat the surface foot section has a substantially reduced thickness anda substantially reduced length when the first and second bladders aredeflated. The surface foot section apparatus also includes a footsection control module for selectively inflating and deflating the firstand second sets of air bladders. The foot section control moduledeflates the first and second sets of air bladders when the articulatingdeck is in the chair configuration, and the foot section control moduleinflates the first and second sets of air bladders when the articulatingdeck is in the bed configuration.

Preferably, the length of the surface foot section is reduced by atleast 40% when the first and second air bladders are deflated and thethickness of the surface foot section is reduced by at least 80% whenthe first and second air bladders are deflated. This feature maintainsan appropriate size for a seat section of the chair and permits apatient's feet to touch the floor when the bed is in the chairconfiguration. The foot deck section is movable from an extendedposition to a retracted position to shorten the foot deck section as thearticulating deck moves to the chair configuration.

Also in the illustrated embodiment, each of the second air bladders isindependently controlled as a separate air zone by the foot sectioncontrol module. The foot section control module selectively inflates anddeflates the second air bladders to provide a heel pressure relief inthe surface foot section. The first set of air bladders is commonlycontrolled as a single air zone by the foot section control module.

According to a further aspect of the present invention, a pulmonaryrotation therapy apparatus is provided for use on a bed having a baseframe, a deck coupled to the base frame, and a support surface locatedon the deck. The pulmonary rotation therapy apparatus includes anormally deflated rotation air bladder located between the supportsurface and the deck. The rotation air bladder remains deflated duringnormal use of the bed. It is understood that the rotation air bladdercan be normally inflated and used as a support surface for the bed, ifdesired. The pulmonary rotation therapy apparatus also includes apulmonary rotation control module coupled to the rotation air bladder.The pulmonary rotation control module selectively inflates and deflatesportions of the rotation air bladder to provide rotational therapy to abody located on the support surface.

In the illustrated embodiment, the rotation air bladder includes aplurality of elongated air bladders extending generally parallel to alongitudinal axis of the bed. The pulmonary rotation control moduleselectively inflates or deflates the plurality of air bladders tocontrol rotation of the patient on the support surface. The rotation airbladders are divided into at least three separate air zones which areindependently controlled by the pulmonary control module.

In an illustrated embodiment of the invention, the support surfaceincludes a plurality of air bladders located on the deck. It isunderstood that any type fluid may be used. The air bladders arepreferably divided into separately controlled air zones corresponding tothe various deck sections of the articulating deck. Therefore, thesupport surface includes separately inflatable head, seat, thigh, andfoot air zones.

Inflation and deflation of the various surface sections is controlled bya surface instrument control module and an air supply module, both ofwhich are coupled to the electrical communication network on the bed.The surface instrument module and the air supply module both receivesignals from the bed articulation control module and from a positionsensing module as the bed begins moving from the bed position to thechair position. The surface instrument module and air supply moduleautomatically partially deflate a seat air zone section of the supportsurface and the foot air zone section of the support surface as the bedmoves to the chair position. For this purpose, the seat section includesnot only the air zone overlying the seat portion of the deck, but alsothe air zone overlying the thigh portion of the deck. In the chairposition, a person's weight is mostly supported by the thigh sections ofthe support surface and deck. Such partial deflation of the seat sectionof the bed is automatically controlled to distribute the person's weightas the bed moves to the chair position. In addition, the bedarticulation control module automatically elevates an end or the thighdeck section closest to a foot end of the bed to maintain the patient ina seated position on the chair bed.

Additional objects, features, and advantages of the invention willbecome apparent to those skilled in the art upon consideration of thefollowing detailed description of the preferred embodiments exemplifyingthe best mode of carrying out the invention as presently perceived.

BRIEF DESCRIPTION OF THE DRAWINGS

The detailed description particularly refers to the accompanying figuresin which:

FIG. 1 is a perspective view of a chair bed in accordance with thepresent invention showing a siderail exploded away from the chair bed,head siderails and foot siderails positioned along longitudinal sides ofthe deck, and a swinging foot gate in a closed position;

FIG. 2 is a view similar to FIG. 1 showing the chair bed in the sittingposition having a head section of an articulating deck moved upwardly toa back-support position, a thigh section of the deck inclined slightlyupwardly, a foot section of the deck moved to a generally verticaldownwardly extending down position, a foot portion of the mattress beingdeflated, and swinging gates moved to an open position with one swinginggate folded next to the chair bed;

FIG. 3 is a diagrammatic view of the chair bed of FIG. 1 showing thechair bed in a bed position including a mattress having anupwardly-facing sleeping surface held a predetermined first distanceabove the floor, the deck being in an initial position supporting thesleeping surface in a generally planar configuration, and the footsection being a first length;

FIG. 4 is a diagrammatic view showing the chair bed in a low position;

FIG. 5 is a diagrammatic view showing the chair bed in a Trendelenburgposition;

FIG. 6 is a diagrammatic view showing the chair bed in a reverseTrendelenburg position;

FIG. 7 is a diagrammatic view showing the chair bed in an intermediateposition having a head end of a head section of the deck pivotedslightly upward from the initial position of the deck, a seat sectionpositioned to lie in the horizontal plane defined by the seat section inthe initial position of the deck, and the foot section being inclinedslightly so that the foot end of the foot section lies below theposition of the foot section when the deck is in the initial position ofthe deck;

FIG. 8 is a diagrammatic view showing the chair bed in a sitting orchair position with the head end of the head section pivoted upwardlyaway from the seat section to a back-support position, the seat sectionlying generally horizontal as in the initial deck position, the thighsection being raised upwardly, the foot section extending downwardlyfrom the thigh section and being a second shorter length, and theportion of the mattress over the foot section being deflated;

FIG. 9 is a perspective view of a first embodiment of a step deck and amattress in accordance with the present invention;

FIG. 10 is a sectional view taken along line 10-10 of FIG. 9 showing thebottom of the step deck beneath the projection;

FIG. 11 is an exploded perspective view of the chair bed of FIG. 1 withportions broken away;

FIG. 12 is a perspective view of the base frame of the chair bed of FIG.1 showing portions of the hydraulic system module mounted on the baseframe;

FIG. 12 a is a perspective view of the power unit for supplying power tomove the portions of the chair bed;

FIG. 13 is a fluid circuit diagram of a hydraulic system module of thechair bed of FIG. 1;

FIG. 14 is an exploded perspective view of the intermediate frame andthe weigh frame of the chair bed of FIG. 1;

FIG. 14 a is a sectional view taken along line 14 a-14 a of FIG. 14showing a load beam cantilevered to the intermediate frame;

FIG. 15 is a sectional view taken along line 15-15 of FIG. 1 having thechair bed in the intermediate position similar to the position shown inFIG. 7;

FIG. 16 is a view similar to FIG. 15 showing portions of the headsection of the articulating deck and the reduced-shear pivot assembly inthe down position shown in FIG. 3;

FIG. 17 is a view similar to FIG. 16 showing portions of the headsection and the reduced-shear pivot assembly in the back-supportposition shown in FIG. 8;

FIG. 18 is a perspective view of a second embodiment of a chair bed in agenerally horizontal bed position;

FIG. 19 is a perspective view of chair bed of FIG. 18 showing the chairbed in a sitting position;

FIG. 20 is a sectional view taken along line 20-20 of FIG. 18 showingthe chair bed of FIG. 18 in the bed position;

FIG. 21 is a view similar to FIG. 20 showing the chair bed in anintermediate position;

FIG. 22 is a view similar to FIG. 21 showing the chair bed in thesitting position;

FIG. 23 is an enlarged view similar to FIG. 20 of the second embodimentof the chair bed showing a telescoping member received by a sheath andriding on a roller while in the fully retracted position;

FIG. 24 is a sectional view taken along line 24-24 of FIG. 1 showing thedeck foot section in an expanded position;

FIG. 25 is a view similar to FIG. 24 showing the deck foot section andthe pivoting member in the contracted position;

FIG. 25 a is a view similar to FIG. 24 of a second embodiment of a deckfoot section in an expanded position;

FIG. 26 is a view taken along line 26-26 of FIG. 25 showing a firsttongue and groove connection between the pivoting member and the slidingmember;

FIG. 27 is a view taken along line 27-27 of FIG. 25 showing a secondtongue and groove connection between the pivoting member and the slidingmember;

FIG. 28 is an exploded perspective view of a second embodiment of a stepdeck and the mattress of the chair bed;

FIG. 29 is a sectional view taken along line 29-29 of FIG. 28 of thestep deck and the mattress and showing a C-arm (in phantom) for holdingmedical equipment such as fluoroscopic equipment;

FIG. 30 is an exploded perspective view of a third embodiment of themattress and the deck showing the foot section of the deck and the footportion of the mattress in a minimized condition having the foot sectionof the deck contracted and the foot portion of the mattress contractedlongitudinally and deflated so that the foot portion of the mattress isthinner and shorter than when foot portion is inflated;

FIG. 31 is a diagrammatic side elevation view of the chair bed of FIG. 1showing the chair bed in the bed position of FIG. 3 and showing a headsection siderail and a body section siderail;

FIG. 32 is a diagrammatic view similar to FIG. 31 showing the headsection of the articulating deck of the chair bed raised to anintermediate position of FIG. 7;

FIG. 33 is a diagrammatic view similar to FIG. 31 showing the headsection in the back-support position of FIG. 8;

FIG. 34 is a sectional view taken along line 34-34 of FIG. 31 of asiderail in a patient-restraining position;

FIG. 35 is a view similar to FIG. 34 of the siderail intermediate thepatient-restraining position of FIG. 34 and a down-out-of-the-wayposition (in phantom) having a top of the siderail beneath the sleepingsurface;

FIG. 36 is an exploded view of a head section of an articulating deck ofthe chair bed of FIG. 1 including a breakaway siderail;

FIG. 37 is a front elevation view from outside of the bed of a headsection siderail in accordance with the present invention having amechanical angle indicator;

FIG. 38 is a sectional view taken along line 38-38 of FIG. 37 showingthe mechanical angle indicator;

FIG. 39 is a perspective view from outside of the bed of a body sectionsiderail in accordance with the present invention having a mechanicalangle indicator and a pivotable display;

FIG. 40 is a sectional view taken along line 40-40 of FIG. 39 showingthe pivotable display;

FIG. 41 is a sectional view taken along line 41-41 of FIG. 39 showingthe patient control buttons on the inside of the siderail;

FIG. 42 is a sectional view taken along line 42-42 of FIG. 41 showingthe patient control buttons;

FIG. 43 is a block diagram illustratively showing major functionalcomponents of the chair bed and some of the mechanical and fluidconnections therebetween;

FIG. 44 is a block diagram of the base module and portions of thehydraulic module illustratively showing some components of the basemodule and illustrating some of the mechanical, fluid, and electricalinterconnections therebetween;

FIG. 45 is a block diagram of the intermediate frame module and portionsof the hydraulic module illustratively showing some components of theintermediate frame module and illustrating some of the mechanical,fluid, and electrical interconnections therebetween;

FIG. 46 is a block diagram of the articulating deck/weigh frame moduleand portions of the hydraulic module illustratively showing somecomponents of the articulating deck/weigh frame module and illustratingsome of the mechanical, fluid, and electrical interconnectionstherebetween;

FIG. 47 is a block diagram of the siderail assemblies illustrativelyshowing some components of the siderail assemblies and illustrating someof the mechanical, fluid, and electrical interconnections therebetween;

FIG. 48 is a block diagram illustrating the electronic control modulesof the present invention connected in a peer-to-peer networkconfiguration and illustrating the additional system components whichare coupled to the various modules by discrete electrical connections;

FIG. 49 is a diagrammatical view illustrating the electrical connectionfrom the communication network cable to a selected module andillustrating a coupler between a pair of network connectors tofacilitate adding another module to the network;

FIG. 50 is a schematic block diagram illustrating the electroniccomponents of a bed articulation control module;

FIG. 51 is a schematic block diagram illustrating the electricalcomponents of the scale instrument module;

FIG. 52 is a schematic block diagram illustrating the mechanical andelectrical components of the bed position sense and junction module;

FIG. 53 is a schematic block diagram illustrating the components of theleft and right standard caregiver interface module for either the leftsiderail or the right siderail;

FIG. 54 is a diagrammatical view of the lockout switches on the siderailcontrol panel to prevent movement of selected sections of the bed; and

FIG. 55 is a schematic block diagram illustrating the mechanical andelectrical components of the graphical caregiver interface module;

FIGS. 56 and 57 are flow charts illustrating details of the automaticmodule recognition feature of the graphical caregiver interface module;

FIG. 58 is a flow chart illustrating the steps performed by thecommunications module for automated data collection from the othermodules connected to the communication network of the bed;

FIG. 59 is a diagrammatical view illustrating a patient status moduleand a gateway module of the present invention;

FIG. 60 is a diagrammatical view illustrating details of a patientcharting module of the present invention;

FIG. 61 is a block diagram illustrating the modular therapy and supportsurface system of the present invention including a plurality of controlmodules for controlling various air therapy devices and surface sectionsof a support surface and illustrating an air supply module forcontrolling an air handling unit and a switching valve to selectivelysupply air pressure and a vacuum to the various therapy devices andsurface sections;

FIG. 62 is a diagrammatical illustration of the configuration of an airtherapy control module;

FIG. 63 is an exploded perspective view illustrating a foam surfacefoundation with side bolsters configured to be positioned on a deck ofthe bed, an upper foam support surface, and an inflatable and deflatablesurface foot section;

FIG. 64 is a perspective view illustrating the surface foot section inan inflated configuration when the bed is in a normal bed position andillustrating the surface foot section in a retracted and collapsedconfiguration when the bed is in a chair position;

FIG. 65 is a diagrammatical view further illustrating how the surfacefoot section retracts or shortens and collapses or thins as the bedmoves from the bed position to the chair position;

FIG. 66 is a diagrammatical view of the control module and bladderconfiguration of the surface foot section;

FIG. 67 is a partial perspective view with portions broken awayillustrating another embodiment of the surface foot section;

FIG. 68 is an exploded perspective view of another embodiment of thepresent invention illustrating a pulmonary therapy rotational bladderlocated between a deck of the bed and the surface foundation andillustrating an upper air bladder support surface located above thesurface foundation in place of the upper foam support surface of FIG.61;

FIG. 69 is a diagrammatical end view illustrating the configuration ofthe modular therapy and support surface of the present invention whenthe pulmonary bladders are all deflated;

FIG. 70 is a diagrammatical view similar to FIG. 66 illustratinginflation of left side pulmonary bladders to rotate a patient to theright;

FIG. 71 is a diagrammatical view similar to FIGS. 66 and 67 illustratinginflation of the right side pulmonary bladders to rotate the patient tothe left;

FIG. 72 is a block diagram illustrating another embodiment of thepresent invention illustrating separate exchangeable surfaces or therapydevices which are each coupled to a control module including pneumaticcontrol valves and sensors, an electrical connection, and a processorfor communicating with an air and power handling unit on the bed andwith a graphical interface display on the bed through the electricalcommunication network of the bed; and

FIG. 73 is a block diagram illustrating the support surface system ofthe present invention including a plurality of a bed articulationcontrol module controlling movement of the articulating deck sectionsand illustrating a surface instrument module and an air supply modulefor controlling an air handling unit and a switching valve toselectively supply air pressure and a vacuum to control inflation anddeflation of zones of the support surface.

DETAILED DESCRIPTION OF THE ILLUSTRATIVE AND PREFERRED EMBODIMENTS

A chair bed 50 in accordance with the present invention having a headend 52, a foot end 54, and sides 56, 58 is illustrated in FIG. 1. Asused in this description, the phrase “head end 52” will be used todenote the end of any referred-to object that is positioned to lienearest head end 52 of chair bed 50. Likewise, the phrase “foot end 54”will be used to denote the end of any referred-to object that ispositioned to lie nearest foot end 54 of chair bed 50.

Chair bed 50 includes a base module 60 having a base frame 62 connectedto an intermediate frame module 300 by lift arms 320, 322, 324, 326 asshown in FIGS. 1, 11 and 43. An articulating deck/weigh frame module 400is coupled to intermediate frame module 300 by load beams 330, 336, 342,348. Siderail assemblies 800, 802, 804, 806 and an extended frame module610 having a swinging foot gate 622 are coupled to articulatingdeck/weigh frame module 400. A mattress 550 is carried by articulatingdeck/weigh frame module 400 and provides a sleeping surface or supportsurface 552 configured to receive a person (not shown).

Chair bed 50 can be manipulated by a caregiver or by a person (notshown) on sleeping surface 552 using hydraulic system module 100 so thatmattress 550, an intermediate frame 302 of intermediate frame module300, and an articulating deck 402 of articulating deck/weigh framemodule 400 assume a variety of positions, several of which are showndiagrammatically in FIGS. 3-7.

Articulating deck 402 includes a head section 404, a seat section 406, athigh section 408, and a foot section 410. Mattress 550 rests on deck402 and includes a head portion 558, a seat portion 560, a thigh portion562, and a foot portion 564, each of which generally corresponds to thelike-named portions of deck 402, and each of which is generallyassociated with the head, seat, thighs, and feet of the person onsleeping surface 552. Details of deck 402 and mattress 550 will beexplained hereinafter.

Chair bed 50 can assume a bed position having deck 402 configured sothat sleeping surface 552 is planar and horizontal, defining an initialposition of deck 402 as shown in FIG. 1 and as shown diagrammatically inFIG. 3. In the bed position, sleeping surface 552 is a predeterminedfirst distance 566 above the floor. Chair bed 50 can also be manipulatedto assume a low position shown diagrammatically in FIG. 4 having deck402 in the initial position and having sleeping surface 552 apredetermined second distance 568 above the floor, the second distance568 being smaller than first distance 566. The foot section 410 of thearticulating deck 402 has a first length 465 when the deck 402 is in theinitial position.

Chair bed 50 can be moved to a Trendelenburg position showndiagrammatically in FIG. 5 having deck 402 in a planar configuration andtilted so that head end 52 of sleeping surface 552 is positioned to liecloser to the floor than foot end 54 of sleeping surface 552. Chair bed50 can also achieve a reverse Trendelenburg position showndiagrammatically in FIG. 6 having deck 402 in a planar configuration andtilted so that foot end 54 of sleeping surface 552 is positioned to liecloser to the floor than head end 52 of sleeping surface 552.

As described above, chair bed 50 is convertible to a sitting positionshown in FIG. 2 and shown diagrammatically in FIG. 8. In the sittingposition, head end 52 of head section 404 of deck 402 is pivotedupwardly away from intermediate frame 302 to a back-support positionproviding a pivotable backrest so that head section 404 and intermediateframe 302 form an angle 512 generally between 55 and 90 degrees. Seatsection 406 of deck 402 is positioned to lie generally horizontally asin the initial position, foot end 54 of thigh section 408 is slightlyupwardly inclined, and foot section 410 of deck 402 extends generallyvertically downwardly from thigh section 408 and has a length 464 thatis shorter than when deck 402 is in the initial position. Foot portion564 of mattress 550 is inflatable and is in a deflated condition whenchair bed 50 is in the sitting position. Foot portion 564 of mattress550 is thinner and shorter when deflated than when inflated.

Chair bed 50 is capable of assuming positions in which head, thigh, andfoot sections 404, 408, 410 of deck 402 are in positions intermediate tothose shown in FIGS. 3 and 8. For example, chair bed 50 can assume anintermediate position shown diagrammatically in FIG. 7 and also shown inFIG. 15, having head end 52 of head section 404 of deck 402 pivotedslightly upwardly from the initial position, seat section 406 positionedto lie in the same generally horizontal plane as in the initialposition, foot end 54 of thigh section 408 raised slightly upwardly fromthe initial position, and foot section 410 being inclined so that footend 54 of foot section 410 lies below head end 52 of foot section 410.

Additionally, articulating deck 402 of chair bed 50 is configured as astep deck 412 as shown illustratively along with illustrative stepmattress 550 in FIGS. 9, 10, and 28-30. The step deck and mattress ofFIGS. 28-30 are those illustrated in FIGS. 3-8. Step deck 412 includesan upper deck 414 and a central, longitudinally extending recess 456defined by a lower deck 430 of step deck 412 and a wall 438 surroundingrecess 456 and connecting lower deck 430 to upper deck 414. Upper deck414 includes longitudinally extending upper deck side portions 417, ahead end upper deck end portion 416, and a foot end upper deck endportion 460.

Mattress 550 includes a generally upwardly-facing sleeping surface 552and a bottom surface 586 that is generally parallel to sleeping surface552 and that is positioned to lie beneath sleeping surface 552. Aperimetral side 578 connects sleeping surface 552 and bottom surface586. A projection 576 is appended to bottom surface 586 and extendsdownwardly therefrom. Preferably, projection 576 is spaced-apart fromsides 578 of mattress 550 and nests in recess 456. Projection 576 mayengage wall 438 of step deck 412 to prevent movement of mattress 550relative to step deck 412 and to maintain the generally central positionof mattress 550 on deck 412.

Preferably, mattress 550 is provided with a thick zone 582 adjacent torecess 456 and projection 576, and a thin zone 580 engaging upper deck414 as shown in FIG. 10. For example, thick zone 582 can be one andone-half times the thickness of thin zone 580. In one preferredembodiment, the thick zone is approximately 7½ inches (19 cm) thick andthe thin zone is 5 inches (12.7 cm) thick. Thick zone 582 is positionedto carry the majority of the weight of a person (shown in phantom)supported on sleeping surface 552 to maximize the comfort of the person.Having perimetral thin zone 580 provides a perimetral portion ofmattress 550 that appears to the person on sleeping surface 552 to befirmer than thick zone 582, facilitating entry onto and exit fromsleeping surface 552 along sides 578 of mattress 550.

As can be seen, step deck 414 and mattress 550 can be used in manyapplications requiring a support surface for supporting a person. Forexample, step deck 414 and mattress 500 can be configured for use as astretcher to be carried by caregivers and as a gurney having step deck414 mounted on a frame with wheels for transporting the person supportedby the gurney.

A general overview of the system architecture will be followed by adescription of the general operation of chair bed 50.

System Architecture

Base module 60, intermediate frame module 300, articulating deck/weighframe module 400, and siderail assemblies 800, 802, 804, 806 areillustratively shown in FIG. 11 and are shown diagrammatically in FIGS.43-47. The solid lines of FIGS. 43-47 represent mechanical connections,the thick short dashed lines represent fluid connections, the thin longdashed lines represent electrical connections, and the double linesrepresent connections to the electronic network. Base module 60,intermediate frame module 300, and articulating deck/weigh frame module400 cooperate with a hydraulic system module 100 to manipulate mattress550 in accordance with commands from the caregiver or from the personsupported by sleeping surface 552. These modules and some connectionstherebetween are described below.

Base Module 60

Base Module 60 includes a base frame 62 on which the components of thechair bed 50 are mounted as shown in FIGS. 11 and 12, anddiagrammatically in FIG. 44. Base module 60 includes a lifting mechanism130 that raises and lowers sleeping surface 552 of chair bed 50 relativeto base frame 62. Much of the electrical, air, and hydraulic componentsof chair bed 50 are located in or on base frame 62.

Head end casters 70, 72, and foot end casters 74, 76 coupled to the baseframe 62. A brake/steer linkage 80 couples the casters 70, 72, 74, 76 tobrake/steer pedals 78 that are connected to base frame 62. Brake/steerpedals 78 are butterfly wheel pedals that can move between a brakingposition locking casters 70, 72, 74, 76 so that casters 70, 72, 74, 76do not rotate, a middle neutral position that allows casters 70, 72, 74,76 to rotate freely, and a steering position having foot end casters 74,76 locked into steer and head end casters 70, 72 free to swivel.

Head end casters 70, 72 are positioned to lie adjacent to head end 52 ofchair bed 50 and foot end casters 74, 76 are spaced-apart from foot end54 of chair bed 50 as shown in FIGS. 11 and 15 to facilitatearticulation of chair bed 50 to the sitting position. Additionally, thisinward positioning of foot end casters 74, 76 closer to the center ofgravity of chair bed 50 maximizes the maneuverability of chair bed 50 inthe steering condition.

Struts 64 are appended to sides 66 of base frame 62 to provide mountingsurfaces for portions of hydraulic system module 100 as shown in FIGS.11-13 and 44. As shown best in FIGS. 12 and 13, illustrative hydraulicsystem module 100 includes lifting mechanism 130 having actuators 132and 142 for individually raising and lowering head end 52 and foot end54 of intermediate frame 302 relative to base frame 62, actuators 150,158, 168, 176 for raising and lowering the head, thigh, and footsections 404, 408, 410 of articulating deck 402 relative to intermediateframe 302, control manifold 186 for selectively controlling actuators132, 142, 150, 158, 168, 176, power unit 112 for providing energy todrive actuators 132, 142, 150, 158, 168, 176, and conduit 122 forconnecting power unit 112 and control manifold 186 to actuators 132,142, 150, 158, 168, 176.

Power unit 112 is preferably a hydraulic power unit and actuators 132,142, 150, 158, 168, 176 are preferably hydraulic cylinders. It will beappreciated, however, that in accordance with the present invention,various mechanical and electromechanical actuators and drivers may beused to raise and lower intermediate frame 302 on base frame 62 as wellas to raise and lower individual deck sections 404, 406, 408, 410relative to intermediate frame 302. As will be explained below, fluidactuators are preferred since they are capable of manual operation witha battery to provide power for electrical control.

It is well known in the hospital bed art that electric drive motors withvarious types of transmission elements including lead screw drives andvarious types of mechanical linkages may be used to cause relativemovement of portions of hospital beds. It is also well known to usepneumatic actuators to actuate and/or move individual portions ofhospital beds. The terms “means for raising or lowering” in thespecification and in the claims, therefore, are intended to cover alltypes of mechanical, electromechanical, hydraulic, and pneumaticmechanisms, including manual cranking mechanisms of all types, forraising and lowering portions of chair bed 50 of the present invention.

The caregiver can adjust the height and angle of inclination of sleepingsurface 552 as shown in FIGS. 3-6 by activating a hydraulically poweredlifting mechanism 130 to control intermediate frame 302 by lift arms320, 322, 324, 326 connected to cylinders 132, 142. A CPR foot pedal 250and emergency Trendelenburg actuator 254 are mounted on base frame 62 tomanually control manifold 186. In addition, CPR foot pedal 250 shown inFIG. 12 may be used as the emergency Trendelenburg actuator 254 whenpivoted upwardly to a raised position.

If chair bed 50 is plugged into an AC outlet (not shown), the caregiveractivates the lifting function with the push of a button. When notplugged in, the caregiver may raise chair bed 50 by pumping one of thehydraulic foot pump pedals 252 located on either side of the base frame64. The caregiver may also place chair bed 50 in the Trendelenburgposition when chair bed 50 is not plugged in or in an emergency byactivating the emergency Trendelenburg actuator 254 located on baseframe 62. If chair bed 50 is equipped with a battery 92, the caregivermay operate any functions of chair bed 50 by pumping the hydraulic footpump pedal 252 and simultaneously pressing the desired function switch.The electrical control of the valves is supported by a battery 92 onbase frame 62.

Base frame 62 also serves as a mounting location for other modules orcomponents such as well as a bed articulation control module 1018,surface electronics, a bed-side communications interface, components ofthe electronic network, bed exit electronics, a night light 1073, apower supply AC/DC converter 1062, and a battery/charge circuit 88.

Hydraulic System Module 100

Hydraulic System Module 100 provides the mechanical power required tomove articulating deck 402 and to raise and lower chair bed 50.Hydraulic system module 100 includes hydraulic cylinders 132, 142, 150,158, 168, 176 that cooperate with linkages to provide these movements.

Movements of head, thigh, and foot sections 404, 408, 410 ofarticulating deck 402 and the raising and lowering of intermediate frame302 of chair bed 50 illustratively shown in FIGS. 3-8 are accomplishedwith hydraulic system module 100. The illustrative system comprises ahydraulic power unit 112, conduit 122, a valve or control manifold 186,and cylinders 132, 142, 150, 158, 168, 176 as shown in FIG. 13.Hydraulic power unit 112 comprises an electric motor 124, a pump 116driven by electric motor 124, a manual pump 118, and a reservoir 120containing hydraulic oil.

Pump 116 pressurizes hydraulic oil when chair bed 50 is connected to ACpower, which in turn moves piston rods 134, 144, 152, 160, 170, 178inside of cylinders 132, 142, 150, 158, 168, 176 to articulate chair bed50. When chair bed 50 is not connected to AC power, manual pump 118 canbe used, via a foot pump pedal 250 mounted on base frame 62 and coupledto manual pump 118, to pressurize the hydraulic oil and cause pistonrods 134, 144, 152, 160, 170, 178 to move. Manually activated valves212, 214 in valve manifold 162 make it possible for the caregiver torapidly lower head section 404 to a horizontal CPR position and to takeadvantage of a manual Trendelenburg feature to manually move chair bed50 to the Trendelenburg position, illustratively shown in FIG. 5, whenAC power is not available.

For chair beds 50 equipped with a battery 92, the caregiver may use anyof the nurse control functions by pumping foot pump pedal 252 andsimultaneously pressing the desired nurse control function on thesiderail assemblies 800, 802, 804, 806. The caregiver supplies thehydraulic power via the foot pump pedal 252, and battery 92 supplieselectrical power to open or close the valves on valve manifold 186 inillustrative chair bed 50.

Intermediate Frame Module 300

Intermediate Frame Module 300 includes intermediate frame 302 which issupported and positioned via lift arms 320, 322, 324, 326 of liftingmechanism 130 of base frame 62. Intermediate frame 302 in turn supportsarticulating deck/weigh frame module 400 and thus couples articulatingdeck/weigh frame module 400 to lifting mechanism 130 as shown in FIG. 11and shown diagrammatically in FIG. 45.

Intermediate frame 302 includes four load beams 330, 336, 338, 342 thatmovably couple weigh frame 506 of articulating deck/weigh frame module400 to intermediate frame 302. Each load beam 330, 336, 342, 348includes a housing 334, 340, 346, 352 and a sensing end 332, 338, 344,350 that is movable relative to housing 334, 340, 346, 352. The detailsof load beam 330 is discussed herein with reference to FIG. 14 a. Eachload beam 330, 336, 342, 348 additionally comprises a transducer (notshown) connected to sensing ends 332, 338, 344, 350 that provides anelectrical signal in response to movement of sensing end 332, 338, 344,350 relative to housing 334, 340, 346, 352. The extent of the movementof sensing ends 332, 338, 344, 350 depends upon the amount of weightsupported by load beams 330, 336, 342, 348, so that the electricalsignal provided by load beams 330, 336, 342, 348 varies in response tothe weight supported by weigh frame 506.

Load beams 330, 336, 342, 348 can be replaced by dummy beams (not shown)that support weigh frame 506 on intermediate frame 302 but that do notprovide for any movement of weigh frame 506 relative to intermediateframe and that do not provide any electrical signals. When chair bed 50has dummy beams instead of load beams 330, 336, 342, 348, weigh frame506 is fixed to intermediate frame 302 and cooperates therewith toprovide a common frame (not shown). The common frame is used with chairbeds 50 that do not include weigh scales 368 but that include otherfeatures of chair beds 50 described herein.

Intermediate frame 302, illustratively shown in FIG. 14, includespermanent IV poles 376, an oxygen tank holder 380, a mount 310 havingopenings 312 for caregivers to mount added-on IV poles (not shown),mounting locations 304 for bumpers, mounting locations 316 for headboard318 adjacent to head end 52 of intermediate frame 302 as shown in FIG.1, and a drainage bag mount 306 for holding drainage bags (not shown)adjacent to foot end 54 of intermediate frame 302 so that the weight ofadded-on oxygen tanks, IV poles, and drainage bags is not included inthe weight measurement of the person (assuming the chair bed 50 isequipped with weigh scales 368). Intermediate frame 302 is the fixedplatform on which load beams 330, 336, 342, 348, which are weightsensing components of the weigh scales 368, are mounted and weigh frame506 is mounted to intermediate frame 302 by load beams 330, 336, 342,348. Any equipment (not shown) mounted to the intermediate frame 302will not be weighed.

Intermediate frame 302 moves upwardly and downwardly relative to baseframe 62, so that weigh frame 506, articulating deck 402, mattress 550,and extended frame module 610 connected to weigh frame 506, which aresupported thereon as shown in FIG. 11, also move upwardly and downwardlyrelative to base frame 62. The movable head, thigh, and foot sections404, 408, 410 of articulating deck 402 move upwardly and downwardlyrelative to weigh frame 506, and seat section 406 is fixed relative toweigh frame 506.

Intermediate frame 302 provides a place off of weigh frame 506 formounting equipment. For chair beds 50 equipped with weigh scales 368,the caregiver may wish to exclude the weights of added-on componentssuch as IV bags (not shown) and drainage bags (not shown) from theweight of the patient. Mounting drainage bag mount 306 and IV pole mount310 on intermediate frame 302 makes this possible.

Articulating Deck/Weigh Frame Module 400

Articulating Deck/Weigh Frame Module 400 includes mattress 550 thatrests on four sections, head section 404, seat section 406, thighsection 408, and foot section 410 of articulating deck 402 as shown inFIGS. 11, 28-30, and 46. The sections 404, 406, 408, 410 of articulatingdeck 402 are movable to change the position of a person supported onsleeping surface 552 of mattress 550. For chair beds 50 equipped withweigh scales 368, deck 402 and a weigh frame 506, which supports deck402 and is interposed between deck 402 and intermediate frame 302, areequivalent to a weigh platform of a platform scale, i.e., anythingresting on deck 402 will be weighed when the weigh scales 368 are used.For chair beds 50 that are not equipped with weigh scales 368, deck 402and weigh frame 506 are fixed together by dummy beams (not shown) toform a common frame (not shown).

Articulating deck 402 is the surface upon which the mattress 550 rests.Deck 402 is illustratively segmented into head, seat, thigh, and footsections 404, 406, 408, 410, three of which, head section 404, thighsection 408, and foot section 410, may be rotated to change the angle ofinclination of the back, thighs, and lower legs of the person (notshown) with respect to seat section 408. Head section 404 has a special“reduced-shear pivot” which is the movement produced by a reduced-shearpivot assembly 650 to be described hereinafter that causes head section404 to pivot about an effective pivot axis 652 that is positioned to lieabove lower deck section 510 and that is preferably at upper deck 414 asshown in FIGS. 16 and 17. Seat section 406 of deck 402 remainshorizontal and the head, thigh, and foot sections 404, 408, 410 of deck402 can move relative to the seat section 406 and relative to eachother, thereby moving the head, thigh, and foot portions 558, 562, 564of mattress 550 relative to seat portion 560 of mattress 550 andrelative to each other.

Articulating deck 402 is mounted to weigh frame 506. Actuators orcylinders 150, 158, 168, 176, that power the movement of head, thigh,and foot sections 404, 408, 410 of deck 402, are also mounted to weighframe 506 as shown in FIGS. 11, 14, and 15. Articulating deck/weighframe module 400 is, in turn, supported by intermediate frame module300. The interface between articulating deck/weigh frame module 400 andintermediate frame module 300 is illustratively limited to fourattachments as shown in FIG. 14. For beds equipped with weigh scales368, load beams 330, 336, 342, 348 are located at these points. Forchair beds that are not equipped with weigh scales 368, or “non-scalechair beds,” the modules are rigidly coupled.

Articulating deck/weigh frame module 400 may also carry other componentsof chair bed 50. For example, details 304 on the articulating deck 402,shown in FIG. 11, make it possible for caregivers to tie restraintstraps (not shown) to deck 402 when required. While head sectionsiderails 808, 810 are mounted to head section 404, body sectionsiderails 812, 814 are mounted to weigh frame 506 by brackets 816, 818.In a preferred embodiment, head siderails 808, 810 are mounted tobreakaway mounting brackets or collateral deck portions 922, 924. Othermodules or components that may be attached to articulating deck/weighframe module 400 include, for example, a removable foot prop 646 forsupporting the feet of the person on sleeping surface 552 duringmovement between the bed position and the sitting position, a footsafety switch 648, and extended frame module 610.

Extended Frame Module 610

Extended Frame Module 610, shown in FIG. 11 and shown diagrammaticallyin FIG. 46, includes an extended U-shaped frame 612 at the foot end 54of the chair bed 50 and mounted to weigh frame 506, extended frame 612providing a location for mounting caregiver controls, traction equipment(not shown), handles for transport (not shown), a utility shelf 644, andbumpering (not shown). The design of chair bed 50 provides for egress oringress of the person at foot end 54 of chair bed 50, particularly whenchair bed 50 is converted to the sitting position shown in Fig. anddiagrammatically in FIG. 8.

Extended frame module 610 includes a foot gate 622 having swinging gates626, 634, one swinging gate 626, 634 mounted on either side of chair bed50 as shown in FIGS. 1, 2, and 11. Gates 626, 634 can swing outwardlyaway from chair bed 50 to provide the person a clear path out of chairbed 50 for easy egress from the sitting position while also providingthe caregiver clear access to the patient. Foot section 410 ofarticulating deck 402 and foot portion 564 of mattress 550 rotatethrough the U-shaped extended frame 612 when foot section 410 movesbetween the up position and the down position.

Siderail Assemblies 800, 802, 804, 806

Siderail Assemblies 800, 802, 804, 806 include siderails 808, 810, 812,814, which are passive restraint devices mounted on both sides of chairbed 50 as shown in FIGS. 1, 2, 11, 31-38, and diagrammatically in FIG.47. In the upward patient-restraining position, siderails 808, 810, 812,814 are vertical barriers extending above sleeping surface 552 torestrain movement of the person past sides 554, 556 of sleeping surface552. Siderails 808, 810, 812, 814 may also be lowered below sleepingsurface 552 of mattress 550 to permit the person to move past sides 554,556 of sleeping surface 552 when entering and exiting chair bed 50 or togive the caregiver clear access to the patient.

Lowering each siderail 808, 810, 812, 814 is accomplished by pulling arelease handle 862. After pulling release handle 862, the caregiver maylet go of release handle 862 and allow siderail 808, 810, 812, 814 torotate downwardly and tuck into the tucked position under deck 402. Therate at which each siderail 808, 810, 812, 814 rotates downwardly ispreferably controlled by a mechanical damper 868. To raise siderails808, 810, 812, 814, the caregiver pulls upwardly on siderails 808, 810,812, 814 until they lock in the patient-restraining position.

Illustratively, there are four siderails 808, 810, 812, 814 on chair bed50. Two head section siderails 808, 810 are mounted to head section 404of articulating deck 402, and two body section siderails 812, 814 aremounted to move or stay with seat section 406 of deck 402, seat section406 and siderails 812, 814 being fixed relative to weigh frame 506.

Siderails 808, 810, 812, 814 can be provided with mechanical angleindicators 938 that provide a visual indication of the angularorientation of siderails 808, 810, 812, 814 relative to the floor. Headsection siderails 808, 810 move with head section 404 of deck 402 ashead section 404 pivots between the down position and the back-supportposition, so that angle indicators 938 mounted to head section siderails808, 810 generally indicate the angular orientation of head section 404.Likewise, body section siderails 812, 814 are generally fixed in anangular orientation relative to intermediate frame 302 so that angleindicators 938 mounted to body section siderails 812, 814 generallyindicate the angular orientation of intermediate frame 302.

Body section siderails 812, 814 can also be provided with a hip pivotguide 694 shown in FIGS. 31-33 to help the caregiver to properlyposition the hip (not shown) of the person (not shown) on sleepingsurface 552. Proper positioning of the hip operates to maximize theeffectiveness of the reduced-shear pivot.

Besides serving as passive restraints, siderails 808, 810, 812, 814 alsoserve as a mounting location for nurse controls 1028, 1030, patientcontrols 1156, 1160 and entertainment modules. These modules arereferred to as human interface control modules. These interface controlmodules output the occurrence of any switch activation into theelectronic network. In addition, siderails 808, 810, 812, 814 maypreferably contain the necessary hardware to allow patient-to-nursecommunications (not shown) and entertainment audio output (not shown).

Detailed Description of Modules and Systems Hydraulic System Module 100

Actuators 132, 142, 150, 158, 168, 176 are preferably hydraulicactuators. For example, head end actuator 132 is a lift cylinder asshown in FIG. 12 having an interior region 133 shown diagrammatically inFIG. 13 and a piston rod 134 slidably received by interior region 133.Head end lift cylinder 132 is formed to include a front port 136 and arear port 138, each of which are in fluid communication with interiorregion 133. When pressurized fluid such as hydraulic oil is received byrear port 138, the pressurized fluid pushes piston rod 134 toward frontport 136 and causes an end 135 of piston rod 134 to extend out of andmove away from lift cylinder 132. At the same time, non-pressurizedfluid escapes from front port 136 and is received by a return conduit185 in fluid communication with a reservoir 120. Likewise, ifpressurized fluid were to be received by front port 136, it would act onpiston rod 136 to slide piston rod 136 toward rear port 138, therebyretracting end 135 into lift cylinder 132 and releasing non-pressurizedfluid into return line 185 and reservoir 120. This allows actuators 132,142, 150, 158, 168, 176 to be hydraulically locked.

Hydraulic power unit 112 is mounted on base frame 62 and includesreservoir 120, pump 116 which is driven by electric motor 124, andmanual pump 118 which is driven by foot pump pedal 252 as shown in FIGS.12, 12 a, and 13. Hydraulic power unit 112 operates to pressurize afluid such as hydraulic oil which is stored at atmospheric pressure inreservoir 120. The pressurized hydraulic oil is supplied to controlmanifold 186 which in turn selectively supplies the pressurizedhydraulic oil to actuators 132, 142, 150, 158, 168, 176.

Pump 116 receives the hydraulic oil from reservoir 120, pressurizes thehydraulic oil, and supplies the pressurized hydraulic oil to apressurized oil manifold 184 of control manifold 186 as shown in FIG.13. Control valves of control manifold 186 receive the pressurizedhydraulic oil and each control valve either supplies the pressurizedhydraulic oil to the actuator or blocks the flow of the hydraulic oil tothe actuator, depending upon the state of the control valve. The controlvalves are typically either three-way valves or they are two-way valvesthat cooperate with companion two-way valves to supply pressurizedhydraulic oil to the actuators or to receive hydraulic oil from theactuators and divert the hydraulic oil from the actuators to returnconduit 185 that returns non-pressurized hydraulic oil to reservoir 120.Thus, the control valves operate to control the flow of pressurizedhydraulic oil between hydraulic power unit 112 and actuators 132, 142,150, 158, 168, 176.

Lifting mechanism 130 includes head end actuator 132 to raise and lowerhead end 52 of intermediate frame 302 and foot end actuator 142 to raiseand lower foot end 54 of intermediate frame 302 as shown in FIG. 13. Ahead end rear first valve 188, a head end rear second valve 190, and anemergency Trendelenburg valve 214 control the flow of fluid between rearport 138 of head end actuator 132 and hydraulic power unit 112. A headend front pilot operated check valve 220 controls the flow of fluidbetween front port 136 of head end actuator 132 and hydraulic power unit112. The raising and lowering of head end 52 of intermediate frame 302will provide the most satisfactory results when the operation of valves188, 190, 214, 220 is coordinated as described below.

First valve 188 is a two-way valve interconnecting pressurized oilmanifold 184 and conduit 122 that is in fluid communication with rearport 138 of head end lift cylinder 132 as shown in FIG. 13. In addition,a head end lift pilot line 236 is in fluid communication with rear port138 so that when first valve 188 is activated, as shown in FIG. 13,first valve 188 blocks the flow of pressurized hydraulic oil frompressurized oil manifold 184 to both pilot line 236 and rear port 138.When first valve 188 is deactivated, fluid communication is restoredbetween pressurized oil manifold 184 and both pilot line 236 and rearport 138 so that pressurized hydraulic oil can flow to both rear port138 and pilot line 236.

Second valve 190 is a two-way valve coupled to return conduit 185 andcoupled by conduit 122 to rear port 138 of head end lift cylinder 132.When second valve 190 is deactivated as shown in FIG. 13, second valve190 blocks the flow of hydraulic oil between rear port 138 and returnconduit 185. When second valve 190 is activated, fluid communication isrestored between rear port 138 and return conduit 185 to allow hydraulicoil to flow from rear port 138 of head end lift cylinder 132 toreservoir 120. Typically when first valve 188 is deactivated to restorefluid communication between pressurized oil manifold 184 and rear port138, second valve 190 is also deactivated to block fluid communicationbetween rear port 138 and return conduit 185.

Emergency Trendelenburg valve 214 is a two-way valve coupled to returnconduit 185 and coupled by conduit 122 to rear port 138 of head end liftcylinder 132. When emergency Trendelenburg valve 214 is deactivated asshown in FIG. 13, emergency Trendelenburg valve 214 blocks the flow ofhydraulic oil from rear port 138 to return conduit 185. When emergencyTrendelenburg valve 214 is activated, fluid communication between rearport 137 and return conduit 185 is restored so that hydraulic oil canflow from rear port 138 to reservoir 120 bypassing second valve 190.Unlike first and second valves 188, 190 which are typicallyelectronically activated, emergency Trendelenburg valve 214 is activatedby a manual actuator 254 such as an emergency Trendelenburg lever, showndiagrammatically in FIG. 13. Emergency Trendelenburg valve can also beactivated by pulling CPR pedal 250 upwardly. Typically, when emergencyTrendelenburg valve 214 is activated to restore fluid communicationbetween rear port 138 and return conduit 185, first valve 188 isactivated to block fluid communication between pressurized oil manifold184 and rear port 138.

Pilot operated check valve 220 is a two-way valve coupled to returnconduit 185 and coupled by conduit 122 to front port 136 of head endlift cylinder 132. Check valve 220 is deactivated when head end liftpilot line 236 is not in fluid communication with pressurized oilmanifold 184 as shown in FIG. 13. When pilot line 236 is in fluidcommunication with pressurized oil manifold 184, pilot operated checkvalve 220 is activated. Thus, check valve 220 is activated when firstvalve 188 is deactivated to restore the fluid communication betweenpilot line 236 and pressurized oil manifold 184, and check valve 220 isdeactivated when first valve 188 is activated to block the fluidcommunication between pilot line 236 and pressurized oil manifold 184.

When pilot operated check valve 220 is deactivated, hydraulic oil canflow through check valve 220 only in a direction from return conduit 185to front port 136 as shown in FIG. 13. When check valve 220 isactivated, hydraulic oil can flow through check valve 220 either fromfront port 136 to return conduit 185 or from return conduit 185 to frontport 136. Thus, when first valve 188 is deactivated to restore fluidcommunication between pressurized oil manifold 184, pilot line 236, andrear port 138, hydraulic oil can flow from front port 136, through checkvalve 220, to return conduit 185 and reservoir 120.

To raise the head end 52 of intermediate frame 302, first valve 188 isdeactivated to restore fluid communication between pressurized oilmanifold 184, pilot line 236, and rear port 138, second valve 190 andemergency Trendelenburg valve 214 are deactivated to block fluidcommunication between rear port 138 and return conduit 185, and pilotoperated check valve 220 is activated to allow the flow of hydraulic oilfrom front port 136 to return conduit 185. As pressurized hydraulic oilflows from pressurized oil manifold 184, through first valve 188,through rear port 138, and into interior region 133, piston rod 134 ispushed toward front port 136 and end 135 of piston rod 134 extends fromlift cylinder 132 lifting head end 52 of intermediate frame 302 throughlinkages between head end 52 of intermediate frame 302 and end 135 ofpiston rod 134 described below. As piston rod 134 is pushed toward frontport 136, hydraulic oil flows out of interior region 133 through frontport 136, through check valve 220, through return conduit 185, toreservoir 120.

To lower head end 52 of intermediate frame 302, first valve 188 isactivated to block the fluid communication between pressurized oilmanifold 184 and both pilot line 236 and rear port 138. Blocking fluidcommunication between pressurized oil manifold 184 and pilot operatedcheck valve 220 deactivates check valve 220 so that check valve 220blocks the flow of hydraulic oil from front port 136 to return conduit185 but allows the flow of hydraulic oil from return conduit 185 tofront port 136. Either one or both of second valve 190 and emergencyTrendelenburg valve 214 are activated to restore fluid communicationbetween rear port 138 and return conduit 185. The weight of intermediateframe 302 and articulating deck/weigh frame module 400 is sufficient topush piston rod 134 toward rear port 138 to retract end 135 of pistonrod 134 into head end lift cylinder 132 and to push hydraulic oil frominterior region 133, through rear port 138, through either one or bothof second valve 190 and emergency Trendelenburg valve 214, and to returnconduit 185 and reservoir 120. The retraction of piston rod 134 intohead end lift cylinder 132 lowers head end 52 of intermediate frame 302through linkages between head end 52 of intermediate frame 302 and end135 of piston rod 134 described below.

Lifting mechanism 130 also includes foot end actuator 142 to raise andlower foot end 54 of intermediate frame 302 as shown in FIG. 13. A footend rear first valve 192, a foot end rear second valve 194, and ableed-off valve 216 control the flow the fluid between rear port 146 offoot end actuator 142 and hydraulic power unit 112. Unlike head endactuator 132, foot end actuator 142 includes no front port.

First valve 192 is a two-way valve coupled to pressurized oil manifold184 and coupled by conduit 122 to rear port 146 of foot end liftcylinder 142. When first valve 192 is activated, as shown in FIG. 13,first valve 192 blocks the flow of pressurized hydraulic oil frompressurized oil manifold 184 to rear port 146. When first valve 192 isdeactivated, fluid communication is restored between pressurized oilmanifold 184 and rear port 146 allowing pressurized hydraulic oil toflow thereto.

Second valve 194 is a two-way valve coupled to return conduit 185 andcoupled by conduit 122 to rear port 146 of foot end lift cylinder 142.When second valve 194 is deactivated as shown in FIG. 13, second valve194 blocks the flow of hydraulic oil from rear port 146 to returnconduit 185. When second valve 194 is activated, fluid communication isrestored between rear port 146 and return conduit 185 so that hydraulicoil can flow from rear port 146 of foot end lift cylinder 142 to returnconduit 185 and to reservoir 120.

Bleed-off valve 216 is a two-way valve coupled to return conduit 185 andcoupled by conduit 122 to rear port 146 of foot end lift cylinder 142 asshown in FIG. 13. When bleed-off valve 216 is closed the flow ofhydraulic oil from rear port 146 to return conduit 185 through bleed-offvalve 216 is blocked. When bleed-off valve 216 is open, fluidcommunication is restored between return conduit 185 and rear port 146to allow hydraulic oil to flow from rear port 146 of foot end liftcylinder 142, through bleed-off valve 216, to return conduit 185 and toreservoir 120. Unlike first and second valves 192, 194 which aretypically electronically activated, bleed-off valve 216 is activatedmanually such as by turning a member (not shown) of bleed-off valve 216to move bleed-off valve 216 between the open and closed positions.

To raise the foot end 54 of intermediate frame 302, first valve 192 isdeactivated to restore fluid communication between pressurized oilmanifold 184 and rear port 146, and second valve 194 is deactivated andbleed-off valve 216 is closed to block fluid communication between rearport 146 and return conduit 185. As pressurized hydraulic oil flows intolift cylinder 142 from pressurized oil manifold 194, through first valve192, and through rear port 146, piston rod 144 is pushed forward toextend therefrom and acts through linkages between foot end 54 ofintermediate frame 302 and piston rod 144 described below to lift footend 54 of intermediate frame 302.

To lower foot end 54 of intermediate frame 302, first valve 192 isactivated to block the fluid communication between pressurized oilmanifold 184 and rear port 146 of foot end lift cylinder 142. Eithersecond valve 194 can be activated or bleed-off valve 216 can be openedto restore fluid communication between rear port 146 and return conduit185. The weight of intermediate frame 302 and articulating deck/weighframe module 400 is sufficient to push piston rod 144 toward rear port146 thereby retracting piston rod 144 into foot end lift cylinder 142,and to push hydraulic oil out of foot end lift cylinder 142, throughrear port 146, and through either one or both of second valve 194 andbleed-off valve 216 to return conduit 185 and reservoir 120. Theretraction of piston rod 144 into foot end lift cylinder 142 lowers footend 54 of intermediate frame 302 through linkages between foot end 54 ofintermediate frame 302 and piston rod 144 described below.

Head section 404 is movable between a generally horizontal down positionand an upward back-support position providing a pivotable backrest. Headsection pivot cylinder 150 is pivotably coupled to weigh frame 506 asshown in FIGS. 15-17 and has a piston rod 152 pivotably coupled to headsection 404 as described below. A head section rear first valve 196, ahead section rear second valve 198, and a CPR valve 212 shown in FIG. 13control the flow of fluid between rear port 154 of head section pivotcylinder 150 and hydraulic power unit 112.

First valve 196 is a two-way valve coupled to pressurized oil manifold184 and coupled by conduit 122 to rear port 154 of head section pivotcylinder 150. When first valve 196 is deactivated, as shown in FIG. 13,fluid communication is restored between pressurized oil manifold 184 andrear port 154 allowing pressurized hydraulic oil to flow thereto. Whenfirst valve 196 is activated, first valve 196 blocks fluid communicationbetween pressurized oil manifold 184 and rear port 154.

Second valve 198 is a two-way valve coupled to return conduit 185 andcoupled by conduit 122 to rear port 154 of head section pivot cylinder150. When second valve 198 is deactivated, as shown in FIG. 13, secondvalve 198 blocks the flow of hydraulic oil between rear port 154 andreturn conduit 185. When second valve 198 is activated, fluidcommunication is restored between rear port 154 and return conduit 185to allow hydraulic oil to flow from rear port 154 of head section pivotcylinder 150 to return line 185 and to reservoir 120. Typically, whenfirst valve 196 is deactivated to restore fluid communication betweenpressurized oil manifold 185 and rear port 154, second valve 198 is alsodeactivated to block fluid communication between rear port 154 andreturn conduit 185.

CPR valve 212 is a two-way valve coupled to return conduit 185 andcoupled by conduit 122 to rear port 154 of head section pivot cylinder150. When CPR valve 212 is deactivated, as shown in FIG. 13, CPR valve212 blocks the flow of hydraulic oil from rear port 154 to returnconduit 185. When CPR valve 212 is activated, fluid communicationbetween rear port 154 and return conduit 185 is restored so thathydraulic oil can flow from rear port 154 to reservoir 120. Unlike firstand second valves 196, 198 which are typically electronically activated,CPR valve 212 is activated by a manual activator such as CPR foot pedal250, shown in FIG. 12 and shown diagrammatically in FIG. 13. Typicallywhen CPR valve 212 is activated to restore fluid communication betweenrear port 154 and return conduit 185, first valve 196 is activated toblock fluid communication between pressurized oil manifold 184 and rearport 154. Preferably, conduit 122 coupling CPR valve 212 to returnconduit 185 has a sufficiently large diameter to cause the hydraulic oilto drain rapidly from head section pivot cylinder 150 resulting in rapidmovement of head section 404 from the back-support position to the downposition when CPR valve 212 is activated.

To move head section 404 from the down position to the back-supportposition, first valve 196 is deactivated to restore fluid communicationbetween pressurized oil manifold 184 and rear port 154 of head sectionpivot cylinder 150. Second valve 198 and CPR valve 212 are deactivatedto block fluid communication between rear port 154 and return conduit185. As pressurized hydraulic oil flows from pressurized oil manifold184 through first valve 196 and then through rear port 154 into headsection pivot cylinder 150, piston rod 152 is pushed outwardly to extendfrom head section pivot cylinder 150, thereby lifting head section 404as the result of connections between piston rod 152 and head section 404described below.

To lower head section 404, first valve 196 is activated to block thefluid communication between pressurized oil manifold 184 and rear port154, and either one or both of second valve 198 and CPR valve 212 areactivated to restore fluid communication between rear port 154 andreturn conduit 185. The weight of head section 404 is sufficient to pushpiston rod 152 toward rear port 154 thereby retracting piston rod 152into head section pivot cylinder 150. As piston rod 152 retracts intohead section pivot cylinder 150, hydraulic oil is pushed through rearport 154, through either one or both of second valve 198 and CPR valve212, and to return conduit 185 and reservoir 120. The retraction ofpiston rod 152 into head section pivot cylinder 150 lowers head section404 as the result of the linkages connecting piston rod 152 and headsection 404 described below.

Thigh section 408 of articulating deck 402 is movable between agenerally horizontal down position and a slightly inclined up positionshown diagrammatically in FIG. 7 and shown in FIGS. 2 and 15. Thighsection pivot cylinder 158 is coupled to thigh section 408 as shown inFIG. 13 to move thigh section 408 between the up position and the downposition. A thigh section front valve 200 and a thigh section frontpilot operated check valve 222 control the flow of fluid between a frontport 162 and hydraulic power unit 112. A thigh section rear valve 202and a thigh section rear pilot operated check valve 224 control the flowof fluid between a rear port 164 and hydraulic power unit 112. Theraising and lowering of thigh section 408 of articulating deck 402 willprovide the most satisfactory results when the operation of valves 200,202, 222, 224 is coordinated as described below.

Rear valve 202 is a three-way valve coupling pressurized oil manifold184 and return manifold 185 to rear port 164 of thigh section pivotcylinder 158. In addition, rear valve 202 couples a thigh section frontpilot line 238 to pressurized oil manifold 184 so that when rear valve202 is activated, as shown in FIG. 13, rear valve 202 restores the flowof pressurized hydraulic oil from pressurized oil manifold 184 to bothrear port 164 and to pilot line 238, thus activating pilot operatedcheck valve 222. When rear valve 202 is deactivated, fluid communicationbetween pressurized oil manifold 184 and both rear port 164 and pilotline 238 is blocked, and fluid communication is restored between rearport 164 and return conduit 185 and reservoir 120 through check valve224.

Front valve 200 is a three-way valve coupling front port 162 of thighsection pivot cylinder 158 to return conduit 185 when front valve 200 isin a deactivated position shown in FIG. 13, and to pressurized oilmanifold 184 when front valve 200 is in an activated position. Whenfront valve 200 is deactivated, front valve 200 blocks the fluidcommunication between front port 162 and pressurized oil manifold 184while restoring the fluid communication between front port 162 andreturn conduit 185. When front valve 200 is activated, fluidcommunication is restored between front port 162 and pressurized oilmanifold 184, while fluid communication between front port 162 andreturn conduit 185 is blocked. In addition, front valve 200 couples athigh section rear pilot line 240 to pressurized oil manifold 184 sothat when front valve 200 is activated fluid communication is restoredbetween pressurized oil manifold 184 and pilot line 240 allowingpressurized hydraulic oil to flow to pilot operated check valve 224 toactivate check valve 224.

Thigh section rear pilot operated check valve 224 is a two-way valvecoupled to rear port 164 and rear valve 202. Check valve 224 isdeactivated when fluid communication between thigh section rear pilotline 240 and pressurized oil manifold 184 is blocked as shown in FIG.13. When pilot line 240 is in fluid communication with pressurized oilmanifold 184, pilot operated check valve 224 is activated. Thus checkvalve 224 is activated when front valve 200 is activated and check valve240 is deactivated when front valve 200 is deactivated as shown in FIG.13.

When check valve 224 is deactivated, hydraulic oil can flow throughcheck valve 224 only in a direction from rear valve 202 to rear port 164as shown in FIG. 13. When check valve 224 is activated, hydraulic oilcan flow through check valve 224 either from rear port 162 to rear valve202 or from rear valve 202 to rear port 162. Thus, when front valve 200is activated to restore fluid communication between pressurized oilmanifold 184, pilot line 240, and front port 162 so that pressurizedhydraulic oil flows from manifold 184 to front port 162, hydraulic oilcan also flow from rear port 164, through check valve 224, to rear valve202. If rear valve 202 is deactivated at the same time that front valve202 is activated, then the hydraulic oil from rear port 264 can flowthrough rear valve 202 to return conduit 185 and reservoir 120.

Likewise, thigh section front pilot operated check valve 222 is atwo-way valve coupled to front port 162 and to front valve 200. Checkvalve 222 is activated when rear valve 202 is activated so that thighsection front pilot line 238 is in fluid communication with pressurizedoil manifold 184 as shown in FIG. 13. When rear valve 202 isdeactivated, pilot line 238 is not in fluid communication withpressurized oil manifold 184 and pilot operated check valve 222 isdeactivated. Thus, check valve 222 is activated when rear valve 202 isactivated and check valve 222 is deactivated when front valve 202 isdeactivated.

When pilot operated check valve 222 is deactivated, hydraulic oil canflow through check valve 222 only in a direction from front valve 200 tofront port 162. When check valve 222 is activated, hydraulic oil canflow through check valve either from front port 162 to front valve 200or from front valve 200 to front port 162. Thus, when rear valve 200 isactivated to restore fluid communication between pressurized oilmanifold 184, pilot line 238, and rear port 164 so that pressurizedhydraulic oil flows from manifold 184 to rear port 164, hydraulic oilcan also flow from front port 162, through check valve 222, to frontvalve 200. If front valve 200 is deactivated when rear valve 202 isactivated, then hydraulic oil from front port 162 can pass through frontvalve 200 to return conduit 185 and reservoir 120.

To raise thigh section 408 of articulating deck 402, rear valve 202 isactivated to restore fluid communication between pressurized oilmanifold 184, pilot line 238, and rear port 164. Front valve 200 isdeactivated to block fluid communication between pressurized oilmanifold 184 and front port 162 and to restore fluid communicationbetween front port 162 and return conduit 185. As pressurized hydraulicoil flows from pressurized oil manifold 184, through rear valve 282,through rear port 164, and into thigh section pivot cylinder 158, pistonrod 160 is pushed toward front port 162 and extends from thigh sectionpivot cylinder 158 to lift thigh section 408 through linkages betweenthigh section 408 and piston rod 160 described below. As piston rod 160is pushed toward front port 162, hydraulic oil flows through front port162, through activated check valve 222, through front valve 200, and toreturn conduit 185 and reservoir 120.

To lower thigh section 408 of articulating deck 402, front valve 200 isactivated to restore the fluid communication between pressurized oilmanifold 184, pilot line 240, and front port 162 of thigh section pivotcylinder 158. Rear valve 202 is deactivated to block the fluidcommunication between pressurized oil manifold 184, pilot line 238, andrear port 164, and to restore fluid communication between rear port 164and return conduit 185. As pressurized hydraulic oil flows frompressurized oil manifold 184, through front valve 200, through frontport 162, and into thigh section pivot cylinder 158, piston rod 160 ispushed toward rear port 164 and is retracted into thigh section pivotcylinder 158, lowering thigh section 408 through linkages between pistonrod 160 and thigh section 408 that are described below. As piston rod160 is pushed toward rear port 164, hydraulic oil flows through rearport 164, through activated check valve 224, through rear valve 202, andto return conduit 185.

Foot section 410 of articulating deck 402 is movable between thegenerally horizontal up position shown in FIGS. 1, 11, and 24 and thegenerally vertically downwardly extending down position showndiagrammatically in FIG. 8 and shown in FIGS. 2 and 25. Foot sectionpivot cylinder 168 is coupled to foot section 410 as shown in FIG. 13 tomove foot section 410 between the up position and the down position. Afoot pivot front valve 204 and a foot pivot front pilot operated checkvalve 226 control the flow of fluid between a front port 172 andhydraulic power unit 112. A foot pivot rear valve 206 and a foot pivotrear pilot operated check valve 228 control the flow of fluid between arear port 174 and hydraulic power unit 112. The raising and lowering offoot section 410 of articulating deck 402 provides the most satisfactoryresults when the operation of valves 204, 206, 226, 228 is coordinatedas described below.

Rear valve 206 is a three-way valve coupling pressurized oil manifold184 and return manifold 185 to rear port 174 of foot section pivotcylinder 168. In addition, rear valve 206 couples a foot pivot frontpilot line 242 to pressurized oil manifold 184 so that when rear valve206 is activated, as shown in FIG. 13, rear valve 206 restores the flowof pressurized hydraulic oil from pressurized oil manifold 184 to bothrear port 174 and to pilot line 242, thus activating pilot operatedcheck valve 226. When rear valve 206 is deactivated, fluid communicationbetween pressurized oil manifold 184 and both rear port 174 and pilotline 242 is blocked, and fluid communication is restored between rearport 174 and return conduit 185 and reservoir 120 through check valve228.

Front valve 204 is a three-way valve coupling front port 172 of footsection pivot cylinder 168 to return conduit 185 when front valve is ina deactivated position, and to pressurized oil manifold 184 when frontvalve 204 is in an activated position shown in FIG. 13. When front valve204 is deactivated, front valve 204 blocks the fluid communicationbetween front port 172 and pressurized oil manifold 184 while restoringthe fluid communication between front port 172 and return conduit 185.When front valve 204 is activated, fluid communication is restoredbetween front port 172 and pressurized oil manifold 184, while fluidcommunication between front port 172 and return conduit 185 is blocked.In addition, front valve 204 couples a foot pivot rear pilot line 244 topressurized oil manifold 184 so that when front valve 204 is activatedfluid communication is restored between pressurized oil manifold 184 andpilot line 244 allowing pressurized hydraulic oil to flow to pilotoperated check valve 228 to activate check valve 228.

Foot pivot rear pilot operated check valve 228 is a two-way valvecoupled to rear port 174 and rear valve 206. Check valve 228 isdeactivated when fluid communication between foot pivot rear pilot line244 and pressurized oil manifold 184 is blocked. When pilot line 244 isin fluid communication with pressurized oil manifold 184, pilot operatedcheck valve 228 is activated as shown in FIG. 13. Thus check valve 228is activated when front valve 204 is activated and check valve 228 isdeactivated when front valve 204 is deactivated.

When check valve 228 is deactivated, hydraulic oil can flow throughcheck valve 228 only in a direction from rear valve 206 to rear port 174as shown in FIG. 13. When check valve 228 is activated, hydraulic oilcan flow through check valve 228 either from rear port 174 to rear valve206 or from rear valve 206 to rear port 174. Thus, when front valve 204is activated to restore fluid communication between pressurized oilmanifold 184, pilot line 244, and front port 172 so that pressurizedhydraulic oil flows from manifold 184 to front port 172, hydraulic oilcan also flow from rear port 174, through check valve 228, to rear valve206. If rear valve 206 is deactivated at the same time that front valve204 is activated, then the hydraulic oil from rear port 264 can flowthrough rear valve 206 to return conduit 185 and reservoir 120.

Likewise, foot pivot front pilot operated check valve 226 is a two-wayvalve coupled to front port 172 and to front valve 204. Check valve 226is activated when rear valve 206 is activated and foot pivot front pilotline 242 is in fluid communication with pressurized oil manifold 184.When rear valve 206 is deactivated, pilot line 242 is not in fluidcommunication with pressurized oil manifold 184 and pilot operated checkvalve 226 is deactivated as shown in FIG. 13. Thus, check valve 226 isactivated when rear valve 206 is activated and check valve 226 isdeactivated when rear valve 206 is deactivated.

When pilot operated check valve 226 is deactivated, hydraulic oil canflow through check valve 226 only in a direction from front valve 204 tofront port 172. When check valve 226 is activated, hydraulic oil canflow through check valve either from front port 172 to front valve 204or from front valve 204 to front port 172. Thus, when rear valve 206 isactivated to restore fluid communication between pressurized oilmanifold 184, pilot line 242, and rear port 174 so that pressurizedhydraulic oil flows from manifold 184 to rear port 174, hydraulic oilcan also flow from front port 172, through check valve 226, to frontvalve 204. If front valve 204 is deactivated when rear valve 206 isactivated, then hydraulic oil from front port 172 can pass through frontvalve 204 to return conduit 185 and reservoir 120.

To raise foot section 410 of articulating deck 402, rear valve 206 isactivated to restore fluid communication between pressurized oilmanifold 184, pilot line 242, and rear port 174. Front valve 204 isdeactivated to block fluid communication between pressurized oilmanifold 184 and front port 172, and to restore fluid communicationbetween front port 172 and return conduit 185. As pressurized hydraulicoil flows from pressurized oil manifold 184, through rear valve 282,through rear port 174, and into foot section pivot cylinder 158, pistonrod 160 is pushed toward front port 172 and extends from foot sectionpivot cylinder 158 to lift foot section 410 through linkages betweenfoot section 410 and piston rod 160 described below. As piston rod 160is pushed toward front port 172, hydraulic oil flows through front port172, through activated check valve 226, through front valve 204, and toreturn conduit 185 and reservoir 120.

To lower foot section 410 of articulating deck 402, front valve 204 isactivated to restore the fluid communication between pressurized oilmanifold 184, pilot line 244, and front port 172 of foot section pivotcylinder 168 as shown in FIG. 13. Rear valve 206 is deactivated to blockthe fluid communication between pressurized oil manifold 184, pilot line242, and rear port 174, and to restore fluid communication between rearport 174 and return conduit 185. As pressurized hydraulic oil flows frompressurized oil manifold 184, through front valve 204, through frontport 172, and into foot section pivot cylinder 168, piston rod 160 ispushed toward rear port 174 and is retracted into foot section pivotcylinder 168, lowering foot section 410 through linkages between pistonrod 160 and foot section 410 that are described below. As piston rod 160is pushed toward rear port 174, hydraulic oil flows through rear port174, through activated check valve 228, through rear valve 206, and toreturn conduit 185.

In addition to pivoting between the up and down positions, foot section410 of articulating deck 402 is also movable between the expandedposition, shown best in FIGS. 11 and 24, and the contracted position,shown best in FIG. 25. Foot section contracting cylinder 176 is coupledto foot section 410 to move foot section 410 between the expandedposition and the contracted position. A foot contracting front valve 208and a foot contracting front pilot operated check valve 230 control theflow of fluid between a front port 180 and hydraulic power unit 112 asshown in FIG. 13. A foot contracting rear valve 210 and a footcontracting rear pilot operated check valve 232 control the flow offluid between a rear port 182 and hydraulic power unit 112. The raisingand lowering of foot section 410 of articulating deck 402 will providethe most satisfactory results when the operation of valve 208, 210, 230,232 is coordinated as described below.

Rear valve 210 is a three-way valve coupling pressurized oil manifold184 and return manifold 185 to rear port 182 of foot section contractingcylinder 176. In addition, rear valve 210 couples a foot contractingfront pilot line 246 to pressurized oil manifold 184 so that when rearvalve 210 is activated the flow of pressurized hydraulic oil frompressurized oil manifold 184 is restored to both rear port 182 and topilot line 246, thus activating pilot operated check valve 230. Whenrear valve 210 is deactivated, as shown in FIG. 13, fluid communicationbetween pressurized oil manifold 184 and both rear port 182 and pilotline 246 is blocked, and fluid communication is restored between rearport 182 and return conduit 185 and reservoir 120 through check valve232.

Front valve 208 is a three-way valve coupling front port 180 of footsection contracting cylinder 176 to return conduit 185 when front valve208 is in a deactivated position and to pressurized oil manifold 184when front valve 208 is in an activated position shown in FIG. 13. Whenfront valve 208 is deactivated, front valve 208 blocks the fluidcommunication between front port 180 and pressurized oil manifold 184while restoring the fluid communication between front port 180 andreturn conduit 185. When front valve 208 is activated, fluidcommunication is restored between front port 180 and pressurized oilmanifold 184, while fluid communication between front port 180 andreturn conduit 185 is blocked. In addition, front valve 208 couples afoot contracting rear pilot line 248 to pressurized oil manifold 184 sothat when front valve 208 is activated fluid communication is restoredbetween pressurized oil manifold 184 and pilot line 248 allowingpressurized hydraulic oil to flow to pilot operated check valve 232 toactivate check valve 232.

Foot contracting rear pilot operated check valve 232 is a two-way valvecoupled to rear port 182 and rear valve 210. Check valve 232 isdeactivated when fluid communication between foot contracting rear pilotline 248 and between pressurized oil manifold 184 is blocked. When pilotline 248 is in fluid communication with pressurized oil manifold 184 asshown in FIG. 13, pilot operated check valve 232 is activated. Thuscheck valve 232 is activated when front valve 208 is activated and checkvalve 232 is deactivated when front valve 208 is deactivated.

When check valve 232 is deactivated, hydraulic oil can flow throughcheck valve 232 only in a direction from rear valve 210 to rear port 182as shown in FIG. 13. When check valve 232 is activated, hydraulic oilcan flow through check valve 232 either from rear port 182 to rear valve210 or from rear valve 210 to rear port 182. Thus, when front valve 208is activated to restore fluid communication between pressurized oilmanifold 184, pilot line 248, and front port 180 so that pressurizedhydraulic oil flows from manifold 184 to front port 180 so thatpressurized hydraulic oil flows from manifold 184 to front port 180,hydraulic oil can also flow from rear port 182, through check valve 232,to rear valve 210. If rear valve 210 is deactivated at the same timethat front valve 208 is activated, then the hydraulic oil from rear port264 can flow through rear valve 210 to return conduit 185 and reservoir120.

Likewise, foot contracting front pilot operated check valve 230 is atwo-way valve coupled to front port 180 and to front valve 208. Checkvalve 230 is activated when rear valve 210 is activated so that footcontracting front pilot line 246 is in fluid communication withpressurized oil manifold 184. When rear valve 210 is deactivated asshown in FIG. 13, pilot line 246 is not in fluid communication withpressurized oil manifold 184 and pilot operated check valve 230 isdeactivated. Thus, check valve 230 is activated when rear valve 210 isactivated and check valve 230 is deactivated when front valve 208 isdeactivated.

When pilot operated check valve 230 is deactivated, hydraulic oil canflow through check valve 230 only in a direction from front valve 208 tofront port 180. When check valve 230 is activated, hydraulic oil canflow through check valve either from front port 180 to front valve 208or from front valve 208 to front port 180. Thus, when rear valve 210 isactivated to restore fluid communication between pressurized oilmanifold 184, pilot line 246, and rear port 182 so that pressurizedhydraulic oil flows from manifold 184 to rear port 182, hydraulic oilcan also flow from front port 180, through check valve 230, to frontvalve 208. If front valve 208 is deactivated when rear valve 210 isactivated, then hydraulic oil from front port 180 can pass through frontvalve 208 to return conduit 185 and reservoir 120.

To expand foot section 410 of articulating deck 402, rear valve 210 isactivated to restore fluid communication between pressurized oilmanifold 184, pilot line 246, and rear port 182. Front valve 208 isdeactivated to block fluid communication between pressurized oilmanifold 184 and front port 180, and to restore fluid communicationbetween front port 180 and return conduit 185. As pressurized hydraulicoil flows from pressurized oil manifold 184, through rear valve 282,through rear port 182, and into foot section contracting cylinder 176,piston rod 160 is pushed toward front port 180 and extends from footsection contracting cylinder 176 to expand foot section 410 throughlinkages between foot section 410 and piston rod 160 described below. Aspiston rod 160 is pushed toward front port 180, hydraulic oil flowsthrough front port 180, through activated check valve 230, through frontvalve 208, and to return conduit 185 and reservoir 120.

To contract foot section 410 of articulating deck 402, front valve 208is activated to restore the fluid communication between pressurized oilmanifold 184, pilot line 248, and front port 180 of foot sectioncontracting cylinder 176. Rear valve 210 is deactivated to block thefluid communication between pressurized oil manifold 184, pilot line246, and rear port 182, and to restore fluid communication between rearport 182 and return conduit 185. As pressurized hydraulic oil flows frompressurized oil manifold 184, through front valve 208, through frontport 180, and into foot section contracting cylinder 176, piston rod 160is pushed toward rear port 182 and is retracted into foot sectioncontracting cylinder 176, contracting foot section 410 through linkagesbetween piston rod 160 and foot section 410 that are described below. Aspiston rod 160 is pushed toward rear port 182, hydraulic oil flowsthrough rear port 182, through activated check valve 232, through rearvalve 210, and to return conduit 185.

Illustratively, the control valves can be configured to selectivelyoperate actuators 132, 142, 150, 158, 168, 176 to move chair bed 50 tovarious positions including the sitting position shown diagrammaticallyin FIG. 13. To move chair bed 50 to the sitting position, the valves areconfigured so that piston rod 134 is retracted into head end liftcylinder 132, piston rod 144 is retracted into foot end lift cylinder142, piston rod 152 is extended from head section pivot cylinder 150,piston rod 160 is extended from thigh section pivot cylinder 158, pistonrod 170 is retracted into foot section pivot cylinder 168, and pistonrod 178 is retracted into foot section contracting cylinder 176. Asdescribed above with respect to each individual actuator 132, 142, 150,158, 168, 176 and as shown diagrammatically in FIG. 13, to attain thesitting position requires that head end rear first valve 188 isactivated, foot end rear first valve 192 is activated, foot retractorfront valve 208 is activated, foot section front valve 204 is activated,thigh section rear valve 202 is activated, and head section rear firstvalve 196 is activated. In addition, all other valves are maintained inthe deactivated position. As can be seen, then, the positions of thehead, thigh, foot sections 404, 408, 410 of articulating deck 402, andthe position of intermediate frame 302 relative to base frame 62 can bemanipulated by manipulating the control valves of control manifold 186.

Of note, in preferred embodiments, only two valves—head end rear firstvalve 188 and foot end rear first valve 192—are normally open, the othervalves being normally closed as shown in FIG. 13, so that when all ofthe control valves are deactivated, pressurized hydraulic oil flows onlythrough valve 188 and valve 192. When pressurized hydraulic oil flowsthrough valve 188, piston rod 134 extends from head end lift cylinder132 to lift head end 52 of intermediate frame 302. When pressurizedhydraulic oil flows through valve 192, piston rod 144 extends from headend lift cylinder 142 to lift foot end 54 of intermediate frame 302.Therefore, if hydraulic oil is pressurized when all control valves aredeactivated, intermediate frame 302 will move to the raised position.

In case of an emergency when intermediate frame 302 is in the lowposition, caregiver can simply pump foot pump pedal 252 to raiseintermediate frame 302 even when chair bed 50 is away from an AC powersource. If intermediate frame 302 is not level when caregiver startspumping foot pump pedal 252, hydraulic system 100 will continue to raiseintermediate frame as long as caregiver pumps foot pump pedal 252 untilboth head end 52 and foot end 54 of intermediate frame 302 are in theraised positions.

In addition, conduit 122 connecting pump 116 to each of the controlvalves includes a variable restrictive orifice 234 as shown in FIG. 13.Each restrictive orifice 234 widens and narrows to maintain the pressuredrop across restrictive orifice 234 at a preselected value. This“pressure compensation” operates to cause uniform articulation ofintermediate frame 302 and head, thigh, and foot sections 404, 408, 410of deck 402 irrespective of the distribution of the weight load on deck402. For example, pressure compensation will cause head end 52 and footend 54 of intermediate frame 302 to raise or lower at the same rate evenif the center of gravity of the person (not shown) on sleeping surface552 is positioned to lie near one of the ends 52, 54 of intermediateframe 302.

Further, it can be seen that by bringing, for example, rear port 154 ofhead section pivot cylinder 150 into fluid communication withpressurized oil manifold 184, that head section 404 can be secured inthe back-support position. In addition, by opening, for example, CPRvalve 212, head section 404 can be released and can move downwardlytoward the bed position. Additionally, by closing CPR valve 212 afterhead section 404 has moved away from the back-support position butbefore head section 404 has moved to the down position, head section 404can be secured in an intermediate position between the back-supportposition and the down position. The ability to secure head section 404in an intermediate position is a characteristic of actuator 150 thatlikewise holds true for actuators 132, 142, 158, 168, 176 so that whenthe actuators cooperate with lifting mechanism 130 and with the linkagesconnecting the actuators to the head, thigh, and foot sections 404, 408,410 of articulating deck 402, chair bed 50 can be secured in manypositions between the bed position and the sitting position providing afull range of positions of chair bed 50 to meet the needs of manydifferent people.

Remote Operation of the Chair Bed (Away from an Electrical Power Source)

Foot pump pedal 252 shown in FIG. 12 can be pumped by the caregiver tooperate manual pump 118, shown best in FIG. 12 a, to pressurize thehydraulic oil. Foot pump pedal 252 can be used, for example, whenelectrical power is not available to electric motor 124 and pump 116 is,therefore, not operating to pressurize the hydraulic oil. Foot pumppedal 252 is pivotably coupled to base frame 62 for movement between anup position and a down position relative to base frame 62. A lever 253is coupled to foot pump pedal 252 so that when foot pump pedal 252 is inthe down position, lever 253 is pulled to a forward position toward footend 54 of chair bed 50, and when foot pump pedal 252 is in the upposition, lever 253 is pushed to a back position toward head end 52 ofchair bed 50.

Manual pump 118 is mounted to control manifold 186 of hydraulic powerunit 112 as shown in FIG. 12 a. Manual pump 118 includes two cylinders104, each cylinder 104 carrying a piston rod 106. Rods 106 areconfigured to pressurize hydraulic oil when rods 106 are pushed to aretracted position toward foot end 54 of chair bed 50, forcingpressurized hydraulic oil out of cylinders 104 and into pressurized oilmanifold 184. As rods 106 move from the retracted position to anextended position toward head end 52 of chair bed 50, unpressurizedhydraulic oil from reservoir 120 moves into cylinders 104.

Manual pump 118 also includes a bar 108 connecting head end 52 of rods106 together as shown in FIG. 12 a and a block 114 coupled to controlmanifold 186. Block 114 is formed to include guide openings 115 that arepositioned to lie so that rods 106 are received by guide openings 115and travel therethrough as rods 106 reciprocate between the retractedand extended positions. A cable 126 has a first end 127 connected tolever 253 as shown in FIG. 12 and a second end 129 connected through athird guide opening 115 formed in block 114 to bar 108 as shown in FIG.13 a.

Control manifold 186 is formed to include an opening 187 that extendsthrough control manifold 186 so that cable 126 can be configured to liein a generally straight line without having cable 126 between first andsecond ends 127, 129 engaging any portion of chair bed 50. Cable 126runs from bar 108, through third guide opening 115 formed in block 114,through opening 187 formed in control manifold 186, and to lever 253 offoot pump pedal 252. Forming opening 187 through control manifold 186additionally allows for compact placement of hydraulic power unit 112and other components on base frame 62 of chair bed 50. A cylindricalreturn spring 110 is received by cable and is positioned to act againstbar 108 and block 114 to yieldably bias bar 108 toward head end 52 ofchair bed 50.

When foot pump pedal 252 is moved downwardly pulling lever 253 towardfoot end of chair bed 50, lever 253 pulls cable 126 toward foot end 54of chair bed and cable 126 pulls bar 108 and rods 106 toward foot end 54of chair bed 50 so that rods 106 retract into cylinders 104 andpressurize hydraulic oil, forcing the hydraulic oil into pressurized oilmanifold 184. When foot pump pedal 252 is released, return spring 110pushes bar 108 toward head end 52 of chair bed 50, pulling rods 106 totheir extended positions and drawing hydraulic oil from reservoir 120into cylinders 104. At the same time, bar 108 pulls cable 126 throughopenings 115, 187, pulling lever 253 toward head end 52 of chair bed 50and moving foot pump pedal 252 upwardly to the up position. Repeatedpumping of foot pump pedal 252 causes manual pump 118 to pressurize thehydraulic oil so that the hydraulic oil can operate the head and footend lift cylinders 132, 142, as well as head, thigh, and foot sectionpivot cylinders 150, 158, 168, and foot section contracting cylinder176.

Typically, the control valves are moved between various configurationsusing electrical power. Chair bed 50 includes a battery 92 configured toprovide electrical power to operate the control valves when electricalpower is not available from a source outside of chair bed 50. Use offoot pump pedal 186 to pressurize the hydraulic oil and the availabilityof electrical power from battery 92 to operate the control valves allowsa caregiver to manipulate lifting mechanism 130 and articulating deck402 to move chair bed 50 to any desired position within its range ofmovement when there is no electrical power supplied to the chair bed 50.

In addition, depressing CPR foot pedal 250 manually moves head section404 from the back-support position to the down position for performingCPR on a person on sleeping surface 552, and the emergency Trendelenburglever 254 manually activates emergency Trendelenburg valve 214 to movesleeping surface 552 to the Trendelenburg position. Both of the CPR footpedals 250 and the emergency Trendelenburg lever 254 operate to changethe position of chair bed 50 when chair bed 50 is away from a powersource, and both operate without the need to pump foot pump pedal 252.

Lifting Mechanism

Lifting mechanism 130 includes a head end axle 258 rotatably mounted tobrackets 260 that are fixed to sides 66 of base frame 62 as shown inFIGS. 11 and 12. A lever 256, and lift arms 320, 322 are fixed to axle258 and piston rod 134 of head end lift cylinder 132 is coupled to lever256. Foot end 54 of base frame 62 carries levers 214 fixed to brackets212, a foot end cross bar 276 fixed to distal ends 294 of levers 214,and piston rod 144 of foot end lift cylinder 142 coupled to foot endcross bar 276.

Head end connector members 262, 264 couple lift arms 320, 322 tointermediate frame 302. Each connector member 262, 264 has a first end266, 268 that is pivotably connected to lift arms 320, 322. Second ends270, 272 of head end connector members 262, 264 are pivotably coupled tointermediate frame 302. Foot end connector members 282, 284 each have afirst end 286, 288 that is pivotably connected to lift arms 324, 326.Second ends 290, 292 of foot end connector members 262, 264 are fixed tointermediate frame 302.

Head end lift cylinder 132 and foot end lift cylinder 142 are eachpivotably mounted to struts 64 of base frame 62 as shown in FIGS. 11 and15. Piston rod 134 of head end lift cylinder 132 is pivotably coupled todistal end 274 of lever 256. When head end lift cylinder 132 isactivated by supplying pressurized hydraulic oil to interior region 133through rear port 138, the pressurized hydraulic oil pushes piston rod134 so that piston rod 134 slides outwardly to extend from head end liftcylinder 132, pushing distal end 274 of lever 256 toward head end 52 ofchair bed 50 and rotating head end axle 258 so that lift arms 320, 322rotate upwardly. As lift arms 320, 322 rotate upwardly, connectingmembers 262, 264 push head end 52 of intermediate frame 302 upwardlyrelative to base frame 62.

Likewise, piston rod 144 of foot end lift cylinder 142 is pivotablycoupled to foot end cross bar 276. When foot end lift cylinder 142 isactivated by supplying pressurized hydraulic oil to foot end liftcylinder 142 through rear port 146, the pressurized hydraulic oil pushespiston rod 144 so that piston rod 144 slides outwardly to extend fromfoot end lift cylinder 142, pushing cross bar 276 and thus distal ends294 of levers 214 toward foot end 54 of chair bed 50, thereby rotatinglift arms 324, 326 upwardly. As lift arms 324, 326 rotate upwardly,connecting members 282, 284 push foot end 54 of intermediate frame 302upwardly relative to base frame 62.

When chair bed 50 is in the standard bed position with articulating deck402 configured to provide a planar sleeping surface 552, liftingmechanism 130 is in the raised position shown in FIG. 15 having liftcylinders 132, 142 activated and piston rods 134, 144 extendedtherefrom, axle 258 and lift arms 320, 322 rotated upwardly, and crossbar 276 pushed toward foot end 54 of chair bed 50 with lift arms 324,326 rotated upwardly, so that lift arms 320, 322, 324, 326 andconnecting members 262, 264, 282, 284 hold sleeping surface 552 firstdistance 566 above the floor as illustratively shown in FIG. 3. Whenchair bed 50 is in the low position, lifting mechanism 130 is in the lowposition shown in FIG. 12 having lift cylinders 132, 142 deactivated andpiston rods 134, 144 retracted into lift cylinders 132, 142, axle 258and lift arms 320, 322 rotated downwardly, and cross bar 276 pulledtoward head end 52 of chair bed 50 with lift arms 324, 326 rotateddownwardly, so that lift arms 320, 322, 324, 326 and connecting members262, 264, 282, 284 hold sleeping surface 552 second distance 568 abovethe floor as illustratively shown in FIG. 4.

Lifting mechanism 130 can also be used when chair bed 50 is in thesitting position to help a person (not shown) on sleeping surface 552 tostand up. When chair bed 50 is in the sitting position, head section 404of articulating deck 402 is in the back-support position, thigh section408 is in the up position, foot section 410 is in the down position, andintermediate frame 302 is in the low position as shown in FIGS. 2 and 7.Typically, the person on sleeping surface 552 can place their feet (notshown) on the floor when chair bed 50 is in the sitting position. afterthe feet of the person are on the floor, lifting mechanism 130 can bemoved from the low position to the raised position to help the person tostand up. Additionally, chair bed 50 can be provided with grip handles632, 640, described below and shown in FIG. 2, that are mounted to movewith intermediate frame 302 to provide additional support for the personstanding up with the aid of chair bed 50.

Reduced-Shear Pivot

Head section 404 is coupled to weigh frame 506 by reduced-shear pivotassembly 650 shown in FIGS. 11 and 14-17. Reduced-shear pivot assembly650 mounts head section 404 to weigh frame 506 for both translationalmovement and pivoting movement of head section 404 relative to seatsection 406 of deck 402 and relative to weigh frame 506. The pivotingand translational movements combine to produce a motion in which headsection 404 pivots relative to weigh frame 506 about an effective pivotaxis positioned to lie above lower deck 430 and immediately adjacentupper deck 414. The shear between the back of the person and thesleeping surface 552 caused by movement of head section 404 is reduced,thereby reducing scrubbing of the sleeping surface 552 against theperson.

Reduced-shear pivot assembly 650 includes brackets 654 mounted to eachside 656 of head section 404 as shown in FIGS. 11 and 15-17. Brackets654 connect flattened U-shaped struts 658 that span head section 404 tosides 656 as shown in FIG. 1. A lever arm 660 having a cap 662 is fixedto struts 658 and extends longitudinally in a direction parallel to thesides 656 of head section 404 toward foot end 54 of chair bed 50,terminating in a tip 664 as shown best in FIGS. 15-17. Two spacer rods666 each have a first end 668 pivotably coupled to struts 658 adjacentto brackets 654 and a second end 670 pivotably connected to weigh frame506 so that spacer rods 666 pivot about a spacer pivot axis 672. Spacerrods 666 maintain the separation between spacer pivot axis 672 andstruts 658 as head section 404 moves between the back-support positionof FIG. 15 and the down position of FIG. 16.

Slotted brackets 674 are fixed to sides 676 of seat section 406 adjacentto foot end 54 of head section 404 as shown in FIGS. 15-17. Each slottedbracket 674 is formed to include a horizontal longitudinal slot 678.Foot end 54 of head section 404 includes pins 680 that are received byslots 678. Pins 680 and slots 678 cooperate to guide the movement offoot end 54 of head section 404 so that foot end 54 of head section 404translates horizontally or longitudinally toward head end 52 of chairbed 50 when head section 404 pivots upwardly to the back-supportposition.

Head section pivot cylinder 150 operates to pivot head section 404between the down position and the back-support position as shown inFIGS. 11 and 15-17. A bracket 682 having a distal end 684 is fixed to anupper deck end portion 460 of thigh section 408. Bracket 682 isgenerally centrally located along weigh frame end portion 460. Headsection pivot cylinder 150 is pivotably coupled to distal end 684 ofbracket 682 and piston rod 152 of head section pivot cylinder 150 ispivotably coupled to tip 664 of lever arm 660 so that head section pivotcylinder 138 and lever arm 660 act between struts 658 of head section404 and weigh frame 506.

When head section 404 is in the down position shown, for example, inFIG. 16, head end pivot cylinder 150 is in a deactivated configurationhaving piston rod 152 in the retracted position. Head section 404 andlever arm 660 are generally parallel to weigh frame 506 when headsection 404 is in the down position.

When head end pivot cylinder 150 moves to the extended position, pistonrod 152 pushes tip 664 of lever arm 660 toward head end 52 of chair bed50. Lever arm 660 pushes against struts 658 to pivot head section 404upwardly to the back-support position as shown in FIG. 17. Pins 680cooperate with slots 678 so that foot end 54 of head section 404 moveslongitudinally toward head end 52 of chair bed 50 a distance 686. At thesame time, spacer rods 666 swing upwardly forcing head section 404 toengage in the motion illustratively shown by arc 688 in FIG. 17combining the pivoting movement of head section 404 and the translatingmovement of head section 404 to provide the reduced-shear pivot. Sincepivot pins 680 are located immediately adjacent the top of side walls438 of step deck 412, the pivot is between sleeping surface 552 andbottom 586 of mattress 550. This reduces the travel required to reduceshear between the person (not shown) and sleeping surface 552.

The longitudinal displacement of the pivot is selected to prevent acrease in mattress 550 between head and seat portions 558, 560. Theeffective point of contact on mattress back portion 558 extends as itpivots upwardly as does the corresponding point on the person onsleeping surface 552 as the person pivots about his or her hip. As aresult of the reduced-shear pivot assembly 650, the point of contact onmattress back portion 558 and the corresponding point on the person movetogether, thus reducing the sliding of the person relative to sleepingsurface 552.

Although the surface of the person's back expands when the person pivotsupwardly to a sitting position, the surface of the back legs of theperson contract as the back legs pivot downwardly. As will be explainedwith respect to FIGS. 24-28 and 30, foot section 410 of deck 402 andfoot portion 564 of mattress 550 are mounted and constructed to shortenin length and mattress 550 thins and shortens in length when pivoting tothe sitting position to effect a reduced-shear pivot.

Chair bed 50 can be provided with hip pivot guide 694 shown in FIGS.31-33 to help the caregiver accurately position the hip (not shown) ofthe person (not shown) on sleeping surface 552. Hip pivot guide 694indicates the position of the hip of the person that will minimize thedistance between effective pivot axis and the axis (not shown) aboutwhich the person's hip pivots, thereby maximizing the effectiveness ofthe reduced-shear pivot. Caregivers providing care to people usingconventional beds having movable head sections typically attempt toplace the hip of the person at the pivot joint of the head section tothe bed. Typically, the only available method for the caregiver toestimate this placement is by viewing the distance between the top ofthe person's head and the head end of the mattress. Providing hip pivotguide 694 on body section siderails 804, 806 of chair bed 50 maximizesthe ability of the caregiver to properly locate the hip of the person onsleeping surface 552.

A reduced-shear pivot assembly 714 is shown included on an examinationtable 700 having a head end 702, a foot end 704, and an articulatingdeck 706, including a head section 708, a seat section 710, and a footsection 712 as shown in FIGS. 18-23. Examination table 700 isconvertible between an examination position having deck 6 in a generallyplanar configuration as shown in FIGS. 18, 20 and a sitting position asshown in FIGS. 19, 22. Head section 708 moves between a generallyhorizontal down position shown in FIG. 18 and an upward back-supportposition shown in FIG. 19, and foot section 712 moves between agenerally horizontal up position shown in FIG. 18 and a generallyvertically downwardly extending down position shown in FIG. 19.

Head section 708 and foot section 712 are both provided with a reducedshear pivot assembly 714, shown best in FIGS. 20-23, that operates topivot head section 708 relative to seat section 710 about an effectivepivot axis 720 that is positioned to lie above an examination or supportsurface 722 and that also operates to pivot foot section 712 relative toseat section 710 about an effective pivot axis 778 that is positioned tolie above examination or support surface 722.

Although the reduced shear pivot assembly 714 is described with respectto an examination table, it can also be used in a bed, a chair bed, astretcher, a gurney or any other device having an articulated deckincluding one or more articulated deck sections wherein the pivotcorresponds to the pivoting of a person on the deck.

Examination table 700 includes a base platform 724 having upstandingposts 726 fixed thereto and extending upwardly therefrom. The upstandingposts 726 are secured to the base 724 by diagonal braces 725. The baseplatform 724 is shown resting on the ground. Wheels 723 are provided atthe back end of the base 724 displaced from the ground when the base 724is in its horizontal position. To move the table, the table is rotatedup such that the base 724 pivots back onto the wheel 723. Then, thetable can be moved to any desired location. This movement is preferablewhen in the chair position of FIG. 19 with an occupant therein. It isnot recommended to transport of the table in its supine position of FIG.18 on wheel 723 with an occupant thereon. Alternatively, wheels may beprovided at the four ends of the base 724 so as to make the tableportable without tilting. This will allow the table to be used as agurney in an emergency department wherein the patient is brought in fromthe ambulance, moved into an emergency bay, then moved out to a room orsurgery center without moving from one conveyance to another.

Reduced-shear pivot assembly 714 includes a frame 716 pivotably attachedto a pair of spaced upstanding posts 726 for pivoting movement relativethereto about a pivot axis 718. A drive motor 728 is pivotably attachedto base platform 724 by bracket 727 for pivoting movement about a pivotaxis 780. Drive motor 728 is configured to rotatably drive a lead screw730 that angles upwardly from drive motor 728 to a sheath 732 that iscoupled to frame 716 for pivoting movement about a pivot axis 734.

Sheath 732 is formed to include an interior region (not shown) thatthreadably receives lead screw 730 as shown in FIG. 20. Extension oflead screw 730 from sheath 732 by rotating causes frame 716 to pivotrelative to base platform 724 about pivot axis 718 with foot end 704 offrame 716 pivoting upwardly and head end 702 of frame 716 pivotingdownwardly. Likewise, retraction of lead screw 730 into sheath 732 causeframe 716 to pivot about pivot axis 718 with foot end 704 of frame 716pivoting downwardly and head end 702 of frame 716 pivoting upwardly.

Head section 708 of articulating deck 706 is fixed to frame 716 byflanges 717 as shown in FIGS. 20-23. As frame 716 pivots from agenerally horizontal initial position shown in FIG. 20 to an inclinedposition shown in FIG. 22 having head end 702 of frame 716 positionedabove foot end 704 of frame 716, head section 708 pivots from agenerally horizontal down position of FIG. 18 to an upward back-supportposition of FIG. 19.

The head end of seat section 710 is connected to upstanding posts 726 bytransverse upper struts 740, transverse lower struts 742, and bracket746. Bracket 746 includes a first end 748 fixed to head end of seatsection 710 and extends downward to terminate at a second end 750. Eachupper strut 740 has a first end 752 pivotably coupled to seat section710 adjacent to first end 748 of bracket 746 and a second end 754pivotably coupled to one of upstanding posts 726. Each lower strut 742has a first end 756 pivotably coupled to second end 750 of bracket 746and a second end 758 pivotably coupled to one of upstanding posts 726beneath second end 754 of upper strut 740.

As can best be seen in FIGS. 20 and 22, the connection of the struts 740and 742 at ends 754 and 758 respectfully to the upstanding post 726 areoffset with respect to a vertical. The connection of the strut 740 and742 at ends 752 and 756 to the bracket 746 are aligned vertically. Thelengths of the struts 740 and 742 are substantially equal. As analternative, the strut 740 and 742 may be of unequal length and theirconnection to the outstanding post 26 may be aligned vertically. As afurther alternative, the connections may be offset and the strutslengths different. The lengths of the struts 740 and 742 and theirconnections to the upstanding posts 726 and to the bracket 726 areselected such that the seat section 710 is horizontal in the planar orhorizontal position of the articulate deck 6 as shown in FIGS. 18 and 20and the foot end of seat section 710 is raised with respect to the headend of seat section 710 in the chair position as illustrated in FIGS. 19and 22. Thus, the struts 740, 742 do not form a true parallelogram withthe upstanding post 726 and bracket 746. The raising of the knee withrespect to the hip secures the occupant to the chair and preventssliding out.

First telescoping members 744 are slidably received by a sheath 760appended to head section 708 and flange 717 of frame 716 as shown bestin FIG. 23 for movement over rollers 762 between a retracted positionshown in FIGS. 20 and 23, and an extended position shown in FIGS. 21 and22. Each first telescoping member 744 includes a foot end 764 that ispivotably coupled to seat section 710 adjacent to first end 748 ofbracket 746 and a head end (not shown) received by sheath 760. As firsttelescoping members 744 move between the retracted position and theextended position, seat section and head section translates relative toeach other. Thus, the pivot point 764 of the seat and head sectionsmoves alone a plane parallel to the frame 716.

Foot section 712 is pivotably coupled at head end 702 of foot section712 to second telescoping members 766 at 776 as shown in FIGS. 20-22.Seat section 770 is formed to include sheaths 770 and each secondtelescoping member 766 is slidably received by a sheath 770 of the seatsection 710 for movement over rollers 768 between an extended positionshown in FIG. 20 and a retracted position shown in FIG. 22. As secondtelescoping members 766 move between the retracted position and theextended position, foot section 712 translates relative to seat section710. Thus, the pivotal connection of the foot section 712 to the seatsection 710 moves in a plane parallel to the seat section transfers tothe plane of the frame 716. A link 782 is pivotably connected at a firstend 784 to frame 716 and at a second end 786 to a bracket 788 extendingfrom foot section 720 pivoting of the frame 716 pivots the foot section712.

A cable 772 has a first end 776 fixed to head end of foot section 712and a second end 774 fixed to flange 717 of head section 708. The lengthof cable 772 is fixed so that second telescoping members 766 move fromthe extended position to the retracted position when first telescopingmembers 744 move from the retracted position to the extended position.Consequently, cable 772, frame 716 and link 782 act to coordinate themovement of head section 708 and foot section 712 relative to seatsection 710 so that as head section 708 translates and pivots upwardlyrelative to seat section 710, foot section 712 simultaneously translatesand pivots downwardly relative to seat section 710.

Seat section 710 translates relative to head section 708 as head section708 pivots from the down position to the back-support position as shownin FIGS. 19-22. The pivoting movement of head section 708 and thetranslational movement of seat section 710 combine to produce a motionin which head section 8 pivots relative to seat section 710 abouteffective pivot axis 720 positioned to lie above support surface 722 andcoincident with a hip (not shown) of a person on the support surface722.

Likewise, seat section 710 translates relative to foot section 712 asfoot section 712 pivots from the up position to the down position asshown in FIGS. 19-22. The pivoting movement of foot section 712 and thetranslational movement of seat section 710 combine to produce a motionin which foot section 712 pivots relative to seat section 710 about asecond effective pivot axis 778 positioned to lie above support surface722 and coincident with a knee (not shown) of a person (not shown) onsupport surface 722.

The head section 708 is fixed to the frame 716 which pivots about afixed pivot point 718 adjacent the foot end of head section 708 fixed tothe base platform 724 and the seat section 710 moves relative to thehead section 722 and frame 716. Thus, when the frame 716 pivots from theplanar position of FIG. 18 to the sixty degree position of FIG. 19, theseat 722 is moved closer to the ground. This allows easy egress.

As can be seen both in bed/chair 50 and table 700, head section 404, 708translates relative to seat section 406, 710 when head section 404, 708pivots from the down position to the back-support position. Thisrelative translation effectively expands the length of deck 402, 706 andsupport surface 552, 722 at the junction of the head and seat sections404, 708 and 406, 710, during the articulation of deck 402, 706. Theeffective expansion of deck 402, 706 and support surface 552, 722 at theseat and head juncture conforms to the lengthening of the back of theperson to minimize the shear that could take place between the personand surface 552, 722. For the foot-seat juncture, the surface 552, 722contracts when moving from a lying position to a sitting position whichcorresponds to the concentration of the back of the legs.

In other words, the expansion of deck 402, 706 and surface 552, 722 atthe back and contraction of the foot allows the lower body of the personto remain stationary relative to surface 552, 722 when tilting the upperbody of the person, which also remains stationary relative to surface552, 722, in order to minimize the scrubbing between the person andsurface 552, 722 during articulation of deck 402, 706.

Thus, the translational movement of seat section 710 of examinationtable 700 illustratively shown in FIGS. 18-23 relative to head and footsections 708, 712 and contemporaneous with the pivoting movement of headand foot sections 708, 712 results in a reduced-shear pivoting movementof head and foot sections 708, 712. The effective pivot axes 720, 778 ofhead and foot sections 708, 712 to lie above support surface 722. Ifeffective pivot axes 720, 778 are approximately co-linear with axis ofrotation of hip and knee respectively, then the scrubbing of supportsurface 722 against the person (not shown) supported by support surface722 will be minimized.

As can be seen in both chair bed 50 and examination table 700, headsection 404, 708 translates relative to seat section 406, 710 when headsection 404, 708 pivots from the down position to the back-supportposition. This relative translation effectively expands the length ofdeck 402, 706 at the junction of the back and seat during thearticulation of deck 402, 706. When the upwardly-facing person (notshown) supported by surface 552, 722 moves from a lying position to asitting position, the back (not shown) of the person lengthen. Theeffective expansion of deck 402, 706 at the juncture of seat section406, 710 and head section 404, 708 and the consequent expansion ofsurface 552, 722 conforms to the lengthening of the back of the personto reduce the shear that could take place between the person and surface552, 722. For the foot-seat juncture, surface 552, 722 contracts whenmoving from a lying position to a sitting position.

In other words, the expansion of deck 402, 706 and surface 552, 722 atthe back and contraction at the foot allows the lower body of the personto remain stationary relative to surface 552, 722 when tilting the upperbody of the person, which also remains stationary relative to surface552, 722, in order to minimize the scrubbing between the person andsurface 552, 722 during articulation of deck 402, 706. The reduced-shearpivot also minimizes the migration of the person on sleeping surface 552toward foot end 54 of chair bed 50 as head section 404 is repeatedlyraised and lowered and minimizes “bunching” of mattress 550 and thepotential corresponding pressure on the hip and shoulder of the person.

CPR Foot Pedal

CPR foot pedals 250 are coupled to hydraulic system module 100 as shownin FIGS. 11 and 12 and are positioned to be operable by the foot of thecaregiver. As described above, hydraulic system module 100 includes CPRvalve 212 shown in FIG. 13 that can be activated to restore fluidcommunication between rear port 154 of head section pivot cylinder 150and return conduit 185 so that hydraulic oil can be released fromcylinder 150 and head section 404 can move from the back-supportposition to the down position. CPR foot pedals 250 are movable betweenan up position and a downward releasing position. When CPR foot pedals250 are in the releasing position, CPR valve 212 is activated and headsection 404 moves from the back-support position to the down position.

CPR foot pedals 250 and CPR valve 212 are configured so that CPR footpedals 250 can be moved from the releasing position to the up positionwhen head section 404 is in an intermediate position after head section404 has moved away from the back-support position but before headsection 404 has reached the down position. CPR valve 212 can thus bedeactivated when head section 404 is in the intermediate position toblock the fluid communication between rear port 154 of head sectionpivot cylinder 150 and return conduit 185. Blocking the fluidcommunication locks head section 404 in the intermediate position. CPRfoot pedals 250 can thereafter be moved back to the releasing positionso that CPR valve is once again activated to restore fluid communicationbetween rear port 154 and return conduit 185 allowing movement of headsection 404 toward the down position. Providing this capability to thecaregiver in an actuator designed as a foot pedal keeps the hands of thecaregiver free to conduct other activities while CPR foot pedals 250 aredepressed and head section 404 moves to the down position.

Thigh Section

The first embodiment of a chair bed 50 in accordance with the presentinvention additionally includes thigh section 408 of articulating deck402 which is configured to pivot relative to weigh frame 506 as shown inFIG. 15. Thigh section 408 pivots about a pivot axis 602 adjacent tohead end 52 of thigh section 408 between a down position in which thighsection 408 is generally horizontal and parallel to weigh frame 506 andan upward position in which foot end 54 of thigh section 408 is elevatedabove weigh frame 506. Thigh section pivot cylinder 158 is connected toweigh frame 506 as shown in FIGS. 14 and 15. Although thigh section 408can move independently of the head and foot sections 404, 410, thighsection 408 preferably moves to the upward position when head section404 moves to the back-support position so that the head and thighsections 404, 408 cooperate to cradle the person (not shown) on sleepingsurface 552 therebetween. Thigh section 408 preferably moves to the downposition when head section 404 moves to the down position.

Foot Section

Foot section 410 of articulating deck 402 is movable from a generallyhorizontal up position parallel to intermediate frame 302 as shown inFIGS. 1 and 3 to a generally vertically downwardly extending downposition to permit the lower legs and feet of the person (not shown) tobe lowered to the sitting position as shown in FIGS. 2 and 8. Footsection 410 can also be contracted from an expanded position having alongitudinal length 465 as shown in FIGS. 3, 24, and 30 to a contractedposition having foot end 54 of foot section 410 drawn inwardly towardhead end 52 of chair bed 50 so that foot section 410 has a longitudinallength 464 that will “clear” the floor when foot section 410 moves tothe down position as shown in FIGS. 8 and 25. Preferably, length 464 offoot section 410 when foot section 410 is contracted is such that footend 54 of foot section 410 clears the floor and is spaced-aparttherefrom sufficiently to permit a base (not shown) of an over bed table(not shown) to fit therebetween.

Foot section 410 is pivotably coupled to an upper deck end portion 460of thigh section 408 by hinge 468 as shown in FIGS. 12, 15, 24, 25, and30. Consequently, foot section 410, when in the down position, can belonger by an amount equal to a vertical offset 514 between lower deck430 and upper deck 414 than it could be if there were no step deck 412,and foot section 410 were instead connected to lower deck 430. Thus, forfoot section 410 to clear the floor when foot section 410 pivots fromthe up position to the down position, foot section 410 can contract alesser amount than would be required if there were no step deck 412.

Foot section 410 includes a pivoting member 466 that is pivotablycoupled to thigh section 408 and a contracting member 462 that can bedrawn inwardly toward head end 52 of foot section 410 from an expandedposition to the contracted position. Foot section pivot cylinder 168 andfoot section contracting cylinder 176 cooperate to move pivoting member466 between the up position and the down position and to movecontracting member 462 between the expanded position shown in FIG. 24and the contracted position shown in FIG. 25.

Contracting member 462 is positioned to slide across top surface 470 ofpivoting member 466 as shown in FIGS. 11 and 15. A folding bracket 472has a first end 474 pivotably coupled to weigh frame 506 and a secondend 476 pivotably coupled to pivoting member 466 as shown in FIGS. 15,24, and 25. Piston rod 170 of foot section pivot cylinder 168 ispivotably coupled to bracket 472. Piston rod 170 pushes against bracket472 as piston rod 170 extends from foot section pivot cylinder 168causing bracket 472 to pivot upwardly from a folded position about apivot axis 478 adjacent to weigh frame 506 and to push pivoting member466 upwardly to the up position. When piston rod 170 is in the extendedposition, bracket 472 is generally unfolded, horizontal, and parallel topivoting member 466.

Foot section 410 further includes first and second linkages 480, 482 anda thruster strut 484 as shown in FIGS. 24 and 25. First linkage 480 hasa first end 486 pivotably coupled to pivoting member 466. A second end488 of first linkage 480 is pivotably coupled to a first end 490 ofsecond linkage 482 and a second end 492 of second linkage 482 ispivotably coupled to foot end 54 of contracting member 462. Thus, firstand second linkages 480, 482 couple pivoting member 466 and contractingmember 462.

Thruster strut 484 has a first end 494 that is pivotably coupled topivoting member 466 and a second end 496 that is pivotably coupled tosecond linkage 482 between the first and second ends 490, 492 of secondlinkage 482 as shown in FIGS. 24 and 25. Foot section contractingcylinder 176 is pivotably coupled to pivoting member 466 near head end52 of pivoting member 466 and piston rod 178 is pivotably coupled tothruster strut 484 between the first and second ends 494, 496 ofthruster strut 484. First and second linkages 480, 482, thruster strut484, and foot section contracting cylinder 176 are generally coplanarand generally operate in a plane that is parallel to foot section 410.

As piston rod 178 moves from the retracted position, shown in FIG. 25,to the extended position, shown in FIG. 24, thruster strut 484 pivotsabout a pivot axis 498 so that second end 496 of thruster strut 484swings toward foot end 54 of chair bed 50. As thruster strut 484 swingstoward foot end 54 of chair bed 50, second linkage 482 is pushed bythruster strut 484 toward foot end 54 of chair bed 50 and second linkage482 pulls second end 488 of first linkage 480 toward foot end 54 ofchair bed 50.

Second end 492 of second linkage 482 pushes contracting member 462toward foot end 54 of chair bed 50 when thruster strut 484 pushes secondlinkage 482 toward foot end 54 of chair bed 50 as shown in FIGS. 24 and25. Likewise, when piston rod 178 moves from the extended position shownin FIG. 24 to the retracted position shown in FIG. 25, thruster strut484 pulls second linkage 482 toward head end 52 of chair bed 50 andsecond linkage 482 pulls foot end 54 of contracting member 462 towardhead end 52 of chair bed 50, causing contracting member 462 to contractand reducing the length of foot section 410 by a distance 500 as shownin FIG. 25.

Contracting member 462 is formed to include downwardly extendinglongitudinal tabs 502 and pivoting member is formed to includelongitudinal channels 504 as shown in FIGS. 24-27. Longitudinal tabs 502are received by longitudinal channels 504 as shown best in FIGS. 26 and27. Tabs 502 cooperate with channels 504 to maintain the transverseposition of contracting member 462 relative to pivoting member 466 ascontracting member 462 slides longitudinally relative to pivoting member466.

As foot section 410 pivots from the up position to the down position,inflatable foot portion 564 of mattress 550 deflates as shown in FIG. 30and shown diagrammatically in FIG. 8 so that foot section 410 ofarticulating deck 402 can move to the down position without interferencefrom foot portion 564 of mattress 550. Deflating foot portion 564 alsoallows the person (not shown) carried by chair bed 50 to sit on chairbed 50 when chair bed 50 moves to the sitting position without havingthe thickness of foot portion 564 of mattress 550 pull the knees andshins of the person forward as foot section 410 of articulating deck 402pivots to the down position. In addition, the deflating action ofdeflating foot portion 564 prevents scrubbing between sleeping surface552 and the legs (not shown) of the person (not shown) on sleepingsurface 552 by allowing sleeping surface 552 adjacent foot portion 564to move with the legs of the person.

A second embodiment of a contracting mechanism 520 for expanding andcontracting the length of foot section 410 can illustratively beoperated using an air control system 522 that also operates to inflateand deflate foot portion 564 of mattress 550 as shown in FIG. 25 a. Aircontrol system 522 includes an air supply 524 for supplying pressurizedair and a controller 526 for controlling the flow of air through conduit528 to inflatable foot portion 564 and to contracting mechanism 520.

Contracting mechanism 520 includes a bellows 530 that is receivedbetween a first wall 534 that is fixed to pivoting member 466 and asecond wall 536 that is fixed to contracting member 462 as shown in FIG.25 a. Contracting member 462 is slidably connected to pivoting member sothat second wall 536 can slide relative to first wall 534. As secondwall 536 moves toward first wall 534, contracting member is drawninwardly to contract foot section 410. As second wall is pushed awayfrom first wall 534, contracting member extends from foot section 410and expands the length of foot section 410. Contracting mechanism 520also includes two extension springs 538 connected to pivoting member 466and contracting member 462 to yieldably bias contracting member 462 tothe contracted position.

As air control system 522 supplies pressurized air to bellows 530,bellows expands and pushes against first and second walls 534, 536moving second wall 536 away from first wall 534 and causing contractingmember to extend from foot section 410 thereby expanding the length offoot section 410. As air control system 522 withdraws air from bellows530, bellows stops pushing against first and second walls 534, 536, andsprings 538 pull contracting member 462 inwardly toward pivoting member466, thus contracting the length of foot section 410.

As described above, illustrative air control system 522 operate tocontrol both the inflation of foot portion 564 and the inflation ofbellows 530 as shown in FIG. 25 a. The illustrative system provides asatisfactory method for coordinating the inflation and deflation of footportion 564 with the contraction and expansion of the length of footsection 410.

Step Deck and Mattress

The head, seat, thigh, and foot sections 404, 406, 408, 410 ofarticulating deck 402 cooperate to define a step deck 412 as shown bestin FIGS. 11, and 28-30. Step deck 412 includes an upper deck 414 havinga head end upper deck portion 416 appended to head end 52 of headsection 404, side upper deck portions 418, 420, 422, 424, 426, 428appended to sides of the head, seat, and thigh sections 404, 406, 408,and a foot end upper deck portion 460 appended to foot end 54 of weighframe 506 adjacent to thigh section 408. The upper deck portions 416,418, 420, 422, 424, 426, 428, 460 and a top surface 411 of foot section410 are coplanar when articulating deck 402 is in the initial positionand cooperate to form upper deck 414 which is generally parallel toweigh frame 506.

Step deck 412 also includes a lower deck 430 having a head slat 432, aseat slat 434, and a thigh slat 436. Head, seat, and thigh slats 432,434, 436, are coplanar when articulating deck 402 is in the initialposition and they cooperate to form lower deck 430 which is generallyparallel to weigh frame 506 and to upper deck 414 when articulating deck402 is in the initial position.

Lower deck 430 is connected to upper deck 414 by a wall 438 including ahead end wall 440 connecting head slat 432 to head end upper deckportion 416, side walls 442, 444, 446, 448, 450, 452 connecting head,seat, and thigh slats 432, 434, 436 to side upper deck portions 418,420, 422, 424, 426, 428, and a foot end wall 454 connecting thigh slat436 to foot end upper deck portion 460 as shown in FIGS. 11 and 28. Stepdeck 412, then, comprises upper deck 414 and is formed to include acentral, longitudinally extending recess 456 defined by lower deck 430and by wall 438 connecting lower deck 430 to upper deck 414. In thepreferred embodiment, foot section 410 of step deck 412 is displacedfrom recess 456 and forms part of upper deck 414, as shown in FIGS. 28and 30.

In preferred embodiments, head section 404 of articulating deck 402 iscoupled to weigh frame 506 by reduced-shear pivot assembly 650immediately adjacent upper deck 414 which causes head section 404 ofarticulating deck 402 to pivot relative to weigh frame 506 between thedown position and the back-support position.

Combining step deck 412 and reduced-shear pivot assembly 650 in chairbed 50 allows reduced-shear pivot assembly 650 to be mounted to wall 438rather than to a bottom of a conventional deck. Consequently, thevertical distance between sleeping surface 552 and reduced-shear pivotassembly 650 is minimized. This minimizing the extent that reduced-shearpivot assembly 650 is required to raise effective pivot axis abovereduced-shear pivot assembly 650.

Mattress 550 is received by articulating deck 402 and includes aprojection 576 sized to be received by recess 456 as shown in FIGS. 28and 29. Consequently, mattress 550 is thinner along sides 580 ofmattress 550 where mattress 550 engages upper deck 414 of step deck 412.Conversely, mattress 550 is thicker in portions adjacent to projection576. Preferably, projection 576 is positioned directly beneath portionsof mattress 550 carrying a majority of the weight of the person onsleeping surface 552. The thick portion of mattress 550 including thethickness of mattress 550 between sleeping surface 552 and a bottomsurface 586 engaging upper deck 414 plus the thickness of projection 576provides greater comfort for the person on sleeping surface 552.Mattress 550, then, has a thinner perimetral zone 580 and a thickerbody-support zone 582 adjacent to projection 576. Preferably, bodysupport zone is 1½ times the thickness of perimetral zone 580. Forexample, perimetral zone can be 5 inches (12.7 cm) thick andbody-support zone 582 can be 7½ inches (19 cm) thick.

Thinner perimetral zone 580 and upper deck side portions 417 cooperateto define “rammed” edges that provide greater firmness around the edgesof sleeping surface 552 as the result of sleeping surface 552 being inclose proximity to upper deck 414. This increased firmness isadvantageous when the person enters and exits the bed along the sides ofthe bed.

Additionally, the rammed edges provide a firm edge that cooperates withsiderail assemblies 800, 802, 804, 806 to minimize the potential forsiderail entrapment, in which an object becomes wedged between sleepingsurface 552 and one of siderails 808, 810, 812, 814. Also, step deck 412cooperates with siderail assemblies 800, 802, 804, 806 to maximize theheight relative to sleeping surface 552 at which siderails 808, 810,812, 814 are mounted as shown in FIGS. 34 and 35. Tops of siderails 808,810, 812, 814 can be higher when in the patient-restraining position forimproved coverage and protection of the person (not shown) on sleepingsurface 552 and bottoms 814 can be higher when in the tucked positionfor improved access to base frame 62 and to the space beneathintermediate frame 302.

Projection 576 includes a side wall 584 that can be configured to engageat least portions of the wall 438 of step deck 412 as shown in FIG. 29,thereby preventing lateral and longitudinal sliding of mattress 550relative to step deck 412. Also, mattress 550 includes sides 578connecting sleeping surface 552 and bottom surface 586. Mattress 550 andstep deck 412 are configured so that sides 578 of mattress 550 areexposed above deck 402 as shown in FIGS. 28 and 29 providing thecaregiver greater and easier access to mattress 550, rather thanengaging a portion of a frame or upstanding walls of a deck as is foundwith conventional mattress and deck systems.

In preferred embodiments, sleeping surface 550 is generally planar andprojection 576 is centrally located beneath sleeping surface 550 to formthick body support zone 582 of mattress 550 surrounded by perimetralzone 580 engaging upper deck 414. Mattress 550 may be provided in morethan one piece, for example, mattress 550 may comprise a first mattresspiece fit into recess 456 and a second mattress piece surrounding andabutting sides of the first mattress piece and engaging upper deck 414,or a first mattress piece could fit into recess 456 and a secondmattress piece having a planar bottom surface could fit over the firstmattress piece so that the bottom of the second mattress piece engagesthe first mattress piece and upper deck 414. However, a one-piecemattress 550 including both body-support zone 582 and perimetral zone580 is preferred.

Inflatable Mattress Portion—Minimizing the Foot Section

Additionally, mattress 550 can include an inflatable portion 574 thatcan assume both an inflated position and a deflated position.Preferably, inflatable portion 574 is positioned to lie in foot portion564 as shown in FIG. 30 so that inflatable portion 574 can be inflatedto serve as sleeping surface 552 when foot section 410 of deck 402 is inthe up position and so that inflatable portion 574 can be deflated andinclined downwardly when the foot section 410 is lowered to the downposition to provide room for the lower legs of the person when chair bed50 is in the sitting position. Foot portion 564 is thinner and shorterwhen deflated than when foot portion 564 is inflated.

Foot portion 564 of mattress 550 and foot section 410 of articulatingdeck 402 cooperate to minimize the length of the foot of chair bed 50 asshown in FIG. 30. Foot section 410 and foot portion 564 are a firstlength 465 when foot section 410 is in the up position and a secondlength 464 when foot section 410 is in the down position, first length465 being greater than second length 464. Also, foot portion 564 is afirst thickness 608 when foot section 410 is in the up position and asecond thickness 609 when foot section 410 is in the down position,first thickness 608 being greater than second thickness 609.

In addition, the width 604 of foot portion 564 of mattress 550 is lessthan the width 606 of head portion 558 of mattress 550, the width 606 ofhead portion 558 typically being a standard mattress width as shown inFIGS. 28 and 30. This difference between the widths 604, 606 permits astandard fitted sheet (not shown) to be tightly installed onto mattress550 while remaining loose adjacent to foot portion 564 so that pressurerelief can be maintained in the section of foot portion 564 receivingthe heels (not shown) of the person (not shown) supported on sleepingsurface 552. The smaller width 604 of foot portion 564, the contractionof foot section 410 and the corresponding contraction of foot portion564, and the deflation of inflatable portion 574 when inflatable portion574 is positioned to lie in foot portion 564, all act to minimize thefoot of chair bed 50 when the foot section 410 moves from the upposition to the down position so that the feet of the person supportedon the sleeping surface 552 can reach the floor (not shown) or foot prop646. The narrow foot section 410 of deck 402 and foot portion 564 ofmattress 550 minimizes the width of foot end 54 of deck 402 so that thewidth of bed 50 adjacent to extended frame 610 is no greater than thewidth of bed 50 adjacent to body section siderails 812, 814.

C-Arm Access

Use of step deck 412 can additionally improve access of equipment toportions of chair bed 50 as shown in FIG. 29. A C-arm 588 carryingequipment 590, 592 and having equipment 590 positioned to lie abovesleeping surface 552 and equipment 592 positioned to lie below step deck412 can be positioned near chair bed 50. C-arm 588 is C-shaped having aninner surface 594 and a point 596 on inner surface 594 that is themaximum lateral distance on inner surface 594 away from equipment 590,592. An edge 598 of upper deck 414 is positioned to lie a distance 600above lower deck 430 of step deck 412. While a conventional deck bottom(not shown) would have an edge (not shown) engaging C-arm 588 away frompoint 596, edge 598 of step deck 412 engages C-arm adjacent to point596, thereby maximizing the area of sleeping surface 552 across whichequipment 590, 592 can be located.

Additionally, head slat 432 can have a radiolucent portion 510 made froma radiolucent material that is transparent to X-rays thereby permittingX-rays to pass therethrough as shown in FIGS. 28 and 29. Equipment 590,592 can be radiography equipment used to produce images such as X-rayimages or photographs of the person (not shown) on sleeping surface 552.Having step deck 412 arranged to engage point 596 of C-arm 588 maximizesthe area of sleeping surface 552 away from edge 598 that equipment 590,592 can be positioned, thereby maximizing the area of sleeping surface552 on which the person can be positioned to lie while fluoroscopicprocedures are performed on the person.

Extended Frame

An extended frame module 610 can be provided for chair bed 50. Extendedframe module 610 includes an extended frame 612 at foot end 54 of chairbed 50 as shown in FIG. 11. Extended frame 612 comprises frame-extendermembers 614, each frame-extender member 614 having a first end 616 fixedto foot end 54 of weigh frame 506 on each side of chair bed 50.Frame-extender members 614 each extend outwardly away from head end 52of chair bed 50 and terminate in a second end 618 positioned to lielongitudinally between thigh section 408 and foot end 54 of foot section410 and along sides 508 of foot section 410.

Extended frame 612 further comprises swing members 620, each swingmember 620 having a first end 624 pivotably coupled to second end 618 offrame-extender members 614. Swing members 620 can swing between a tuckedposition beside frame-extender members 614 and an extended positionbeside foot section 410 of articulating deck 402 as shown in FIG. 2.Each swing member 620 is preferably provided with a foot safety switch648 as shown in FIGS. 2 and 11 to prevent entrapment of objects underswing members 620 during movement of intermediate frame 302.

Extended frame 612 additionally comprises a foot gate 622 includingswinging gates 626, 634, each swinging gate 626, 634 having a first end628, 636 rotatably coupled to swing members 620 as shown in FIG. 11.Gates 626, 634 can rotate a full 360 degrees relative to swing members620. Gates 626, 634 cooperate with swing members 620 to move gates 626,634 to several positions relative to weigh frame 506. For example, gates626, 634 can “close” foot end 54 of chair bed 50 as shown in FIG. 1 bymoving to a closed position in which gates 626, 634 are positioned tolie transversely across foot end 54 of chair bed 50 having second ends630, 638 of gates 626, 634 in juxtaposition. Gates 626, 634 provide aprotective “crib-like” perimeter when gates 626, 634 are closed andchair bed 50 is in the sitting position.

Foot gate 622 can also be moved to a side-grip position shown in FIG. 2by first swinging gates 626, 634 inwardly along arc 642 as shown in FIG.11 so that gates 626, 634 are positioned to lie directly above swingmembers 620 and then swinging swing members 620 along arc 732 so thatswing members 620 and gates 626, 634 are positioned to lie besideframe-extender members 614. Including both fixed frame-extender members614 and swing members 620 in extended frame 612 allows gates 626, 634 toboth close foot end 54 of chair bed 50 while at the same time reducingthe radius through which swing members 620 swing when moving from theclosed position to the side-grip position. As a result, the spacerequired around chair bed 50 to permit the movement of gates 626, 634 isminimized. Gates 626, 634 are provided with grip handles 632, 640 thatprovide support for a person on sleeping surface 552 moving from aseated position to a standing position when chair bed 50 is in thesitting position and foot gate 622 is in the side-grip position as shownin FIG. 2.

Gates 626, 634 perform the function of a conventional footboard whengates 626, 634 are closed and chair bed 50 is in the bed position. Gates626, 634 can swing outwardly from the closed position to an openposition having each gate 626, 634 positioned to lie away from foot end54 of chair bed 50. When gates 626, 634 are in the open position, thecaregiver has clear access to foot section 410 of chair bed 50.Additionally, gates 626, 634 act as support aids for the person (notshown) supported by sleeping surface 552 when the person stands or istransferred to a wheelchair (not shown) or other equipment (not shown)when chair bed 50 is in the sitting position, swing members 620 areextended, and gates 626, 634 are angled back toward the person. Also,gates 626, 634 can be removed entirely from foot end 54 of chair bed 50to clear foot end 54 of chair bed 50 for caregivers and equipment (notshown) when swing members 620 are folded back and gates 626, 634 arefolded back. Safety switches (not shown) can be included to limit thearticulation of deck 402 and intermediate frame 302 when gates 626, 634are in selected positions to prevent limb entrapment between gates 626,634 and either deck 402 or intermediate frame 302.

Typically, extended frame 612 is carried by weigh frame 506. Forembodiments of chair bed 50 that do not include weighing capability,extended frame 612 is carried by the common frame, which typicallyincludes intermediate frame 302 and weigh frame 506 fixed together.Weigh frame 506 and the common frame also carry articulating deck 402.Carrying extended frame 612 on weigh frame 506 or the common framecauses extended frame 612 to move with articulating deck 402 whenintermediate frame 302 is raised and lowered relative to base frame 62.Consequently, extended frame 612 and gates 626, 634 remain stationaryrelative to the person (not shown) supported by sleeping surface 552.For example, when chair bed 50 is in the sitting position and extendedframe 612 is in the side-grip position, intermediate frame 302 can beraised from the low position to the raised position to help the personto stand. Extended frame 612 is stationary relative to sleeping surface552 so that the person can use grip handles 632, 640 for support.

Siderail Assemblies

Chair bed 50 is typically provided with siderail assemblies 800, 802,804, 806 as shown in FIGS. 11 and 31-38 and shown diagrammatically inFIG. 47. Siderail assemblies 800, 802, 804, 806 include head sectionsiderails 808, 810 mounted to head section 404 of articulating deck 402,and body section siderails 812, 814 mounted to weigh frame 506 adjacentto thigh section 408 of deck 402.

Head section siderails 808, 810 are mounted to move with head section404 as head section 404 pivots relative to weigh frame 506 between thedown position and the back-support position as shown in FIGS. 11 and31-33. Body section siderails 812, 814 are mounted to weigh frame 506and do not move relative to weigh frame 506 and seat section 406 whenhead, thigh, and foot sections 404, 408, 410 of articulating deck 402move. Head section siderails 808, 810 are shorter than body sectionsiderails 812, 814 and extend only adjacent head section 404, whereasbody section siderails 812, 814 extend adjacent head and body (seat andthigh) sections 404, 406, 408. Both of the head section and body sectionsiderails 808, 810, 812, 814 are configured to maintain a between-railgap 866 of approximately 2-3 inches as head section 404 moves betweenthe back-support position and the down position.

In addition, having short head section siderails 808, 810 ideallypositions head section siderails 808, 810 to provide support to a person(not shown) entering or exiting chair bed 50 on one of sides 554, 556when appropriate head section siderail 808, 810 is in thepatient-restraining position and body section siderail 812, 814 is inthe tucked position. This configuration allows the person to enter andexit by sitting on sleeping surface 552 while holding head sectionsiderail 808, 810 for support, and pivoting off of or onto sleepingsurface 552 so that the person does not have to “scoot” along sleepingsurface 552. Also, a hip pivot guide 694 on body section siderails 812,814 helps to optimize the positioning of the hip (not shown) of theperson on chair bed 50 after entering chair bed 50 from one of sides554, 556.

Siderails 808, 810, 812, 814, are passive restraint devices mounted onboth sides of chair bed 50 as shown in FIGS. 11, 34, and 35. In theupward patient-restraining position shown in FIGS. 31-34, siderails 808,810, 812, 814 are vertical barriers that can abut sides 554, 556 ofmattress 550 and extending above sleeping surface 552 to restrainmovement of the person past sides 554, 556 of sleeping surface 552.Siderails 808, 810, 812, 814 may also be lowered below sleeping surface552 of mattress 550 to a tucked position shown in phantom in FIG. 35beneath side portions 418, 420, 422, 424, 426, 428 of upper deck 414 topermit the person to move past sides 554, 556 of sleeping surface 552when entering or exiting chair bed 50. Lowering siderails 808, 810, 812,814 also provides the caregiver with clear access to the patient.

Lowering each siderail 808, 810, 812, 814 is accomplished by pullingrelease handle 862 as shown in FIGS. 34 and 35. After pulling releasehandle 862, the caregiver may let go of release handle 862 and allowsiderail 808, 810, 812, 814 to rotate downwardly into the tuckedposition. The rate at which each siderail 808, 810, 812, 814 rotatesdownwardly is preferably controlled by a mechanical damper 868. To raisesiderails 808, 810, 812, 814, the caregiver pulls up on siderails 808,810, 812, 814 until they lock in the patient-restraining position.Siderail assemblies 800, 802, 804, 806 are configured so that siderails808, 810, 812, 814 are generally vertical and generally parallel to thesides of chair bed 50 at all positions between the tucked position andthe patient-restraining position as shown in FIGS. 34 and 35.

Siderail assemblies 800, 802, 804, 806 are of similar construction. Theprinciples discussed below with respect to body section siderailassembly 806 pertains to each siderail assembly 800, 802, 804, 806unless the description herein specifically states otherwise.

Siderail assembly 806 includes body section siderail 814, a siderailmounting mechanism 816, and a mounting bracket 818 connecting mountingmechanism 816 to sides 508 of weigh frame 506 as shown in FIGS. 34 and35. Mounting bracket 818 is positioned to lie beneath upper deck 414 andis attached to weigh frame 506 as shown in FIGS. 34 and 35. Similarly,head section siderail assemblies 800, 802 are connected to walls 442,444 of head section 404, and body siderail assembly 804 is connected toside 508 of weigh frame 506 as shown in FIG. 11.

Mounting bracket 818 includes an upstanding support wall 820 attached towall 508 of weigh frame 506 and outwardly extending walls 822 attachedthereto and attached to weigh frame 506 as shown in FIGS. 34 and 35.Walls 822 of mounting bracket 818 are formed to include upper openings824 and lower openings 826. Siderail mounting mechanism 816 is aparallelogram connecting mechanism that connects siderail 814 tomounting bracket 818 for movement between the patient-restrainingposition and the tucked position while maintaining siderail 814 in agenerally vertical orientation. Siderail mounting mechanism 816 includesthree curved parallel bars 828, 830, 832 having first ends 834, 836,838, and second ends 840, 842, 844. Curved bar 830 is laterallypositioned to lie between curved bars 828, 832 and vertically positionedto lie above curved bars 828, 832. Bracket mounting pins 848 areappended to a first end 836 of curved bar 830 and are rotatably receivedby upper openings 824 of walls 822. Bracket mounting pins 846, 850 areappended to first ends 834, 838 of curved bars 828, 832 and arerotatably received by lower openings 826 of walls 822. Curved bars 828,830, 832 are mounted to pivot relative to weigh frame 506.

Curved bars 828, 830, 832 each include a first section extendingperpendicular to and above upper deck section 428 and a second sectionextending transverse to the first bar section below upper deck section428 when siderail 814 is in the patient-restraining position as shown inFIG. 34. This curved structure in combination with the raised pivotconnection to step deck 412 allows siderail 814 to be raised abovebottom surface 586 of mattress 550 while being immediately adjacentsides 578 with minimum gap.

Siderail 814 is also formed to include upper openings 852 and loweropenings 854 as shown in FIGS. 34 and 35. Siderail mounting pins 858 areappended to second end 842 of curved bar 830 and are received by upperopenings 852 of siderail 814. Siderail mounting pins 856, 860 areappended to second ends 840, 844 of curved bars 828, 832 and arereceived by lower openings 854 of siderail 814. Curved bars 828, 830,832 are mounted to pivot relative to siderail 814. Upper and loweropenings 824, 826 of mounting bracket 818 are spaced apart and upper andlower openings 852, 854 of siderail 814 are spaced apart an equal amountso that curved bars 828, 830, 832 are positioned in parallel relationbetween siderail 814 and mounting bracket 818.

Siderail 814 can thus rotate between an upper patient-restrainingposition abutting side 556 of mattress 550 as shown in FIG. 34 to atucked position beneath section 428 of upper deck 414 shown in FIG. 35(in phantom). Parallel curved bars 828, 830, 832 cooperate with upperand lower openings 824, 826 of mounting bracket 818 and upper and loweropenings 852, 854 of siderail 814 to keep siderail 814 generallyparallel to wall 452 of step deck 412 and generally perpendicular tosleeping surface 552 as siderail 814 rotates between thepatient-restraining position and the tucked position.

Siderail assembly 806 also includes a latching mechanism 870 including arelease handle 862 rotatably mounted to curved bars 828, 832 formovement between a forward latched position shown in FIG. 34 and arearward released position shown in FIG. 34 (in phantom). Latchingmechanism additionally includes links 872 and latches 878, each linkhaving a first end 874 pivotably connected to release handle 862 and asecond end 876 that is pivotably connected to a latch 878. Each latch878 is formed to include a first end 880 that is pivotably connected tocurved bars 828, 832, a second end 882 spaced apart from first end 880,a rod-gripper recess 884 adjacent to second end 882, and aspring-receiving opening 886 spaced apart from both ends 880, 882 oflatch 878.

Tension springs 888 each have a first end 890 connected tospring-receiving openings 886 of latches 878 and a second end 892connected to brackets 894 fixed to curved bars 828, 832 as shown in FIG.34. As release handle 862 is pulled outwardly by the caregiver, releasehandle 862 pulls links 872 outwardly and upwardly which in turn pulllatches 878 upwardly to pivot latches 878 against the bias of springs888.

A rod 896 is connected to walls 822 of mounting bracket 818 and isarranged to be received by rod-gripper recesses 884 when siderail 814 isin the patient-restraining position shown in FIG. 34 so that rod 896 andlatches 878 cooperate to retain siderail 814 in the patient-retrainingposition. When release handle 862 is pulled outwardly, as shown inphantom in FIG. 34, latches 878 disengage from rod 896, thereby allowingsiderail 814 to rotate downwardly as shown in FIG. 35 until siderail 814reaches the tucked position beneath upper deck 414 of articulating deck402, as shown for siderail 808 in FIG. 1 and siderail 814 in FIG. 35 (inphantom).

To raise siderail 814, the caregiver simply lifts siderail 814 to rotatesiderail 814 upwardly to the patient-restraining position. Each latch878 has second end 882 having a camming surface 898 as shown in FIGS. 34and 35 that engages rod 896. As siderail 814 advances toward thepatient-restraining position, camming engagement of camming surfaces 898and rod 896 forces latches 878 to pivot upwardly against the bias ofsprings 888. Latches 878 ride over rod 896 as siderail 814 advances tothe patient-restraining position until rod 896 is adjacent torod-gripper recesses 884. Springs 888 then pull latches 878 downwardlyto capture rod 896 in rod-gripper recesses 884, thereby holding siderail814 in the patient-restraining position.

Siderail 814 cooperates with siderail mounting mechanism 816 to controlthe gap between mattress 550 and siderail 814. Because siderail 814rotates upwardly from the tucked position to the patient-restrainingposition toward side 556 to abut side 556 of mattress 550, a gap thatcould form between mattress 550 and siderail 814 is minimized.Additionally, siderail 814 cooperates with step deck 412 to minimize thedistance between a bottom 864 of siderail 814 and section 428 of upperdeck 414, further maximizing the effectiveness of siderail 814 as apassive restraint. In addition, siderail mounting mechanism 816 providesa one-step release and auto-tuck movement as siderail 814 rotates fromthe patient-restraining position to the tucked position.

Each siderail assembly 800, 802, 804, 806 operates in a manner similarto siderail assembly 806 described above to move siderails 808, 810,812, 814 between the tucked position and the patient-restrainingposition. Head section siderails 808, 810 can additionally be providedwith breakaway siderails 920 that move from the tucked position to agenerally vertically downwardly extending down-out-of-the-way positiondescribed below.

Breakaway Siderails

Breakaway siderails 920 allow the caregiver to move the siderailassemblies from the generally horizontal tucked position to a generallyvertically downwardly extending down-out-of-the-way position to provideclear access to chair bed 50 beneath intermediate frame 302 as shown inFIG. 36 and also to provide clear access beneath intermediate frame 302for equipment mounted on a C-arm. Breakaway siderails 920 accomplishthis by moving the siderail to a down-out-of-the-way position away fromthe side of chair bed 50 and by narrowing the width of the section ofchair bed 50 adjacent to the siderail for deeper C-arm insertion.

When chair bed 50 is provided with breakaway siderails 920, head sectionupper deck side portions 418, 420 include collateral head frames 922,924 as shown in FIG. 36. Each collateral head frame 922, 924 ispivotably mounted to upper deck side portion 418, 420 by a hinge 926,928. Each collateral head frame 922, 924 can swing between an upposition, as shown, for example, by collateral head frame 924 in FIG.36, and a generally vertically downwardly extending down-out-of-the-wayposition, as shown, for example, by collateral head frame 922 in FIG.36. Preferably, hinges 926, 928 are connected to head end 52 ofcollateral head frames 922, 924 so that collateral head frames 922, 924are adjacent to head end 52 of chair bed 50 when collateral head frames922, 924 are in the down-out-of-the-way position. Each collateral headframe 922, 924 can be locked into the up position by a pin 930configured to be received by an opening (not shown) in upper deck sideportion 418, 420 and an opening 932 in collateral head frame 922, 924.

Mounting brackets 818 are fixed to collateral head frame 922, 924 andare configured to move with collateral head frames 922, 924 so thatsiderails 808, 810 swing between the generally horizontal tuckedposition and the generally vertically downwardly extendingdown-out-of-the-way position when collateral head frames 922, 924 movebetween the up position and the down-out-of-the-way position as shown inFIG. 36. When a caregiver wishes to move head section siderails 808, 810to the down-out-of-the-way position, such as when preparing chair bed 50for use during a procedure including the use of equipment mounted on aC-arm, the caregiver can raise intermediate frame 302 to the raisedposition, rotate the appropriate head section siderail 808, 810 to thetucked position, remove pin 930 from opening 932 in collateral headframe 922, 924 and from the opening (not shown) in upper deck sideportions 418, 420, and swing siderail 808, 810 from the tucked positionto the down-out-of-the-way position.

Mechanical Angle Indicators

Siderails 808, 810, 812, 814 can additionally be provided with angleindicators 938 as shown, for example, in FIGS. 37-39. Head sectionsiderails 808, 810 include indicators 938 as shown in FIG. 37 thatgenerally indicate the angular orientation of head section 404 of deck402, and body section siderails include angle indicators 938 as shown inFIG. 39 that generally indicate the angular orientation of intermediateframe 302 relative to base frame 62. Thus, angle indicators 938 on bodysection siderails 812, 814 are sometimes referred to as Trendelenburgindicators or Trend indicators. Mounting angle indicators 938 onsiderails 808, 810, 812, 814 prominently displays angle indicators 938so that the caregiver can quickly and easily judge the status of chairbed 50.

Each angle indicator 938 includes a housing 940 having an interiorregion 942 defined by a rear wall 944 formed in siderail 808, 810, 812,814 and a front wall 946 connected to siderail 808, 810, 812, 814 asshown in FIG. 38. An indicator member 948 is received by interior region942 for movement therein relative to housing 940 as the angularorientation of siderail 808, 810, 812, 814 and angle indicator 938changes. The position of indicator member 948 relative to housing 940indicates the angular orientation of angle indicator 938. Housing 940can be formed so that rear wall 944 is arcuate across the face ofsiderail 808, 810, 812, 814 as shown in FIG. 37 and indicator member 948can be spherical and can be positioned to lie on and to roll alongarcuate rear wall 944 as the angular orientation of angle indicator 938changes.

Preferably, indicator member 948 includes an indicator surface 950 thatis visible through front wall 946 of housing 940. Markings 952 that arestationary relative to housing 940 can be positioned to lie adjacent tofront wall 946 so that markings 952 and indicator member 948 cooperateto indicate the position of indicator member 948 relative to housing940, thus indicating the angular orientation of siderails 808, 810, 812,814.

Angle indicator 938 mounted to head section siderail 808, 810 includes afirst end 954 positioned to lie toward head end 52 of siderail 808, 810and a second end 956 positioned to lie toward foot end 54 of siderail808, 810 and positioned vertically higher than first end 954 as shown inFIG. 37. When head section 404 is in the down position, shown in FIG.37, indicator member 948 is toward first end 954. When head section 404moves from the down position to the back-support position, indicatormember 948 moves from first end 954 toward second end 956. Indicatormember 948 is infinitely positionable relative to housing 940 betweenfirst end 954 and second end 956 and the positions of indicator member948 correspond to positions of head section 404 between the downposition and the back-support position.

Angle indicator 938 mounted to body section siderail 812, 814 issubstantially identical to angle indicator 938 on head section siderail808, 760, except that first and second ends 954, 956 are positioned tolie on generally the same horizontal plane as shown in FIG. 39. Whenintermediate frame 302 is generally horizontal, body section siderail812, 814 is generally horizontal and indicator member 948 is positionedto lie generally half-way between first end 954 and second end 956. Whenintermediate frame 302 moves to the Trendelenburg position, intermediateframe 302, body section siderail 812, 814, and angle indicator 938 moveso that indicator member moves toward first end 954 of housing 940. Whenintermediate frame 302 moves to the reverse Trendelenburg position, bodysection siderail 812, 814 and angle indicator 938 move so that indicatormember moves toward second end 956 of housing 940. Indicator member 948is infinitely positionable relative to housing 940 between first end 954and second end 956 and the positions of indicator member 948 correspondto positions of intermediate frame 302 between the Trendelenburgposition and the reverse Trendelenburg position.

Alternatively, an angle indicator can be a spirit level having a housingfilled with a fluid to form a liquid-filled bulb type bubble spiritlevel. In such a spirit level, the position of the bubble relative tothe housing changes as the angular orientation of the spirit levelchanges, the position of the bubble relative to the housing indicatingthe angular orientation of the spirit level.

Controls on Siderails

Siderails 808, 810, 812, 814 can additionally be provided with controlsfor operating bed 50 and moving bed 50 to various positions. Controlscan include control buttons 960 on a bed side of the siderail 960 foruse by a person (not shown) on sleeping surface 550 as shown in FIGS. 40and 41. Typically, the person's head will rest on head end 52 ofsleeping surface 550. To accommodate the person on sleeping surface andallow the person to easily locate and view control buttons 960, controlbuttons 960 can be angled toward head end 52 of deck 402 as shown inFIGS. 40 and 41 so that faces 961 of buttons 960 are toward head end 52of deck 402. Bed 50 can also be provided with a second plurality ofcontrol buttons (not shown) on an outside of the siderail for use by aperson outside of bed 50 as described below.

Siderail 812 is coupled to the side of deck 402 for movement between thepatient-restraining position and the tucked position. A pad 962 having adisplay screen 964 can be provided on a side of siderail 812 outside ofbed 50 as shown in FIGS. 39 and 42 for use by the caregiver. Preferably,pad 962 is mounted to siderail 812 to pivot outwardly for easy viewingof display screen 964 as shown in FIG. 42. For example, pad 962 can bemounted to the outside of siderail 812 and can be configured to pivotupwardly about a pivot axis 966 adjacent to the top of pad 962. Thismovement of pad 962 particularly allows for easy viewing of displayscreen 964 by a person standing next to the bed 50 even when siderail812 is in the tucked position.

FIG. 48 is a block diagram illustrating the plurality of electroniccontrol modules for controlling operation of the hospital bed. Asdiscussed above, the plurality of modules are electrically coupled toeach other using a twisted pair network channel in a peer-to-peerconfiguration. The peer-to-peer network extends between first and secondnetwork terminators 1012 and 1013. The network connections areillustrated by the solid black lines in FIG. 48. Discrete connections toeach of the modules are illustrated by the dotted lines in FIG. 48. Thebold line of FIG. 48 illustrates an AC power connection.

Network terminator 1012 is coupled to an air supply module 1014. Airsupply module 1014 is coupled via the network cable to accessory portmodule 1016. Accessory port module 1016 is coupled to the bedarticulation control module (BACM) 1018. BACM 1018 is coupled to acommunications module 1020. Communications module 1020 is coupled toscale instrument module 1022. Scale instrument module 1022 is coupled tosurface instrument control module 1024. Surface instrument module 1024is coupled to position sense and junction module 1026. Position sensemodule 1026 is coupled to the network terminator 1013. A left sidestandard caregiver interface module 1028 is also coupled to the networkby a connection in position sense module 1026. The right side standardcaregiver interface module 1030 and the graphic caregiver interfacemodule 1032 are also coupled to the network using a connection in theposition sense module 1026.

It is understood that the modules can be rearranged into a differentposition within the peer-to-peer network. The modules are configured tocommunicate with each other over the network cable without therequirement of a master controller. Therefore, modules can be added orremoved from the network without the requirement of reprogramming orredesigning a master controller. The network recognizes when a module isadded to the network and automatically enables a control interface suchas graphic caregiver interface module 1032 to display specific modulecontrols for the added module. This eliminates the requirement forcontrols on individual modules. The module recognition feature isdiscussed in detail below.

Each module is connected to its appropriate sensors and actuators sothat it can perform its dedicated function. The following is a briefdescription of each electronic module:

Power for the communication network is supplied by a power supply andbattery charge module 1062. Power supply 1062 is coupled to a powerentry module 1063 and an AC main plug 1065. Power Supply/Battery chargemodule (PSB) 1062 converts the AC Mains input 1065 to DC levels to beused by the electronic modules. PSB 1062 contains filtering for the ACMains 1065 at the Mains entry point 1063. The PSB 1062 also providespower for limited bed functionality upon removal of the AC Mains powerinput via a battery 1067. The PSB 1062 contains an automatic batterycharging circuit with output to indicate battery status (i.e., batterydead, battery low, battery OK). PSB 1062 also controls the hydraulicpump 1055.

Bed Articulation Control Module (BACM) 1018—The BACM 1018 primarilycontrols the hydraulic system used to articulate the bed. BACM 1018accepts inputs from various user interfaces located throughout the bedto control bed articulations. This control input is qualified with aposition sensing input representing the actual locations of the bed decksections, along with patient lockout controls, to determine whether thebed should articulate. The BACM 1018 is present in every bed. BACMincludes a real time clock circuit to set the time for various othermodules.

Position Sense module 1026 detects the angles of all the appropriate beddeck sections. In addition, it interfaces to the bed exit detect, andthe four (4) siderail UP sensors. The position sense module 1026 outputsthis information to the network. These functions may be incorporatedinto the BACM 1018 and Bed-Side Communications Interface module 1020.The position sense module 1026 also provides the interconnections of thebed network and hospital communications links to the siderail standardcaregiver interface 1028 and 1030 modules.

Siderails (SIDE)—The siderails will contain standard caregiver interfacemodules 1028 and 1030 consisting of input switch controls, output statusindicators, and an audio channel. The standard caregiver interfacemodules 1028 and 1030 are coupled to patient control mechanisms for bedarticulations, entertainment, surface, lighting, Bed Exit, and NurseCall.

Scale Instrument Module 1022 translates the signals from the embeddedload beams into actual weight measured on the weigh frame. Scale module1022 outputs this weight to the Graphic Caregiver Interface Module (GCI)1032 for display purposes. This weight is also available to thecommunications module 1020 for transmittal to the hospital informationnetwork. Scale module 1022 includes Bed Exit and weight gain/loss alarmdetection capability.

Surface Instrument Module 1024 controls the dynamic air surface. It willaccept input from the GCI 1032 to dictate system performancecharacteristics. Surface module 1024 uses the GCI 1032 to displayoutgoing system information. Surface instrument module 1024 alsointerfaces with the air supply module 1014 to control the air handlingunit 1046.

Sequential Compression Device (SCD)—This module will control theoptional compression boots. It will use the GCI 1032 for interfacing tothe caregiver.

Graphic Caregiver Interface Module (GCI) 1032 controls the scale 1022and surface module 1024 (including SCDs). In addition, GCI 1032 providescontrol input and text and graphic output capability for future designconsiderations. GCI 1032 utilizes a graphic display along with asoftware menu structure to provide for full caregiver interaction.

Communications module 1022 is the gateway between the patient'senvironment controls and bed status information residing on the bed, andthe hospital information/control network.

Bed Exit Sensor (BES) 1069 exists on non-scale beds. The BES connects tothe position sense module 1026 to detect a patient bed exit.

Brake-Not-Set Sensor (BNS) 1056 detects the state of the Brake/SteerPedal. It is connected to the BACM 1018.

Bed-Not-Down Sensor (BND) 1058 detects if the bed is fully down (bothHead and Foot Hilo). It is connected to the BACM 1018.

Siderail Up Detect Sensors (SUD) 1071 consists of four switches todetect the secure UP position of the siderails. The SUD 1071 isconnected to the position sense module 1026.

Night Light 1073 is a stand alone unit providing the night lightfunction. It is powered by low voltage AC coming from the PowerSupply/Battery module 1062.

Pendant 1048 provides for bed articulation control input throughaccessory port module 1016.

Patient Assist Arm Control 1050 is a functional equivalent of thestandard caregiver interface modules 1028 and 1030 controls in adifferent physical embodiment. The assist arm includes a control padcoupled to the accessory module 1016.

The air supply module 1014, the bed articulation control module 1018,the power supply module 1062, and the power entry module 1063 are allcoupled to the base frame of the hospital bed. The communications module1020, the scale instrument 1022, and the remote information interface1124 are all coupled to the intermediate frame. The left standardcaregiver interface 1028 and patient interfaces 1154 and 1156 are allcoupled to the left siderail. The right standard caregiver interface1030 and patient interfaces 1158 and 1160 are all coupled to the rightsiderail. Graphical caregiver interface module 1032 may either becoupled to the left siderail or the right siderail. The position sensemodule 1026 and surface module 1024 are each coupled to the weigh frame.It is understood that the position of each module can be changed.

FIG. 49 diagrammatically illustrates how the various modules are addedand removed from the network. The electronic network uses an EchelonLonTalk serial communications protocol for module to modulecommunication in the bed. The cable 1034 illustrated in FIG. 49 containspower and a twisted pair connection. The preferred protocol is RS-485with a transmission speed of 78 kbs. The cable 1034 is provided withconnectors 1036. Extra connectors 1036 are provided for moduleadditions. When the connectors 1036 are not coupled to a module, acoupler 1038 is provided to interconnect adjacent connectors 1036. Inorder to connect a particular module 1040 to the network, the coupler1038 is removed and connectors 1036 are coupled to mating connectors1042 of the module 1040. Connectors 1042 are electrically coupled withinthe module 1040 as illustrated by dotted line 1044.

Referring again to FIG. 48, air supply module 1014 is coupled to an airhandling unit 1046 by a discrete electrical connection. Air supplymodule 1014 controls compressor 1046 to inflate and deflate the mattresssurface of the bed as discussed in detail below (or in mainapplication).

The accessory port module 1016 provides connections to the network for apendant 1048, an assist arm control 1050, or a diagnostic tool 1052.Pendant 1048 is a hand held control unit which is movable from bed tobed. Therefore, pendant 1048 may be coupled and uncoupled from accessoryport module 1016 to control various functions of the bed. For example,the accessory port module 1016 can communicate with BACM 1018 to controlmovement of the bed. Assist arm controls 1050 provide input to accessoryport module 1016 from a control pad coupled to an assist arm extendingout over the patient support surface of the bed. The assist arm 1050 canbe used to control movement of the bed, as well as for other desiredfunctions. The pendant 1048 and assist arm control 1050 may include allthe controls of the right and left standard caregiver interface modulesdiscussed below.

Diagnostic tool 1052 is used for servicing the bed, either at the bedsite or from a remote location. A modem is coupled to accessory portmodule 1016 to provide a telephone line connection to the hospital bed.This permits information related to the bed from any module to beretrieved from the peer-to-peer network at a remote location. Forinstance, the amount of time that the surface of the bed is in use maybe detected at the remote location through the modem for billingpurposes. The diagnostic tool 1052 permits a remote operator tointerrogate every module of the electrical control network. Thediagnostic tool 1052 checks application dependent parameters, runs eachof the modules through a test procedure, and fully accesses all networkinformation. Diagnostic tool 1052 may be a hand held tool such as a laptop computer which is coupled directly to accessory port module 1016. Inaddition, a remote computer can be coupled to accessory port 1016 withthe modem link to provide a data link to the network. A Voice Mate™control system available from Hill Rom, Inc. may also be coupled toaccessory port module 1016 to control the bed.

The bed articulation control module (BACM) 1018 is the module thatcontrols movement of the bed. BACM 1018 controls actuation of aplurality of solenoids 1054 which open and close valves coupled tohydraulic cylinders to move the articulating deck sections of thehospital bed relative to each other. BACM 1018 is also coupled to aBreak Not Set sensor 1056 and a Bed Not Down sensor 1058. When BACM 1018receives an input signal from the network requesting movement of the bedto a predetermined position, the BACM 1018 first reads the position ofthe bed provided from position sense module 1026. If movement of aportion of the bed is necessary, BACM 1018 checks for a lockout signalfrom the left and right standard caregiver interface modules 1028 and1030. If the lockouts are not set, BACM 1018 controls activation of theselected solenoid 1054 and then BACM 1018 turns on the hydraulic pump1055 (gravity may also be used if appropriate) to actuate a selectedcylinder if necessary.

Details of the BACM 1018 are illustrated in FIG. 50. BACM 1018 includesa neuron controller 1060. Illustratively, neuron controller 1060 is aMC143150FU echelon neuron networking microprocessor available fromMotorola. Controller 1060 is coupled to the network through an RS-485transceiver 1061. BACM 1018 operates to move a plurality of solenoids1054 in a hydraulic manifold to open and close control valves coupled tothe hydraulic cylinders and articulate the bed based on various networkcommands received from the peer-to-peer network. Neuron controller 1060receives commands from the right and left siderail standard caregiverinterface modules 1028 and 1030, the graphic caregiver interface 1032,or from another input device to articulate the bed. Neuron controller1060 also receives other information from the network regarding theposition of the head, seat, thigh, and foot deck sections of thearticulating deck of the bed. Therefore, neuron controller 1060 controlsthe solenoids and pump to stop articulating the bed as a limit isreached or when the particular bed section reaches its desired orselected position.

Both the articulating deck of the bed and the height of the deck arecontrolled by the BACM 1018. Upon receiving a bed function command fromthe network, the BACM 1018 energizes the appropriate solenoids andprovides a control signal to the Power Supply/Battery Module 1062illustrated in FIG. 48 to power the hydraulic pump, if necessary. BACM1018 may use bed position information provided by the remotely mountedbed position transducers. Alternatively, the position of the varioussections of the articulating deck may be supplied to BACM 1018 by theposition sense module 1026. BACM 1018 also instructs air supply module1014 and surface control module 1024 via the network to partiallydeflate a seat section and a foot section of the mattress when the bedmoves to a chair position. BACM 1018 also receives lockout informationfrom the siderail standard caregiver interface modules 1026 and 1028 todetermine whether or not a particular section of the articulating deckshould move.

Neuron controller 1060 executes code stored in EPROM 1064.Illustratively, EPROM 1064 is a 27C256-70 EPROM available from AMD. Inorder to conserve power, BACM 1018 uses a pulse width modulation (PWM)control system to minimize the current draw required to actuate thesolenoids 1054. Conventional control systems simply turn the solenoid1054 full on or full off and, as the voltage varies, current consumptiongoes up and down accordingly. With the PWM control design of the presentinvention, as the voltage varies BACM 1018 controls the power that isapplied to the solenoid 1054 to maintain substantially the same currentlevel to minimize power consumption. Neuron controller 1060 controls atiming generator 1066 through a memory map address decoder 1068. Memorymap address decoder 1068 provides a signal to timing generator 1066 online 1070 to start PWM and provides a signal on line 1072 to timinggenerator 1066 to stop PWM. Neuron controller 1060 provides a 5 or 10MHz clock signal to timing generator 1066 on line 1074.

Timing generator 1066 provides six different time periods in which toactuate one of six pairs of solenoids 1054 used to control the valves ofthe hydraulic cylinders. Each time period is about 50 milliseconds. Onlyone solenoid 1054 can be pulled during any one time period. Thisminimizes the maximum current draw on the power supply or battery at anygiven time. It is understood that a different number of solenoid pairsmay be controlled in accordance with the present invention. The numberof time periods and the time period intervals may be changed, ifdesired. In the illustrated embodiment, six pairs of solenoids arecontrolled by the BACM 1018. One solenoid of each pair is used to open afirst valve to control movement of a deck section in a first direction,and the other solenoid of each pair is used to open a second valve tocontrol movement of the particular section in an opposite direction.Therefore, a pair of solenoids is provided for the head sectioncylinder, the foot section cylinder, the foot Hi Lo cylinder, the headHi Lo cylinder, the knee section cylinder, and the foot retractingsection cylinder.

Timing generator 1066 supplies a PWM enable signal on line 1076 to asolenoid PWM select logic control circuit 1078. Timing generator 1066also provides time division terms to PWM control circuit 1078 on line1080.

Illustratively, there are twelve different solenoids 1054 powered by FETdrivers 1090. Neuron controller 1060 can provide three separate commandsfor each solenoid. The commands include an extend command, a retractcommand, and a pull-in command. The extend command is used to select thecorrect solenoid which when energized will extend the appropriatecylinder. Steady-state control of the FET which powers the solenoids ispulsed ON and OFF at the PWM rate. The retract command is used to selectthe opposing solenoid which when energized retracts the cylinder. It toois turned ON and OFF at the PWM rate. When a solenoid is initiallyactivated or turned on, it is desirable to actuate the selected solenoidat “full on” for a predetermined time. Therefore, the pull-in commandoverrides the PWM control circuit.

Data including the control commands (pull-in, extend, or retract) for aselected solenoid 1054 transmitted from the neuron controller 1060 iswritten to buffer register 1084. To synchronize the commands stored inthe buffer register 1084 with the timing pulses from timing generator1066, the commands are shifted into a holding register 1088. Therefore,asynchronous information is received in buffer register 1084. Thisasynchronous information is synchronized into the holding register 1088using a timing generator pulse on line 1094. The timing signal 1094synchronizes the pull-in latch 1082 in buffer register 1084 and thepull-in latch 1086 in the holding register 1088 with the timinggenerator 1066. Timing signal 1094 also synchronizes the solenoid“extend” latches 1096 and 1098 and the solenoid 1054 “retract” latches1100 and 1102 with the timing generator 1066.

The PWM select logic control circuit 1078 receives commands from theholding register 1088 and provides signals to drive a discrete FETthrough FET drivers 1090 during each timing interval of the PWM timinggenerator 1066. Driver 1090 pulls the selected solenoid 1054 down toground and applies a voltage across the selected solenoid 1054 tocontrol the solenoid. A voltage clamp 1104 is coupled to each of thesolenoids 1054. When power is removed from a particular FET an inductivesignal is supplied to the solenoids 1054. Voltage clamp 1104 clamps theinductive signal to the voltage rail. Therefore, voltage clamp 1104provides voltage spike suppression.

A diagnostic block 1106 also receives current signals related to eachpair of solenoids 1054 from voltage clamp 1104 on line 1105. Only onesolenoid 1054 in each pair can be controlled or actuated at any giventime. Diagnostic block 1106 also receives a data command signal fromneuron controller 1060 on line 1108 indicating the particular solenoids1054 which are designated by the controller 1060 for activation.Therefore, diagnostic block 1106 compares the actual informationreceived from the solenoid 1054 pairs to the data received on lines1108. If the actual solenoid 1054 current does not match the desiredsolenoid 1054 activation data from controller 1060, diagnostic block1106 sends a signal to neuron controller 1060 on line 1110. A signal online 1110 actuates a signal on supervisory line 1112 coupled to a masterFET 1114 to turn off the master FET 1114 and shut off power to all thesolenoids 1054. The master FET 1114 is coupled in line with all twelvesolenoids 1054. Therefore, supervisory FET must be turned on to providepower to any one of the solenoids 1054.

A current sense resister 116 is coupled to the FET drivers 1090. Thecurrent sense resister 116 is coupled to the first input terminal of acomparator 1118. A second input terminal of comparator 1118 is coupledto a reference voltage. The output of comparator 1118 provides PWMfeedback signal to timing generator 1066 on line 1120. In order toprovide PWM, the current must be measured in each solenoid 1054.Therefore, the current sense resister 116 measures the current in eachof the six time slots used for controlling the solenoids 1054. Dependingon the measured current, the signal on line 1120 adjusts the timinggenerator 1066 to control the pulse width of the driver signal.Therefore, if too much current is being drawn, then timing generator1066 shortens the width of the driver pulse in order to bring thecurrent down.

Referring again to FIG. 48, communications module 1020 provides aninterface needed for bed-to-hospital or hospital-to-bed informationtransfer. Communications module 1020 is a gateway between the bednetwork and the hospital information/control network. Communicationsmodule 1020 is connected to a standard side-com interface 1122.Interface 1122 also provides direct hard wired links between the nursecall switches on the siderails of the bed and the hospital prioritynurse call network. Signals from these nurse call switches can also besent over the network. On beds without a scale, a switch input port isprovided to accept a bed exit signal coming from a bed exit sensor.

Interface 1122 supports all existing discrete wire protocols. Interface1124 will support newly defined serial protocols, both to hospitalnetwork and other hospital room equipment. Any other hospital roomequipment can use the GCI module 1032 as its user interface controlmodule.

Communications module 1020 also provides entertainment functions.Television, radio, or the like may be controlled by communicationsmodule 1020 based on input/output signals received/sent from the left orright siderail standard caregiver interface modules 1028 and 1030 overthe network or via discrete connections.

Communications module 1020 is directly coupled to the hospitalinformation electrical network to transmit and receive signals from aremote location. Communications module 1020 receives weight informationfrom scale instrument module 1022. Communications module also receivessurface setting information, including pressures and other parametersfrom surface instrument module 1024. Communications module 1020 alsoreceives bed position information from position sensing module 1026. Inaddition, communications module 1020 can receive all informationtraveling on the network.

The hospital network can drive a display on the graphic caregiverinterface 1032 using signals transmitted from the remote locationthrough a remote information interface 1124, to communications module1020, and then to graphic caregiver interface 1032 over the network.Therefore, communications module 1020 provides an interactive data linkbetween the remote location and the graphic caregiver interface module1032. Requests for weight acquisition can be automatically sent from aremote location through remote information interface 1124 andcommunications module 1020. Communications module 1020 then communicateswith scale instrument 1022 to determine the weight and then transmitsthe weight to the remote location via the remote information interface1124.

The scale instrument module 1022 receives input signals from load beamscoupled to a weigh frame of the bed. Specifically, scale instrumentmodule 1022 receives input signals from a left head load beam 1126, aright head load beam 1128, a right foot load beam 1130, and a left footload beam 1132. The scale module 1022 transmits weight information andoperation parameters to the GCI module 1032 and communications module1020. Load beams 1126, 1128, 1130, and 1132 are bolted to theintermediate frame. The articulating deck and weigh frame module is thenbolted to the load bearing ends of the load beams. Any item attached toor resting on the articulating deck and weigh frame will be weighed bythe load beams. Scale instrument module 1022 receives information fromthe network via a nurse caregiver interface unit or a graphic caregiverinterface module 1032. The scale acquires data from the load beamtransducers 1126, 1128, 1130, and 1132 and automatically factors in thetare weight to calculate a patient weight. Scale module 1022 transmitsan output signal to the network representing the patient weight. Scalemodule 1022 can detect bed exit and alert the hospital via thecommunications module 1020 and remote information interface 1124.

Scale module 1022 also provides a weight change alarm. Scale module 1022accepts a set point weight from the network. Scale module 1022 detectsif a patient's weight change has exceeded or dropped below a presetlevel from the initial set point weight. If a preset weight change hasoccurred, scale module 1022 provides an alarm message to the network.Scale module 1022 stores all data critical to the functioning of thescale in non-volatile memory. Scale module 1022 has built in diagnosticcapability to detect hardware integrity and data integrity.

Details of scale module 1022 are illustrated in FIG. 51. The four loadcells 1126, 1128, 1130, and 1132 are coupled to a four channel analog todigital converter 134. Illustratively, analog to digital converter is aCS5516, 4 MHz analog to digital converter available from CrystalSemiconductor. Analog to digital converter 134 converts analog signalsfrom the load cells 1126, 1128, 1130, and 1132 into digital signals andinputs the signals into the echelon neuron controller 1136. Neuroncontroller 1136 is a MC143150, 10 MHz networking microprocessoravailable from Motorola. Controller 1136 executes code stored in anEPROM 1138. Illustratively, EPROM 1138 is a 32K×8, model 27HC256 EPROMavailable from AMD.

Neuron controller 1136 stores calibration data related to each of theload cells 1126, 1128, 1130, and 1132 either in its internal memory orin external EEPROM 1140. Calibration data is necessary because each loadbeam 1126, 1128, 1130, and 1132 has slightly different gain or offsetconstant associated with it. Calibration/excitation relay 1142 transmitsthe calibration data from neuron controller 1136 to analog to digitalconverter 1134. Two connectors 1148 and 1150 are provided to couplescale module 1022 to the peer-to-peer communication network. Connector1148 is hard wired to connector 1150. An RS-485 transceiver 1149 iscoupled between connectors 1148 and 1150 and controller 1136.Transceiver 1149 takes logic inputs and outputs and converts them toRS-485 level signals for the network. For each of the modules on thepeer-to-peer network, a connecter such as connector 1148 is hard wiredto another connector such as connector 1150 that goes onto the next nodeor module in a daisy chain configuration. Scale module 1022 alsoincludes a +5VDC regulated power supply 1152.

Referring again to FIG. 48, the surface instrument module 1024 isprovided for controlling operation of the mattress or support surface.Details of this module are discussed below with reference to the surfacedesign (or in main application).

The bed includes position transducers mounted throughout the bed tosense any needed positions of individual bed sections for articulationand caregiver interface purposes. The position sense module 1026 alsointerfaces a Siderail Up Detect Sensor, and a Bed Exit Sensor.

Details of the position sense module 1026 are illustrated in FIG. 52.Illustratively, the position transducers are discrete tilt sensors onvarious deck sections of the bed. The sensors include a Trendelenburglimit sensor at 13° relative to earth, a reverse Trendelenburg sensor at−13° relative to earth, and a bed-level at 0° relative to earth. Inaddition, the articulating deck sections include position transducerswhich are also discrete tilt sensors. Illustratively, the tilt sensorsare model A½ sensors available from AEC. The patient head limit sensordetects the head section at 55° relative to earth. The head contourlimit sensor detects the head section at 30° relative to earth. The kneecontour limit detects the knee section at 12° relative to earth. Thepatient foot limit detects the position of the foot section at 30°relative to earth.

The sensor inputs are coupled to the position sense module 1026. Thesensor input signals are signed conditioned using a RC filter 1154. Theoutput of RC filter 1154 is coupled to a neuron controller networkingmicroprocessor 1156. An output from controller 1156 drives a local alarm1158. Input power on line 1160 is coupled to a regulated power supply1162 which produces a +5V output. The output from power supply 1162 iscoupled to neuron controller 1156 and to a network transceiver 1164. Theposition transducers illustratively switch from a logic high to a logiclow upon detection of the particular angle relative to earth.

Controller 1156 transmits and receives network information throughtransceiver 1164. Network transceiver 1164 is coupled to a first networkconnector 1165 via lines 1166. Position sense module 1126 also providesthe connection points to the network for the left and right standardcaregiver interface modules 1028 and 1030. Network connector 1165 alsocoupled to a left siderail network connector 1170 which is coupled tothe left siderail standard caregiver interface module 1128. Leftsiderail connector 1170 is coupled to a right siderail connector 1172 bylines 1171. Connector 1172 is coupled to a right siderail standardcaregiver interface module 1030. Connector 1172 is also coupled to asecond network connector 1173 by lines 1175. Therefore, position sensemodule 1026 is also a junction module for connection to the left andright siderail standard care giver interface modules 1028 and 1030.

During operation, neuron controller 1156 interprets the sensor signalsreceived from RC filter 1154 and sends an output signal indicative ofthe state of each sensor to the network through network transceiver1164. Network transceiver 1164 is a RS-485 protocol transceiver. Alarm1158 contains a piezo device so that any alarms on the bed that aretransmitted through the network turn on the piezo alarm on the positionsense module 1026. These alarms may include bed exit, patient weightgain, weight loss, surface pressure loss, or other desired alarms. Alarm1158 can also be used to alert an operator when catastrophic failuresare detected in the bed by the diagnostic tools.

The left and right standard caregiver interface modules 1028 and 1030are substantially identical. The left standard caregiver interfacemodule 1028 is coupled to patient controls including an articulation andentertainment interface in the left siderail as illustrated at block1154 of FIG. 48. Standard caregiver interface module 1028 is alsocoupled to a surface patient interface on the left siderail asillustrated at block 1156. The standard caregiver interface module 1030for the right side is coupled to articulation and entertainment patientinterface module on the right siderail as illustrated at block 1158. Theright standard caregiver interface module 1030 is also coupled to asurface patient interface caregiver interface on the right siderail asillustrated at block 1160.

Details of the left standard caregiver interface module 1028 isillustrated in FIG. 53. The standard caregiver interface module includesan echelon controller 1162 which is a networking microprocessor. Echeloncontroller 1162 is coupled to a +5.0V supply voltage from power supply1164. Echelon controller 1162 is also coupled to a network transceiver1166. Transceiver 1166 is an RS-485 protocol transceiver. Transceiver1166 couples controller 1162 to the peer-to-peer communication networkas illustrated at line 1168. A network connection for the graphiccaregiver interface module 1032 is provided at line 1170 for both theleft and right standard caregiver interface modules 1128 and 1030.Graphic caregiver interface module 1032 can be connected on either theleft or right side of the bed. Echelon controller 1162 interprets thenetwork messages. Network controller 1162 also detects switch activationfrom the articulation and entertainment patient interface 1154 and thesurface patient interface 1156 and transmits output signals to thenetwork on line 1168. The switches can be dead function switches,lockout switches, bed exit switches, nurse call backlit switches, and soon. Controller 1162 drives a LED driver 1172 to light indicator LEDS1174 related to various bed status functions, such as bed-not-down,brake-not-set, battery low, and service required.

The LED driver 1172 is also coupled to a backlighting switch 1176 of thearticulation and entertainment patient interface 1154. Backlightingswitch 1176 is coupled to backlighting LEDs 1178. Backlighting switch1176 is also coupled to backlighting LEDs 1180 on the surface patientinterface 1156.

The standard caregiver modules 1028 and 1030 connect all the caregiverinterfaces switches in a row/column type architecture to provide a 4×10matrix. A keyboard row selection logic circuit is used to detect switchpresses as illustrated at block 1182.

The standard caregiver interface (SCI) modules 1028 and 1030 include thenetwork circuitry for interfacing all caregiver and patient siderailcaregiver interfaces to the communication network. The patient caregiverinterfaces are separated into modules which can be connected to the SCImodule 1028 or 1030 in a modular fashion.

Each SCI module 1028 and 1030 includes bed articulation switches 1184.These include head up, head down, knee up, knee down, foot up, footdown, bed up, bed down, chair in, chair out, Trendelenburg, and reverseTrendelenburg. In the case of a switch closure, a signal is periodicallyoutput to the network until the opening of the switch occurs. The SCImodules 1028 and 1030 further include lockout switches 1186 as discussedbelow, bed exit switches 1188, nurse call switches 1190, andbacklighting switches 1192. Control buttons for the switches 1184, 1186,1188, 1190, and 1192 are typically on an outside portion of the siderailfor use by a nurse.

The articulation and entertainment patient interface 1154 also includesa nurse call switch 1194, interactive TV switches and a light switch1196, and bed articulation switches 1198. Surface patient interface 1156includes nurse call LEDs 1200, mattress switches 1202, and a nurse callswitch 1204.

As discussed above, the lockout control switches are located on the leftand right siderail control interfaces. As illustrated in FIG. 54, thelockout control includes a global enable lockout activation switch 1205which must be pressed in order to activate any of the other lockouttoggle switches for the foot control lockout 1207, the knee controllockout 1209, the head control lockout 1211, or the lockout for allcontrols at 1213. This double lockout activation reduces the likelihoodof the accidental deactivation of one of the lockout control switches.Therefore, the global enable switch 1205 must be pressed in order toturn any of the other lockout controls on or off. The global enableswitch 1205 automatically deactivates after about 5 seconds ofinactivity. After the global enable is deactivated, the lockout statuscannot be changed. Since the caregiver controls are within reach of apatient, the global enable switch may be used to enable and disable boththe patient and caregiver bed articulation control switches.

A graphic caregiver interface (GCI) module 1032 is illustrated in detailin FIG. 55. The GCI module 1032 provides an enhanced menu-drivencaregiver input and output for bed articulation, scale, surfacecaregiver interface, and sequential compression device controller, andall other modules needing this type of user interface. The GCI module1032 includes a LCD display 1206, which is illustratively a 320×240,model DMF 50081 available from Optrex. Display 1206 may also be a320×240, model G321EX available from Seiko. Display 1206 outputsgraphical information to the caregiver. A switch panel 1208 permits thecaregiver to input information into the GCI module 1032. Switch panel1208 may be a series of discrete switches or an alpha/numeric keypad.Switch panel 1208 is coupled to a connector 1210. Connector 1210 iscoupled to an input of CPU 1212. CPU 1212 is illustratively an 80C188XL,10 MHz CPU available from Intel. The input device for the caregiver mayalso be an encoder 1214 which is coupled to a connector 1216. Connector1216 is coupled to CPU 1212. Illustratively, encoder 1214 is a rotaryencoder.

Connection to the peer-to-peer communication network is provided atterminal 1218. The network connection is made to a RS-485 transceiver1220. Transceiver 1220 is coupled to a +5 VDC regulated power supply1222. Transceiver 1220 is also coupled to a +12VDC regulated powersupply 1224. Transceiver 1220 is coupled to an echelon neuron controllernetworking microprocessor 1226. Controller 1226 is illustratively anAMC143120, 10 MHz networking microprocessor available from Motorola.Neuron controller 1226 is coupled to an I/O test port 1228. Controller1226 is also coupled to CPU 1212. Software code for operating CPU 1212is stored in an EPROM memory 1230. Illustratively, memory 1230 is a512K×8 flash EPROM memory. Data is stored in static RAM memory 1232.Illustratively, memory 1232 is a 128K×8 memory chip. Additional memoryis provided in a 2K×8 EEPROM 1234. An output from CPU 1212 is coupled toa LCD backlight inverter 1236. Backlight inverter 1236 is coupled to LCDdisplay 1206 by connector 1238. Backlight inverter facilitates viewingof display 1206 in all types of room lighting. Inverter 1236 isconfigured to match the particular display 1206 selected.

CPU 1212 is also coupled to a LCD controller 1240. LCD controller 1240drives the display 1206 through a connector 1242. Controller 1240 iscoupled to a 32K×8 static video RAM 1244. As the CPU 1212 writes animage to LDC controller 1240, the controller 1240 stores the image inVRAM 1244 and then continuously refreshes the display screen 1206 withthe image stored in the VRAM 1244.

Contrast of the display 1206 is controlled by software contrastadjustment as illustrated at block 1246. A LCD bias supply voltage atblock 1248 is coupled to connector 1242. Supply 1248 converts a +5Vinput or a +12V input into a −22V output. An external watchdog timer1250 monitors CPU 1212. If the CPU 1212 does not pulse the particularline on a periodic basis, timer 1250 resets the system.

GCI module 1032 also includes a diagnostic port 1252. Diagnostic port1252 is coupled to CPU 1212 through a serial port 1254. Serial port 1254is a RS-232 UART. Therefore, a laptop may be connected at port 1252 tointerrogate the CPU 1212. CPU 1212 can access and send information tothe network through controller 1226.

The GCI module 1032 provides an enhanced menu-driven caregiver input andoutput control for bed articulation, scale, surfaces, sequentialcompression devices, and all other modules needing this user interfacecapability. The GCI module 1032 is intended to be a drop in replacementfor Scale/Surface Nurse Control Unit. GCI module 1032 interacts withscale module 1022. Specifically, GCI module 1032 can transmit a requestfor patient weight to the scale module 1022. In addition, the GCI module1032 can also zero the scale and perform other scale module functions.

GCI module 1032 stores predetermined graphics data and caregiverinterface data in memory 1230. This predetermined graphics data isstored in the GCI module 1032 at the time of production. Additionally,other modules on the peer-to-peer communication network can downloadscreen formats to the GCI module into static RAM 1232. The GCI modulethen retrieves the stored graphic screen formats either from memory 1230or static RAM 1232 and displays the output on display 1206. By providingstored built-in graphics in memory 1230, the GCI module 1032 can supportproducts or other modules that may later be connected to thepeer-to-peer communication network. By providing the storedpredetermined graphic formats, the GCI module 1032 does not have to beupdated each time a new module is added to the system. If the desiredgraphics format is not present in memory 1230, then the newly addedmodule must download the desired graphic formats into RAM 1232 at runtime.

The specific graphic formats stored in the GCI module 1032 can includecharting formats such as bar graphs, X-Y graphs, pie charts, etc., iconsor pictures representing each of the modules in the communicationnetwork, or any other type of graphical format desired. Graphic formatsfor use by the modules are stored in two different ways in the GCImodule 1032. Typically, these various graphic formats are stored inEPROM 1230 at the time of manufacture. In other words, these graphicalformats are typically designed into the GCI module 1032. If a particularGCI module 1032 does not include the desired graphic format stored inmemory 1230, then the particular graphic format for the new module addedto the system is downloaded into the static RAM 1232 of GCI module 1032after the bed is powered up. For instance, if GCI module 1032 does notinclude a X-Y graphic format in memory 1230, this graphic format can bedownloaded into RAM 1232 after the bed is powered up. Once a particulargraphic format is stored in GCI module 1032, in either memory 1230 orRAM 1232, the new module transmits only data to the GCI module 1032during operation. The GCI module 1032 uses the received data and thestored graphic format to produce an appropriate screen output on display1206. For instance, after the X-Y graphic format is stored in eithermemory 1230 or RAM 1232, the particular module transmits only the X-Ydata to the GCI module 1032 over the network. The GCI module 1032 thenuses this data along with the stored X-Y graphic format to provide anoutput to display 1206. Each new module will also download a particularicon representative of the new module for the menu-driven display 1206of GCI module 1032 as discussed below.

Updating of the graphic formats and menu information of the GCI module1032 can be accomplished in one of three ways. The particular graphicformat and menu information can be downloaded into static RAM 1232 atpower up of the bed. The graphic format and menu information can also bedownloaded to EEPROM 1234 during installation of a new module. Finally,EPROM 1232 can be changed to include the new graphic format and menuinformation at the time the new module is installed.

Details of the operation of GCI module 1032 for automaticallyrecognizing and controlling newly added modules on the communicationnetwork are illustrated in FIGS. 56 and 57. Bed power up is illustratedat block 1260. A graphics status flag and a menu saved status flag areboth cleared at block 1262. These flags provide an indication of whethera particular graphic format or menu information for the module must bedownloaded to the GCI module 1032. For each module on the network, menuscreens will be provided on display 1206. Therefore, if a particularmodule is selected using the GCI module 1032, control options for thatmodule will appear as menu items on display 1206. Once a particularcontrol option is selected, additional menu items for the selectedcontrol option may appear, and so on.

GCI module 1032 performs a system query at block 1264. GCI module 1032first determines whether any modules are present on the communicationnetwork which use the GCI module 1032 as illustrated at block 1266. Ifno modules are present on the network which use the GCI module 1032, theGCI module 1032 returns to block 1264. The system query is carried outat predetermined time intervals.

If modules are present which use the GCI module 1032 at block 1266, theGCI module 1032 determines whether any of the modules need to downloadgraphic formats to the GCI module 1032 as indicated at block 1268. If nomodules need to download graphic information, GCI module 1032 advancesto block 1274. If any of the modules need to download graphic formats,the graphic formats are downloaded to static RAM 1232 of GCI module 1032as illustrated at block 1270. The graphics status flag for the module isthen updated as illustrated at block 1272. The graphics status flag isinitially generated at block 1266 during detection of any modules whichuse the GCI module. Therefore, after step 1270 the status flag 1272indicates that all the graphic format data for the particular module isnow stored on the GCI module 1032.

GCI module 1032 next determines whether any of the modules need todownload menu structure information to the GCI module. If not, GCImodule 1032 advances to block 1280 in FIG. 57. If any of the modulesneed to download menu structure information, the appropriate menustructure information is downloaded to the static RAM 1232 of GCI module1032. This menu structure information provides the appropriatemenu-driven control for each module. For instance, once the module iconis selected using the switch panel 1208 or encoder 1214 of the GCImodule 1032, the GCI module 1032 automatically displays a menu screen ofoptions on display 1206 associated with the particular module. Once aparticular option is selected, another menu screen may be provided todisplay 1206 giving further options. Button sizes and text fonts areincluded in the graphics format data stored in the GCI module 1032. Themenu structure information provides the actual textural material to beincluded with the menu-screen buttons.

The GCI module 1032 next updates a menu saved status flag at block 1278.This status flag provides an indication that all the menu structureinformation for the particular module has been downloaded. GCI module1032 then proceeds to block 1280 of FIG. 57.

GCI module determines whether this particular loop is the first timethrough after power up or if a new module has been added as illustratedat block 1280. If not, GCI module 1032 proceeds to block 1286. If it isthe first time through or a new module has been added, GCI module 1032reconfigures an opening menu to include icons of all the modules presentas illustrated at block 1282. In other words, the main menu initialdisplay screen of display 1206 is updated to include an iconrepresenting each of the controllable modules. GCI module 1032 thenreconfigures existing menus to include the new options of added modulesas illustrated at block 1284. The code stored in the GCI module 1032 isaltered, in real time, to merge new menu information for the newly addedmodules with existing menu information of the previous modules.

GCI module 1032 then performs an integrity check on RAM 1232 based savedinformation as illustrated at block 1286 (i.e. checksum). If theintegrity of the stored information in RAM 1232 is not correct at block1288, GCI module 1032 changes an appropriate saved status flag at block1290. GCI module 1032 then proceeds back to block 1268 to download theappropriate graphical format information or menu structure informationfor the particular module again.

If the integrity of the information saved in RAM 1232 is correct atblock 1288, GCI module 1032 determines whether an input switch fromswitch panel 1208 or encoder 1214 has been pressed at block 1292. If noinput has been pressed, GCI module returns to block 1264 of FIG. 56 toperform another system query at the next predetermined time interval.

If an input switch has been pressed at block 1292, GCI module 1032updates the display screen 1206 as illustrated at block 1294. The GCImodule 1032 then transmits an appropriate network command to theparticular module to perform any selected application or specificfunction as illustrated at block 1296. For instance, GCI module 1032 cantransmit a signal to scale module 1022 to weigh a patient, to surfaceinstrument module 1024 and air supply module 1014 to adjust the pressurewithin a particular bladder of the bed surface, or to perform any othermodule function.

It is understood that the hospital network can use the GCI module 1032in an identical way to the other network modules. The hospital networkcan send menu driven control options to the GCI if desired. Either thepatient or the caregiver can use the GCI module 1032 to control bedfunctions and interact with the hospital network or another remotelocation.

The automated data collection feature of communications module 1020 isillustrated in further detail in FIG. 58. A request for bed informationand/or bed control is received as illustrated at block 1300. The requestis either from the hospital information network or from a remote dataacquisition system. In other words, the hospital bed may be connected tothe hospital network through wiring in a wall as discussed above. Inaddition, the bed may be connected to another piece of equipment in theroom which can be connected to a remote location through the hospitalnetwork, a modem, or other data link. Finally, the request forinformation and/or control can be from an on-board bed data acquisitionsystem.

The particular command or status request is then mapped to a networkvariable or value as illustrated at block 1302. In other words, thereceived request or command is changed to a usable network format atblock 1302. Illustratively, a table is used to transform the receivedrequest for information and/or control to an appropriate andunderstandable network command.

A message is then issued to the bed modules over the communicationnetwork as illustrated at block 1304. Communications module 1020determines whether the particular module responded over the network withan acknowledgment of the message at block 1306. Once a particular modulereceives a message, an acknowledgment of the message is transmitted backover the network before the particular function is carried out by themodule. If the acknowledgment is not received, the communication module1020 sets an error status indicator as illustrated at block 1308. If theacknowledgment is received at block 1306, communications module 1020next determines whether the module responds over the network with aparticular status that was requested or with an acknowledgment that aparticular control has been implemented as illustrated at block 1310. Ifnot, communications module 1020 sets the error status indicator asillustrated at block 1308. If the module did respond over the networkwith the particular status requested or with the acknowledgment that thecontrol was implemented, the network response is mapped to the off bednetwork as illustrated at block 1310. The communications module 1020transforms the response received from the bed network format to theoff-bed network format for transmission at block 1312. Thecommunications module 1020 then sends the off-bed network command or anerror message to the remote network as illustrated at block 1314. Anerror message sent to the hospital network or other remote locationprovides an indication that something went wrong with the particularrequest for status information or control. This request can then beretransmitted. A persistent error message indicates problems with one ofthe modules. Therefore, corrective action to repair the module can beimplemented.

Each of the modules on the hospital bed can store specific statusinformation related to operation and control of the bed or related tothe module functions in an internal memory present on each module. Forinstance, the BACM 1018 can store all bed articulations and positions ina memory of the BACM 1018. In addition, the surface instrument module1024 can store all surface positions and settings or therapy moduleusages in memory on the surface instrument module 1024. This informationcan be retrieved using the automated data collection feature discussedabove to indicate patient activity. The standard caregiver interfacemodules 1028 and 1030 can store all entertainment patient controlinteractions in memory. These interactions can be retrieved via theautomated data collection feature for billing or other monitoringpurposes. Each module has a capability of storing all patientinteraction with controls on the module. This stored information isavailable to the GCI module 1032 and to the off bed information systemvia the automated data collection feature.

As discussed above, the hospital network can retrieve status informationthrough the communications module 1020. In addition, status informationcan be retrieved from a remote location through a data link coupled toaccessory port module 1016. This status information may be bed statusinformation stored in any of the modules. Each module can store statusinformation related to switch presses, and specific movements, controls,or functions performed by the module.

Another module which can be coupled to the peer-to-peer communicationnetwork is a patient status module 1320. This patient status module 1320is illustrated in FIG. 59. The patient status module 1320 monitors andrecords vital statistics from the patient received from a selectedpatient monitoring device 1322. Such body monitors may include, forexample, temperature sensors, blood pressure detectors, heart ratemonitors, or any other body monitor. Data from these monitors 1322 isstored in memory of the patient status module 1320 and can betransmitted over the network to the hospital network or to a remotelocation through a data link coupled to accessory port 1016. Patientmonitoring devices 1322 are discretely coupled to the patient statusmodule 1320.

Another module coupled to the bed peer-to-peer communication network isa gateway module 1324. The gateway module 1324 provides an interface tothe network for an application specific module 1326. Specifically,gateway module 1324 provides echelon network interface circuitry forcommunicating with the peer-to-peer network of the hospital bed. Gatewaymodule 1324 also includes application specific interface circuitry forcommunicating with the application specific module 1326 for performing adedicated function on the bed or elsewhere. Therefore, gateway module1324 provides a format change for the data so that understandableinformation and commands are transmitted and received by both the bednetwork and the application specific module 1326.

Another feature of the present invention is that each of the bed modulescan be upgraded over the network using a data link through accessoryport 1016 or using communications module 1020. Upgrade information canbe transmitted from the remote location to the peer-to-peer network. Inother words, a remote location can be used to download new software toall the modules connected to the communication network of the bed. Thispermits an operator to reprogram the bed modules from a remote locationover the peer-to-peer communication network.

Yet another feature of the present invention is that each module is ableto perform internal diagnostics. After a module performs its dedicatedfunction, a diagnostic check can be performed to make sure that themodule is functioning correctly. If an error is detected, an errormessage can be transmitted over the network to another module or to aremote location through communications module 1020 or accessory port1016.

Another module of the present invention is illustrated in FIG. 60. FIG.60 illustrates an automatic charting module 1330. The automatic chartingmodule 1330 includes an echelon controller 1332 which is a networkingmicroprocessor. Controller 1332 accesses memory 1334. Memory 1334includes an EEPROM, and EPROM, and a static RAM. Controller 1332 iscoupled to a RS-485 transceiver 1336. Transceiver 1336 is coupled tofirst and second network connectors 1338 and 1340. Module 1330 includesan internal power supply 1342 coupled to a power input. Illustratively,power supply 1342 supplies a +5V supply voltage to controller 1332 online 1344. Power supply 1342 also supplies power to a bar code interface1346, a display interface 1348, and a keyboard interface 1350. Displayinterface 1348 and keyboard interface 1350 are optional elements ofcharting module 1330.

Bar code interface 1346 receives an input from bar code scanner 1352. Anoutput of bar code interface 1346 is coupled to controller 1332 on line1354. Controller supplies information to display interface 1348 on line1356. An output from display interface 1348 is coupled to a suitabledisplay 1358. Keyboard interface 1350 receives an input from a keyboard1360. An output of keyboard interface 1350 is coupled to controller 1332by line 1362.

Charting module 1330 provides an apparatus for automatically chartingpatient information. Bar code scanner 1352 and keyboard 1360 provideinput devices for inputting information into charting module 1330. It isunderstood that any type of input device can be used in connection withthe present invention. The patient or caregiver can input information tothe network using the bar code scanner 1352 or keyboard 1360. Thisinformation can remain locally on the peer-to-peer communication networkof the hospital bed. In addition, the information can be sent to thehospital network through transceiver 1336 and communication module 1020or to another remote location via accessory module 1016.

An output device such as display 1358 is provided to display informationto the user. The display 1359 can be a series of LEDS or a displaypanel, such as a LCD display.

The memory of 1334 of charting module 1330 is loaded in a manner similarto the GCI module 1032 discussed above. Memory 1334 contains code thattranslates raw bar code scanner information and keyboard inputinformation from keyboard 1360 into specific network commands, eitherfor local on-bed use or for hospital network off-bed use. For instance,the nurse can scan bar codes directly from prescription medicine orinput various information into keyboard 1360 related to the patient.This input is used to generate an internal chart of the medical historyof the patient for use on the hospital bed. This chart data can bedisplayed on display 1358. In addition, this chart can be transmittedover the hospital network or transmitted to a remote location using adata link coupled to accessory port 1016.

It is understood that the GCI module 1032 discussed above may bemodified to include an input interface such as bar code interface 1346.The functionality of charting module 1330 is similar to the GCI module1032 except for the scanning device 1352 and the bar code interface1346.

Another use of charting module 1330 is for inputting a control sequenceused to control a module to perform a dedicated function on the bed. Forinstance, a doctor can prescribe a certain surface therapy for pulmonaryor other type of treatment of the patient on the bed. This treatmentprescription can specify a period of time for percussion and vibrationtherapy or for rotational therapy of the patient on the bed. Theprescription can include a specific period of time for the therapy withvarying rates of rotation or a varying frequency of percussion andvibration. This specific control sequence or prescription is encodedonto a bar code or other appropriate input scanning device format andscanned or otherwise input into charting module 1330. Charting module1330 then automatically executes the prescribed control sequence bytransmitting appropriate commands at appropriate times throughtransceiver 1336 to the network and to the selected modules to controlthe selected modules in the prescribed control sequence.

As discussed above, each of the network modules includes a echelonneuron networking microprocessor or controller. Each of the networkingcontrollers has a unique serial number which is different from theserial number on any other controller. At manufacturing time, a database is created to associate each unique serial number with the moduletype and manufacturing date. Any other desired information related tothe particular module may also be stored in the data base. Therefore,the hospital bed of the present invention provides an inventory controlfeature both in the plant prior to shipment of the beds and in the fieldat remote customer locations. A diagnostic tool coupled to accessoryport module 1016 through a data link or the hospital network coupled tocommunications module 1020 can instantly query a bed over thepeer-to-peer communication network to retrieve the unique serial numberassociated with all the modules on the network of the bed. Therefore, anoperator has access to an instantaneous inventory of all the modules andassociated features of a particular bed from a remote location formaintenance, repairs, recalls, upgrades, etc. An operator at a remotelocation can quickly determine the exact modules on the bed at any time.

The apparatus of the present invention can automatically poll beds at aremote location over the network by providing a query to all modules andretrieving all the serial numbers over the network. Therefore, by usingthe stored data base, an operator can determine an inventory of all bedmodules present in a hospital or other remote location.

Details of the modular therapy and support surface apparatus of thepresent invention are illustrated in FIG. 61. The support surface of thepresent invention is configured to be positioned over a bed deck 1596 ofa hospital bed. The support surface includes a surface foundation 1500located on the bed deck. An inflatable and deflatable surface footsection 1502 is located adjacent surface foundation 1500. For certainapplications, an upper foam support surface 1504 is located onfoundation 1500. Upper foam support 1504 is typically used for shorthospital stays. An upper air bladder 1506 can also be positioned oversurface foundation 1500. A rotation bladder 1508 is located between thesurface foundation and the bed deck. An optional percussion bladder 1510may be inserted in place of a section of upper air bladder 1506. Asequential compression device 1512 for venous compression therapy of apatient is also provided.

A plurality of separate treatment and surface control modules areprovided for interconnecting the various treatment devices and supportsurface bladders to the communication network of the bed and to on-boardair handling unit 1046. Specifically, the present invention includes afoot section control module 1014, a decubitus prevention control module1516, and a decubitus treatment control module 1518. The modular therapyapparatus further includes a pulmonary rotation control module 1520, asequential compression device air control module 1522, and a pulmonarypercussion and vibration control module 1524. An auxiliary air portcontrol module 1526 is also provided. The air port control module 1526provides for auxiliary air output for manual filling of auxiliarybladder systems for positioning, safety barriers, clinical treatmentssuch as burn contractures, and other purposes.

Each of the modules is designed to physically and functionally connectthe various bladders and treatment devices to both the communicationnetwork of the hospital bed through the surface instrument module 1024and to the air handling unit 1046 which is controlled by air supplymodule 1014. Air supply module 1014 is coupled to the peer-to-peercommunication network. Air supply electronics 1528 are connected to airsupply module 1014 for controlling air handling unit 1046 and switchingvalve 1530 based on network commands for controlling the various surfaceand treatment modules illustrated in FIG. 61.

Air handling unit 1046 is configured to supply air under pressure toswitching valve 1530 on line 1532. Air handling unit 1046 also applies avacuum to switching valve 1530 through line 1534. An output of switchingvalve 1530 is coupled to a connector block 1536. Connector block 1536provides an air and vacuum supply line to each of the surface controland treatment control modules as illustrated in block 1538 of FIG. 61.It is understood that dual control lines for both air and vacuum can besupplied to each of the surface control and treatment control modules ofFIG. 61. This dual control allows each module to apply pressure andvacuum simultaneously to different zones of a bladder or treatmentdevice.

The surface instrument module 1024 which is also coupled to thepeer-to-peer communication network is electrically coupled to each ofthe surface control modules and treatment control modules as illustratedin block 1540 of FIG. 61. This network connection permits all themodules to receive input commands from other network modules and tooutput information to the network.

Details of a therapy or support surface control module 1542 areillustrated in FIG. 62. It is understood that the details of footsection module 1514, prevention module 1516, treatment module 1518,pulmonary rotation module 1520, SCD air module 1522, pulmonarypercussion/vibration module 1524, and air port module 1526 include thesame or similar structural components as module 1542 illustrated in FIG.62. The FIG. 62 embodiment illustrates the air handling unit 1046coupled directly to connector block 1536 by both an air pressure supplyline 1544 and a vacuum supply line 1546. As discussed above, lines 1549and 1546 from air handling unit may be coupled to a switching valve 1530and only a single pressure/vacuum tube may be coupled to connector block1536 as illustrated in FIG. 61.

The connector block 1536 is coupled to module connector 1548 located onthe hospital bed. Specifically, connector block 1536 is coupled tomodule connector 1548 by a pressure supply line 1550 and a vacuum supplyline 1552. It is understood that a single supply line for both pressureand vacuum could also be used.

Module connector 1548 is also coupled to one of the surface or therapydevices as illustrated by a block 1554 by a pressure supply line 1556, avacuum supply line 1558, and a sensor supply line 1560. Depending uponthe particular surface or therapy device, more than one pressure,vacuum, and sensor lines may be connected between the connector block1548 and the surface or therapy device 1554. Typically, each separateair zone of the surface or therapy device will have its own pressure,vacuum, and sensor lines. For illustration purposes, however, only asingle set of supply lines will be discussed.

The bed also includes an electrical connector 1562 coupled to surfaceinstrument module 1024 of the peer-to-peer communication network of thebed by suitable cable 1564. The therapy or surface control module 1542illustrated in FIG. 62 is designed to facilitate coupling of the controlmodule 1542 to the bed. Each of the surface and treatment optionsillustrated in FIG. 61 is provided in the bed with a pneumatic connectorsuch as connector 1548 and an electrical connector such as connector1562 provided for each of the surface and therapy devices. The module1542 is easily installed by coupling connector 1548 on the bed to amating connector 1566 of module 1542. In addition, a mating electricalconnector 1568 is provided on module 1542 for coupling to electricalconnector 1562 on the hospital bed. The configuration of module 1542permits a simple “slide in” connection to be used to install the module1542 and activate the surface of therapy device 1554.

An air pressure input from pneumatic connector 1566 is coupled to anelectrically controlled valve 1570 by a supply line 1572. An output ofvalve 1570 is coupled to a pressure output port 1571 by line 1574. Port1571 is coupled to the surface or therapy device 1554 by pressure supplyline 1556.

The vacuum supply line 1552 from connector block 1536 is coupled to anelectrically controlled valve 1576 by line 1578 of control module 1542.An output of valve 1576 is coupled to a vacuum port 1577 of connector1566 by line 1580. Vacuum port 1577 is coupled to the surface or therapydevice 1554 by the vacuum supply line 1558. The electrically controlledvalves 1570 and 1576 are controlled by output signals on lines 1582 and1584, respectively, from a control circuit 1586 of module 1542. Controlcircuit includes a microprocessor or other controller for selectivelyopening and closing valves 1570 and 1576 to control surface or treatmentdevice 1554.

It is understood that several valves may be used for each surface ortreatment device. For instance, the upper air bladder 1506 may have aplurality of different air zones which are independently controlled. Inthis instance, separate pressure and vacuum and sensor lines are coupledto each zone of the air bladder. A electrically controlled valve isprovided for each pressure and sensor line in each zone to provideindependent controls for each zone.

Module 1542 also includes a pressure sensor 1588. Pressure sensor 1588is coupled to sensor supply line 1560 by line 1590. Pressure sensor 1588generates an output signal indicative of the pressure in the particularzone of the surface or therapy device 1554. This output signal frompressure sensor 1588 is coupled to the control circuit 1586 by line1592.

Control circuit 1586 is also coupled to an electrical connector 1568 bya suitable connection 1594 to couple the control circuit 1586 of module1542 to the surface instrument module 1024. Therefore, control circuit1586 can receive instructions from the other modules coupled to thepeer-to-peer communications network illustrated in FIG. 48. Controlcircuit 1586 can also output information related to the particularsurface or therapy device 1554 to the network. Specifically, thegraphical interactive display 1664 or the graphic caregiver interfacemodule 1032 is coupled to the electrical communication network fortransmitting command signals for the plurality of air therapy devicesover the electrical communication network to control operation of theplurality of air therapy devices. The graphical interactive displayincludes a display and a user input. Each control module transmitsdisplay commands to the display related to the corresponding air therapydevice. The display commands from the control modules provide a menudriven list of options to the display to permit selection of controloptions for the plurality of air therapy devices from the user input.

Details of the structural features of the modular therapy and supportsurface are illustrated in FIGS. 63-72. FIG. 63 illustrates a deckportion 1596 of a hospital bed. Illustratively, deck portion 1596 is astep deck having a cross-sectional shape best illustrated in FIGS.69-71. Illustratively, deck 1596 includes a head section 1598, a seatsection 1600, and a thigh section 1602. Sections 1598, 1600, and 1602are all articulatable relative to each other.

The modular therapy and support surface system of the present inventionincludes surface foundation 1500 including a foundation base 1606 andside bolsters 1608 and 1610. Preferably, side bolsters 1608 and 1610 arecoupled to opposite sides of foundation base 1606. Foundation base 1606includes foldable sections 1612 and 1614 to permit the foundation 1500to move when the step deck 1596 articulates.

The hospital bed also includes an expanding and retracting foot section410 to facilitate movement of the hospital bed to the chair position.Surface foot section 1502 is located over the retracting mechanical footportion 410. Surface foot section 1502 is described in detail below withreference to FIGS. 64-67.

The FIG. 63 embodiment includes an upper foam surface insert 1504configured to the positioned on the foam foundation base 1606 betweenside bolsters 1608 and 1610. Foam surface 1504 provides a suitablesupport surface for a patient who is mobile and whose length of stay isexpected to be less than about two days.

The surface foot section 1502 is particularly designed for use with thechair bed of the present invention. The foot section 1502 includes afirst set of air bladders 1618 and a second set of air bladders 1620alternately positioned with air bladders 1618. Air bladders 1618 and1620 are configured to collapse to a near zero dimension when air iswithdrawn from the bladders 1618 and 1620. The first set of bladders1618 are oriented to collapse in a first direction which is generallyparallel to the foot section 410 of the bed deck as illustrated bydouble headed arrow 1622. The second set of bladders 1620 are configuredto collapse in a second direction generally perpendicular to the footdeck section 410 as illustrated by double headed arrow 1624. Thisorientation of bladders 1618 and 1620 in foot section 1502 causes thefoot section 1502 to retract or shorten and to collapses or thin as thebladders 1618 and 1620 are deflated by the foot section control module1514 as the hospital bed moves from a bed orientation to a chairorientation. In the chair orientation, the foot deck section 410 andsurface foot section 1502 move from a generally horizontal position to agenerally vertical, downwardly extending position. Preferably, the footdeck section 410 moves from a retracted position to an extended positionto shorten the foot deck section as the articulating deck of the bedmoves to a chair configuration.

The minimizing foot section 1504 is further illustrated in FIG. 65. Thesurface foot section 1502 deflates as it moves from the bed position tothe chair position in the direction of arrow 1626. In the bed position,the surface foot section 1502 has a length of about 27 inches (68.6 cm)and a thickness of about 5 inches (12.7 cm) when the bladders 1618 and1620 are fully inflated. When in the downwardly extended chair positionillustrated at location 1628 in FIG. 65, the surface foot section isfully deflated and has a length of about 14 inches (35.6 cm) and athickness of preferably less than one inch (2.54 cm). The length of thesurface foot section is preferably reduced by at least 40% and thethickness of the surface foot section is preferably reduced by at least80% as the bed moves to the chair configuration. The width of thesurface foot section 1502 remains substantially the same in both the bedorientation and the chair orientation.

Pressure control in the surface foot section 1502 is illustrateddiagrammatically in FIG. 66. Each of the vertically collapsible bladders1620 are separately coupled to foot section control module 1514 bypressure/vacuum supply lines 1630 and sensor lines 1632. Therefore, eachof the three bladders 1620 are independently coupled to and controlledby foot section control module 1514. Each of the three horizontallycollapsing bladders 1618 are commonly connected to a commonpressure/vacuum source of the foot section control module as illustratedline 1634. A single sensor line 1636 is used to determine the pressurein the common zone of the interconnected bladders 1618. The controlconfiguration illustrated in FIG. 66 permits independent inflation anddeflation of bladders 1620 to provide heel pressure relief in footsection 1502. Details of the heel pressure management apparatus areillustrated in copending U.S. Pat. No. 5,666,681, filed Jan. 3, 1995,owned by the assignee of the present application, the disclosure ofwhich is hereby expressly incorporated by reference into the presentapplication.

Another embodiment of the foot section 1502 is illustrated in FIG. 67.In this embodiment, bladders 1618 have been replaced by diamond shapedbladders 1640. It is understood that any shape which collapses in aspecified direction upon deflation may be used in foot section 1502 ofthe present invention to provide the shortening or retracting andthinning or collapsing features discussed above.

Additional surface and treatment options of the modular air therapy andsupport surface apparatus are illustrated in FIG. 68. In FIG. 68, anupper air bladder 1506 is located on foam foundation base 1606 betweenside bolsters 1608 and 1610. Upper air bladder 1506 includes a pluralityof adjacent air tubes or bladders 1642 oriented transverse to alongitudinal axis of the bed. Illustratively, bladders 1642 areconnected in three commonly controlled zones 1644, 1646, and 1648. It isunderstood that more zones may be provided. If desired, each bladder1642 may be controlled independently.

The surface instrument module 1024 receives commands from the BACM 1018and the position sense module 1026 to reduce the pressure in a seatsection defined by zone 1644 of the upper air bladder 1506 as the bedmoves to the chair configuration in order to distribute a patient'sweight. A thigh section of the deck is angled upwardly to help maintainthe patient in a proper position on the seat when the bed is in thechair configuration.

For the upper surface decubitus prevention, the three supply tubes 1650of upper air bladder 1506 are all connected to a common pressure sourcethrough prevention module 1516. For the upper surface decubitustreatment, the three supply lines 1650 are coupled to three separatevalves in treatment module 1518 to control each of the zones 1644, 1646,and 1648 of upper air bladder 1506 independently.

A pulmonary rotation bladder 1508 is located between foundation base1606 and step deck 1596. It is understood that rotation bladder 1508 maybe positioned between foundation base 1606 and upper air bladder 1506 ifdesired. Rotation bladder 1508 includes separate bladders 1650 which areoriented to run parallel to a longitudinal axis of the hospital bed.Illustratively, three separate pressure zones 1652, 1654, and 1656 areprovided in rotation bladder 1508. In the illustrated embodiment, eachof the pressure zones 1652, 1654, and 1656 are independently controlledby pressure supply lines 1658. Each pressure supply line is coupled to aseparate valve in pulmonary control module 1520 illustrated in FIG. 61.A separate sensor line (not shown) for each zone 1652, 1654, and 1656 isalso coupled to pulmonary rotation control module 1520.

Pulmonary rotation bladder 1508 is stored in a deflated position withinthe bed until it is desired to treat the patient with rotationaltherapy. In this embodiment, the rotation bladder 1508 does not providea support surface for the patient. The support surface is provided byeither upper foam mattress 1504 or upper air bladder 1506. Therefore,rotation bladder 1508 can be stored flat in the bed during normaloperation of the bed as illustrated in FIG. 69. It is understood that inanother embodiment of the invention, the rotation bladder 1508 may benormally inflated to provide a support surface for the patient.

When it is desired to provide rotational treatment to the patient, apulmonary rotation control module 1520 is coupled to the bed. Thegraphical interactive display 1664 of the bed or the graphic caregiverinterface module 1032 automatically recognizes that the pulmonaryrotation control module 1520 is attached to the bed. Therefore, controlsfor the pulmonary rotation therapy device can be actuated from thegraphical interactive display 1664 or the graphic caregiver interface1032.

FIG. 69 illustrates the configuration of rotation bladder 1508 in itsdeflated position during normal operation of the bed with the upper foammattress 1504 in place of upper air bladder 1506. In FIG. 69, all threezones 1652, 1654, and 1656 of rotation bladder 1508 are deflated orflat.

FIG. 70 illustrates actuation of the rotation bladder 1508 to rotate apatient situated on foam mattress 1504 to the right. Pulmonary rotationcontrol module 1520 controls airflow to fully inflate zone 1656 topartially inflate zone 1654, and to deflate zone 1652 of rotationbladder 1508. FIG. 71 illustrates actuation of the rotation bladder 1508to rotate the patient to the left. Pulmonary rotation control module1520 fully inflates zone 1652, partially inflates zone 1656, anddeflates zone 1654 to rotate the patient.

Another embodiment of the modular therapy and support surface inventionis illustrated in FIG. 72. In this embodiment, separate exchangeablesurfaces are provided. The bed is illustrated by dotted line 1660. Asdiscussed above, the bed includes a peer-to-peer communication network1662 which is coupled to a graphical interactive display 1664. It isunderstood that graphical interactive display 1664 may be the graphiccaregiver interface module 1032 discussed above. In addition, graphicalinterface display 1664 may be a display with control switches embeddedin a foot board or at another location of the bed to provide a usercontrol for all therapy and surface options. As discussed above, thenetwork 1662 automatically recognizes when a specific therapy module isconnected to the bed 1660 and automatically provides control options tothe graphical interactive display 1664. The open architecture of theelectrical communication network 1662 allows interaction between theadded module and the graphical interactive display 1664 withoutredesigning the system. Bed 1660 includes a surface header connector1664 coupled to the air handling unit 1046 and to the electricalcommunication network 1662 by line 1668. In addition, bed 1660 includestherapy header connectors illustrated at block 1670 which are connectedto the air and power handling unit 1046 and to the electricalcommunication network 1662 as illustrated by line 1672.

In this embodiment of the present invention, separate surfaces areprovided, including a decubitus treatment surface 1674 and a separatedecubitus prevention surface 1676. The decubitus treatment surface 1674has its own attached control module 1678 for connecting to surfaceheader 1666. Decubitus prevention surface 1676 has its own controlmodule 1680 configured to be coupled to surface header connector 1666.Header connector 1666 is connected to modules 1678 or 1680 in a mannersimilar to module 1542 in FIG. 62.

Separate therapy modules are also provided. A pulmonary rotation therapysurface 1682 can be added to bed 1660. Rotation therapy surface 1682 iscoupled to its own control module 1684 which is configured to beconnected to therapy header connector 1670. A sequential compressiontherapy device 1686 is also provided. Sequential compression device 1686is coupled to its own control module 1688 which is configured to beconnected to therapy header connector 1670. The present inventionpermits the sequential compression device to use an on board airhandling unit 1046 and control system. This eliminates the requirementfor a separate air pump and control panel which takes up valuable floorspace near the bed and makes the bed difficult to move.

A separate pulmonary percussion and vibration therapy surface 1690 isalso provided. Pulmonary percussion and vibration therapy surface isadded to bed 1660 in place of a portion of the support surface of thebed. Pulmonary percussion and vibration therapy surface 1690 is coupledto its own control module 1692. Control module 1692 is configured to becoupled to a therapy header connector 1670.

The separate control modules are used to control power and airdistribution, and to control user options displayed on the graphicalinteractive display 1664 for each therapy or surface option. Asdiscussed above in detail with reference to FIG. 62, each control module1678, 1680, 1684, 1688 and 1692 contain valves, sensors, and electroniccontrol circuits specific to the particular surface or therapyapplication. All control features are implemented as a menu driveninteractive control for the selected therapy or surface module of thepresent invention on the graphical interface display 1664 or on thegraphic care giver interface 1023.

All surface related parameters can be transmitted from surfaceinstrument module 1024 to communications module 1020 and then to aremote location via the hospital network. Surface instrument 1024 can beinterrogated by a diagnostic tool coupled to accessory port 1016 ifdesired. Information related to the surface modules can also be receivedvia modem from a remote location through accessory port 1016.

Further details of the air support surfaces, the articulating deck, andthe control modules of the present invention are illustrated in FIG. 73.The support surface of the present invention is configured to bepositioned over a bed deck 402 of a hospital bed. The support surfaceincludes a surface foundation 1500 located on the bed deck 402. Aninflatable and deflatable surface foot section 1502 is located adjacentsurface foundation 1500. An upper air bladder 1506 is positioned oversurface foundation 1500.

As discussed above, the articulating deck includes separate,independently movable deck sections. Specifically, deck 402 includes ahead deck section 404, a seat deck section 406, a thigh deck section408, and a foot deck section 410. Upper air bladder 1506 includes aplurality of separate air bladders. The air bladders are preferablyconnected in three independently controlled air zones corresponding tothe different sections of deck 402. Specifically, air bladder 1506 isdivided into a head air zone 1648, a seat air zone 1646, and a air thighzone 1644. The separate surface foot section 1502 which overlies footdeck section 410 is also independently controlled.

An air surface control module 1517 is provided for selectively couplingthe various air zones 1644, 1646, and 1648 to the air handling unit1046. Air surface control module 1517 includes separate valves andpressure sensors for each air zone 1644, 1646, and 1648 of air bladder1506. When a command to move the bed deck is transmitted to the networkfrom a user input control on one of the standard caregiver interfacemodules 1028 and 1030, the graphic caregiver interface module 1032, orfrom another control device, the BACM 1018 actuates appropriatecylinders to articulate the deck 402. The BACM 1018 also providessignals to surface instrument module 1024 and air supply module 1014 forcontrolling inflation and deflation of the surface foot section 1502 andthe independent air zones 1644, 1646, and 1648 of upper air bladder 1506automatically as the bed articulates.

The surface instrument module 1024 sends signals to a controller insidethe air surface control module 1517 to open and close valves atpredetermined intervals to control inflation and deflation of the airzones 1649, 1646, and 1648. The surface instrument module 1024 and theair supply module 1014 also receive signals over the network from theposition sense module 1026 to indicate the position of the articulatingdeck sections 409, 406, 408 and 410.

As discussed above, the surface foot section 1502 is deflated as thedeck 402 moves to the chair position. In addition, seat air zone 1646and thigh air zone 1644 are partially deflated to distribute the weightof the person in the chair. When in the chair position, the surfacethigh bladder 1644 and the thigh deck section 408 support most of apatient's weight. This partial deflation of the chair seat section iscontrolled automatically by surface instrument module 1024, air supplymodule 1014, and air surface control module 1517 as the bed deck movesfrom the bed position of FIG. 1 to the chair position of FIG. 2. In someinstances, a single air bladder may be provided for seat air zone 1646and thigh air zone 1644. In other instances, a plurality of individualair zones may be all separately controlled. In other words, each of theair zones of air bladder 1506 may have several independently controlledair bladders 1642.

Separate valves and pressure sensors in air surface control module 1517are provided for interconnecting the various air zones 1644, 1646, and1648 to the communication network of the bed and to on-board airhandling unit 1046. The present invention also includes a foot sectioncontrol module 1514 which includes valves and pressure sensors for eachair zone of the surface foot section 1502.

Each of the control modules 1514, 1517 is designed to physically andfunctionally connect the various air zone bladders and to both thecommunication network of the hospital bed through the surface instrumentmodule 1024 and to the air handling unit 1046 which is controlled by airsupply module 1014. Air supply module 1014 is coupled to thepeer-to-peer communication network. Air supply electronics 1528 areconnected to air supply module 1014 for controlling air handling unit1046 and switching valve 1530 based on network commands for controllingthe various surface and treatment modules illustrated in FIG. 73.

Air handling unit 1046 is configured to supply air under pressure toswitching valve 1530 on line 1532. Air handling unit 1046 also applies avacuum to switching valve 1530 through line 1534. An output of switchingvalve 1530 is coupled to a connector block 1536. Connector block 1536provides an air and vacuum supply line 1515 to the foot section controlmodule 1514 and provides an air and vacuum supply line 1519 to the airsurface control module 1517. It is understood that dual control linesfor both air and vacuum can be supplied to each of the foot sectioncontrol module 1514 and the air surface control module 1517. This dualcontrol allows each module to apply pressure and vacuum simultaneouslyto different zones of a bladder or treatment device.

The surface instrument module 1024 receives commands from the BACM 1018and the position sense module 1026 to control the air surface controlmodule 1517 to reduce the pressure in a seat section defined by zones1644 and 1646 of the upper air bladder 1506 automatically as the bedmoves to the chair configuration in order to distribute a patient'sweight. An end of the thigh deck section 408 closest to foot end 54 isangled upwardly automatically as illustrated in FIG. 8 to help maintainthe patient in a proper position on the seat when the bed is in thechair configuration.

Although the invention has been described in detail with reference topreferred embodiments, variations and modifications exist within thescope and spirit of the invention as described and defined in thefollowing claims.

1. A bed comprising: a plurality of casters, a base supported by thecasters, a frame supported by the base, a lift mechanism configured toraise and lower the frame relative to the base, a deck supported by theframe, the deck including movable sections configurable for the deck toassume at least a bed position and a chair position, the movablesections including at least a head section and a foot section spacedfrom the head section, the foot section being movable from an upposition to a down position, the foot section having a longitudinallength, the foot section comprising a pivoting member pivotable relativeto the frame, and a contracting member coupled to the pivoting member,the contracting member including a pair of spaced-apart longitudinaltabs, the longitudinal tabs being movable between an expanded positionand a contracted position, wherein the longitudinal length of the footsection is shorter when the longitudinal tabs are in the contractedposition and the longitudinal length of the foot section is longer whenthe longitudinal tabs are in the expanded position.
 2. The bed of claim1, wherein the pivoting member is pivotably coupled to another sectionof the deck located between the head section and the foot section. 3.The bed of claim 2, wherein the pivoting member is pivotably coupled toa thigh section of the deck.
 4. The bed of claim 2, wherein the pivotingmember is pivotable between a generally horizontal position and agenerally vertically downwardly extending position.
 5. The bed of claim1, wherein the pivoting member has a top surface and the contractingmember is positioned to slide across the top surface of the pivotingmember.
 6. The bed of claim 5, wherein the contracting member slideslongitudinally relative to the pivoting member.
 7. The bed of claim 1,wherein the foot section is parallel to the frame when the foot sectionis in the up position.
 8. The bed of claim 7, wherein the foot sectionis at an angle relative to the frame when the foot section is in thedown position.
 9. A bed comprising: a frame, a deck configured tosupport a mattress in at least a horizontal position and a chairposition, the deck comprising at least a head section and a foot sectionspaced from the head section, the head and foot sections each beingmovable relative to the frame, the foot section having a longitudinallength, the foot section comprising a pivoting member pivotable betweena horizontal position and a non-horizontal position, a contractingmember coupled to the pivoting member, the contracting member beingslidable relative to the pivoting member to expand and contract thelongitudinal length of the foot section, a first actuator configured topivot the pivoting member relative to the frame, and a second actuatorspaced from the first actuator, the second actuator being movable forexpanding and contracting the longitudinal length of the foot section.10. The bed of claim 9, wherein the pivoting member slidably receivesthe contracting member.
 11. The bed of claim 9, wherein the contractingmember is slidable toward the head section to reduce the length of thefoot section.
 12. The bed of claim 9, wherein the contracting member isslidable away from the head section to extend the length of the footsection.
 13. The bed of claim 12, wherein the contracting member isslidable toward the head section when the pivoting member is in thehorizontal position.
 14. The bed of claim 9, wherein the contractingmember is slidable inwardly toward the head section.
 15. The bed ofclaim 9, wherein the pivoting member has a top surface and thecontracting member is positioned to slide relative to the top surface ofthe pivoting member.
 16. A bed comprising: a frame, a deck configured tosupport a mattress in at least a horizontal position and a chairposition, the deck comprising at least a head section and a foot sectionspaced from the head section, the head and foot sections each beingmovable relative to the frame, the foot section having at least a firstlength and a second length greater than the first length, the footsection comprising a pivoting member pivotable from a horizontalposition to a non-horizontal position relative to the frame, acontracting member arranged to contract toward and expand away from thepivoting member in a longitudinal direction relative to the pivotingmember, the contracting member including at least one tab, and at leastone channel located in the pivoting member, the at least one tab of thecontracting member cooperating with the at least one channel of thepivoting member as the contracting member moves relative to the pivotingmember to change the length of the foot section.
 17. The bed of claim16, wherein each tab of the contracting member is slidable in a channelof the pivoting member.
 18. The bed of claim 16, wherein each channel ofthe pivoting member receives a tab of the contracting member.
 19. Thebed of claim 16, wherein the at least one tab is nearer to the headsection when the foot section has the first length and the at least onetab is further away from the head section when the foot section has thesecond length.
 20. The bed of claim 16, wherein the contracting memberis formed to include a pair of spaced apart longitudinal tabs and thepivoting member includes a pair of spaced apart longitudinal channelsthat cooperate with the longitudinal tabs.